-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar 23 17:04:20 2023 UTC
-- File 20211129_120221_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.951290 to 45.321509
-each separately from 9s to 45s and from 321220us to 951579us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.407694 in 1 packets: 18
      11.374789 in 1 packets: 36
      10.586580 in 1 packets: 54
MIU packet time from 2021-11-29T12:02:33.951.290.000 to 2021-11-29T12:03:09.321.509.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.407694 in 1 packets: 18
      11.374789 in 1 packets: 36
      10.586580 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.7886386 to 46.7885166
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999525 in 1 packets: 18
      10.9999627 in 1 packets: 36
      10.9999628 in 1 packets: 54
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C0 -> 35.6 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 72 packets (all packets)
Number of generated alerts: 
      0 in 72 packets (all packets)
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20211129_120221_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      56 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.7399291 to 11.7402181
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.4076470  -- OK for SL --
   MIU time from 2021-11-29T12:02:33.951.290.000 to 2021-11-29T12:02:33.951.579.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.407694

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.1478651 to 25.1481541
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.3747510  -- OK for SL --
   MIU time from 2021-11-29T12:02:47.359.273.000 to 2021-11-29T12:02:47.359.562.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.374789

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      66 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.5229051 to 36.5231941
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.5865430  -- OK for SL --
   MIU time from 2021-11-29T12:02:58.734.351.000 to 2021-11-29T12:02:58.734.640.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.586580

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      64 in 18 packets (all packets)  -- OK --
   10 MHz counter from 47.1097371 to 47.1100261
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-11-29T12:03:09.321.220.000 to 2021-11-29T12:03:09.321.509.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.739929100 corrected by 40bit 10MHz counter of 10.788638600 obtained at 1Hz counter of 9. Result: 9.951293734
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-11-29T12:02:33.951.149.725
   Max 1st sample MIU time among all SL packets: 2021-11-29T12:03:09.321.368.725
   Min 1st sample XO time among all SL packets: 9.951153459
   Max 1st sample XO time among all SL packets: 45.321370318
   TM values from -187 to 204 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-11-29T12:02:33.951.149.725 (XO: 9.951153459 - 9.951459460, t=35.6 degC)  Rank: 00056
   TM values: Max =   190  Min =  -176  Ave =    -4.46  Std =    15.98
   Min TM value:-176 at an offset of 117.475 usec -> 2021-11-29T12:02:33.951.267.200 (XO: 9.951270934)
   Max TM value:190 at an offset of 114.925 usec -> 2021-11-29T12:02:33.951.264.650 (XO: 9.951268384)
   Max dTM/dt [1/us]:1.51e+03 at an offset of 114.838 usec -> 2021-11-29T12:02:33.951.264.562 (XO: 9.951268297)
   Min dTM/dt [1/us]:-1.62e+03 at an offset of 115.013 usec -> 2021-11-29T12:02:33.951.264.737 (XO: 9.951268472)

 - Block number: 2 starting at 2021-11-29T12:02:47.359.132.725 (XO: 23.359134946 - 23.359440947, t=35.6 degC)  Rank: 00076
   TM values: Max =   106  Min =  -179  Ave =    -4.65  Std =    17.78
   Min TM value:-179 at an offset of 128.288 usec -> 2021-11-29T12:02:47.359.261.012 (XO: 23.359263234)
   Max TM value:106 at an offset of 131.675 usec -> 2021-11-29T12:02:47.359.264.400 (XO: 23.359266622)
   Max dTM/dt [1/us]:2.76e+02 at an offset of 237.976 usec -> 2021-11-29T12:02:47.359.370.700 (XO: 23.359372922)
   Min dTM/dt [1/us]:-3.61e+02 at an offset of 128.075 usec -> 2021-11-29T12:02:47.359.260.800 (XO: 23.359263021)

 - Block number: 3 starting at 2021-11-29T12:02:58.734.210.725 (XO: 34.734213521 - 34.734519522, t=35.6 degC)  Rank: 00066
   TM values: Max =   200  Min =  -168  Ave =    -4.56  Std =    14.39
   Min TM value:-168 at an offset of 128.500 usec -> 2021-11-29T12:02:58.734.339.225 (XO: 34.734342022)
   Max TM value:200 at an offset of 128.250 usec -> 2021-11-29T12:02:58.734.338.975 (XO: 34.734341772)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 128.200 usec -> 2021-11-29T12:02:58.734.338.925 (XO: 34.734341722)
   Min dTM/dt [1/us]:-1.80e+03 at an offset of 130.663 usec -> 2021-11-29T12:02:58.734.341.387 (XO: 34.734344184)

 - Block number: 4 starting at 2021-11-29T12:03:09.321.079.725 (XO: 45.321081317 - 45.321387318, t=35.6 degC)  Rank: 00064
   TM values: Max =   204  Min =  -187  Ave =    -4.57  Std =    19.77
   Min TM value:-187 at an offset of 129.850 usec -> 2021-11-29T12:03:09.321.209.575 (XO: 45.321211167)
   Max TM value:204 at an offset of 129.600 usec -> 2021-11-29T12:03:09.321.209.325 (XO: 45.321210917)
   Max dTM/dt [1/us]:1.70e+03 at an offset of 129.513 usec -> 2021-11-29T12:03:09.321.209.237 (XO: 45.321210830)
   Min dTM/dt [1/us]:-2.11e+03 at an offset of 129.738 usec -> 2021-11-29T12:03:09.321.209.462 (XO: 45.321211055)
