-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Feb 15 04:37:45 2022 UTC
-- File 20211114_093155_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.538370 to 19.356826
-each separately from 3s to 19s and from 356537us to 538659us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.817878 in 1 packets: 18
MIU packet time from 2021-11-14T09:32:01.538.370.000 to 2021-11-14T09:32:17.356.826.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.817878 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.7289676 to 20.7289123
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999447 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 36 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 3 packets: 19, 20, 21
      2 in 6 packets: 22, 23, 24, 25, 26, 27
      3 in 8 packets: 28, 29, 30, 31, 32, 33, 34, 35
      4 in 1 packets: 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20211114_093155_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      53 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.2673381 to 5.2676271
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.8178230  -- OK for SL --
   MIU time from 2021-11-14T09:32:01.538.370.000 to 2021-11-14T09:32:01.538.659.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.817878

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      181 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.0854501 to 21.0857391
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-11-14T09:32:17.356.537.000 to 2021-11-14T09:32:17.356.826.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.267338100 corrected by 40bit 10MHz counter of 4.728967600 obtained at 1Hz counter of 3. Result: 3.538372384
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2021-11-14T09:32:01.538.229.725
   Max 1st sample MIU time among all SL packets: 2021-11-14T09:32:17.356.685.725
   Min 1st sample XO time among all SL packets: 3.538232109
   Max 1st sample XO time among all SL packets: 19.356687773
   TM values from -350 to 417 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-11-14T09:32:01.538.229.725 (XO: 3.538232109 - 3.538538110, t=35.6 degC)  Rank: 00053
   TM values: Max =   173  Min =  -149  Ave =    -4.55  Std =    14.27
   Min TM value:-149 at an offset of 106.300 usec -> 2021-11-14T09:32:01.538.336.025 (XO: 3.538338409)
   Max TM value:173 at an offset of 108.500 usec -> 2021-11-14T09:32:01.538.338.225 (XO: 3.538340609)
   Max dTM/dt [1/us]:1.49e+03 at an offset of 108.400 usec -> 2021-11-14T09:32:01.538.338.125 (XO: 3.538340509)
   Min dTM/dt [1/us]:-1.38e+03 at an offset of 108.600 usec -> 2021-11-14T09:32:01.538.338.325 (XO: 3.538340709)

 - Block number: 2 starting at 2021-11-14T09:32:17.356.396.725 (XO: 19.356398772 - 19.356704773, t=35.6 degC)  Rank: 00181
   TM values: Max =   417  Min =  -350  Ave =    -4.65  Std =    68.46
   Min TM value:-350 at an offset of 191.313 usec -> 2021-11-14T09:32:17.356.588.038 (XO: 19.356590086)
   Max TM value:417 at an offset of 188.463 usec -> 2021-11-14T09:32:17.356.585.188 (XO: 19.356587236)
   Max dTM/dt [1/us]:1.61e+03 at an offset of 188.363 usec -> 2021-11-14T09:32:17.356.585.088 (XO: 19.356587136)
   Min dTM/dt [1/us]:-9.84e+02 at an offset of 189.176 usec -> 2021-11-14T09:32:17.356.585.900 (XO: 19.356587948)
