-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar 23 16:59:27 2023 UTC
-- File 20211129_031454_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.815682 to 64.943592
-each separately from 2s to 64s and from 762172us to 943592us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.946201 in 1 packets: 18
      13.063291 in 1 packets: 36
      8.033690 in 1 packets: 54
      15.083283 in 1 packets: 72
MIU packet time from 2021-11-29T03:14:59.815.682.000 to 2021-11-29T03:16:01.943.592.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.946201 in 1 packets: 18
      13.063291 in 1 packets: 36
      8.033690 in 1 packets: 54
      15.083283 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.8618224 to 65.8616105
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.9999111 in 1 packets: 18
      12.9999555 in 1 packets: 36
      7.9999727 in 1 packets: 54
      14.9999488 in 1 packets: 72
Temperature FPGA: 
      05 -> 48.5 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      06 -> 47.2 degC in 36 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, ...
Temperature XO: 
      C0 -> 35.6 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 73 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Number of generated alerts: 
      0 in 73 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20211129_031454_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      47 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.6775051 to 4.6777941
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.9461110  -- OK for SL --
   MIU time from 2021-11-29T03:14:59.815.682.000 to 2021-11-29T03:14:59.815.971.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.946201

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      50 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.6239051 to 30.6241941
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.0632470  -- OK for SL --
   MIU time from 2021-11-29T03:15:25.762.172.000 to 2021-11-29T03:15:25.762.461.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.063291

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      57 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.6874411 to 43.6877301
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.0336630  -- OK for SL --
   MIU time from 2021-11-29T03:15:38.825.752.000 to 2021-11-29T03:15:38.826.041.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.033690

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      49 in 18 packets (all packets)  -- OK --
   10 MHz counter from 51.7213931 to 51.7216821
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0832310  -- OK for SL --
   MIU time from 2021-11-29T03:15:46.859.731.000 to 2021-11-29T03:15:46.860.020.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.083283

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 66.8049131 to 66.8052021
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-11-29T03:16:01.943.303.000 to 2021-11-29T03:16:01.943.592.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.677505100 corrected by 40bit 10MHz counter of 3.861822400 obtained at 1Hz counter of 2. Result: 2.815685473
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-11-29T03:14:59.815.541.725
   Max 1st sample MIU time among all SL packets: 2021-11-29T03:16:01.943.451.725
   Min 1st sample XO time among all SL packets: 2.815545198
   Max 1st sample XO time among all SL packets: 64.943454533
   TM values from -184 to 376 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-11-29T03:14:59.815.541.725 (XO: 2.815545198 - 2.815851199, t=35.6 degC)  Rank: 00047
   TM values: Max =   138  Min =  -151  Ave =    -4.58  Std =    14.80
   Min TM value:-151 at an offset of 122.313 usec -> 2021-11-29T03:14:59.815.664.037 (XO: 2.815667511)
   Max TM value:138 at an offset of 122.538 usec -> 2021-11-29T03:14:59.815.664.262 (XO: 2.815667736)
   Max dTM/dt [1/us]:1.44e+03 at an offset of 122.450 usec -> 2021-11-29T03:14:59.815.664.175 (XO: 2.815667648)
   Min dTM/dt [1/us]:-1.29e+03 at an offset of 119.675 usec -> 2021-11-29T03:14:59.815.661.400 (XO: 2.815664873)

 - Block number: 2 starting at 2021-11-29T03:15:25.762.031.725 (XO: 28.762033916 - 28.762339917, t=35.6 degC)  Rank: 00050
   TM values: Max =   159  Min =  -148  Ave =    -4.62  Std =    13.54
   Min TM value:-148 at an offset of 111.313 usec -> 2021-11-29T03:15:25.762.143.037 (XO: 28.762145229)
   Max TM value:159 at an offset of 111.050 usec -> 2021-11-29T03:15:25.762.142.775 (XO: 28.762144966)
   Max dTM/dt [1/us]:1.27e+03 at an offset of 110.975 usec -> 2021-11-29T03:15:25.762.142.700 (XO: 28.762144891)
   Min dTM/dt [1/us]:-1.44e+03 at an offset of 111.175 usec -> 2021-11-29T03:15:25.762.142.900 (XO: 28.762145091)

 - Block number: 3 starting at 2021-11-29T03:15:38.825.611.725 (XO: 41.825614632 - 41.825920633, t=35.6 degC)  Rank: 00057
   TM values: Max =   179  Min =  -184  Ave =    -4.58  Std =    22.01
   Min TM value:-184 at an offset of 108.150 usec -> 2021-11-29T03:15:38.825.719.875 (XO: 41.825722782)
   Max TM value:179 at an offset of 110.263 usec -> 2021-11-29T03:15:38.825.721.987 (XO: 41.825724895)
   Max dTM/dt [1/us]:1.52e+03 at an offset of 110.163 usec -> 2021-11-29T03:15:38.825.721.887 (XO: 41.825724795)
   Min dTM/dt [1/us]:-1.84e+03 at an offset of 110.375 usec -> 2021-11-29T03:15:38.825.722.100 (XO: 41.825725007)

 - Block number: 4 starting at 2021-11-29T03:15:46.859.590.725 (XO: 49.859594048 - 49.859900049, t=35.6 degC)  Rank: 00049
   TM values: Max =   154  Min =  -148  Ave =    -4.66  Std =    13.48
   Min TM value:-148 at an offset of 119.488 usec -> 2021-11-29T03:15:46.859.710.212 (XO: 49.859713536)
   Max TM value:154 at an offset of 121.638 usec -> 2021-11-29T03:15:46.859.712.362 (XO: 49.859715686)
   Max dTM/dt [1/us]:1.44e+03 at an offset of 119.638 usec -> 2021-11-29T03:15:46.859.710.362 (XO: 49.859713686)
   Min dTM/dt [1/us]:-1.30e+03 at an offset of 121.738 usec -> 2021-11-29T03:15:46.859.712.462 (XO: 49.859715786)

 - Block number: 5 starting at 2021-11-29T03:16:01.943.162.725 (XO: 64.943165532 - 64.943471533, t=35.6 degC)  Rank: 00086
   TM values: Max =   376  Min =  -135  Ave =    -4.64  Std =    26.46
   Min TM value:-135 at an offset of 111.500 usec -> 2021-11-29T03:16:01.943.274.225 (XO: 64.943277032)
   Max TM value:376 at an offset of 108.900 usec -> 2021-11-29T03:16:01.943.271.625 (XO: 64.943274432)
   Max dTM/dt [1/us]:1.83e+03 at an offset of 108.813 usec -> 2021-11-29T03:16:01.943.271.537 (XO: 64.943274345)
   Min dTM/dt [1/us]:-1.86e+03 at an offset of 107.138 usec -> 2021-11-29T03:16:01.943.269.862 (XO: 64.943272670)
