-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Feb 14 23:31:54 2022 UTC
-- File 20211113_194958_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.060433 to 18.803579
-each separately from 11s to 18s and from 60433us to 803579us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.742568 in 1 packets: 18
MIU packet time from 2021-11-13T19:50:12.060.433.000 to 2021-11-13T19:50:19.803.579.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.742568 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.7195516 to 19.7195274
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.9999758 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 36 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 3 packets: 19, 20, 21
      3 in 15 packets: 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20211113_194958_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      162 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.7799858 to 12.7802748
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.7425430  -- OK for SL --
   MIU time from 2021-11-13T19:50:12.060.433.000 to 2021-11-13T19:50:12.060.722.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.742568

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      172 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.5228178 to 20.5231068
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-11-13T19:50:19.803.290.000 to 2021-11-13T19:50:19.803.579.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 12.779985800 corrected by 40bit 10MHz counter of 12.719551600 obtained at 1Hz counter of 11. Result: 11.060434405
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-11-13T19:50:12.060.292.725
   Max 1st sample MIU time among all SL packets: 2021-11-13T19:50:19.803.438.725
   Min 1st sample XO time among all SL packets: 11.060294130
   Max 1st sample XO time among all SL packets: 18.803441937
   TM values from -714 to 527 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-11-13T19:50:12.060.292.725 (XO: 11.060294130 - 11.060600131, t=35.6 degC)  Rank: 00162
   TM values: Max =   370  Min =  -426  Ave =    -4.49  Std =    58.85
   Min TM value:-426 at an offset of 90.288 usec -> 2021-11-13T19:50:12.060.383.012 (XO: 11.060384418)
   Max TM value:370 at an offset of 91.988 usec -> 2021-11-13T19:50:12.060.384.712 (XO: 11.060386118)
   Max dTM/dt [1/us]:4.35e+03 at an offset of 4.675 usec -> 2021-11-13T19:50:12.060.297.400 (XO: 11.060298805)
   Min dTM/dt [1/us]:-3.85e+03 at an offset of 103.738 usec -> 2021-11-13T19:50:12.060.396.462 (XO: 11.060397868)

 - Block number: 2 starting at 2021-11-13T19:50:19.803.149.725 (XO: 18.803152936 - 18.803458937, t=35.6 degC)  Rank: 00172
   TM values: Max =   527  Min =  -714  Ave =    -4.23  Std =    64.91
   Min TM value:-714 at an offset of 119.413 usec -> 2021-11-13T19:50:19.803.269.137 (XO: 18.803272349)
   Max TM value:527 at an offset of 119.813 usec -> 2021-11-13T19:50:19.803.269.537 (XO: 18.803272749)
   Max dTM/dt [1/us]:6.78e+03 at an offset of 62.363 usec -> 2021-11-13T19:50:19.803.212.087 (XO: 18.803215299)
   Min dTM/dt [1/us]:-5.76e+03 at an offset of 235.501 usec -> 2021-11-13T19:50:19.803.385.225 (XO: 18.803388437)
