-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Feb 14 23:10:48 2022 UTC
-- File 20211113_193950_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.389953 to 29.913704
-each separately from 0s to 29s and from 389953us to 913704us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      29.523173 in 1 packets: 18
MIU packet time from 2021-11-13T19:39:53.389.953.000 to 2021-11-13T19:40:22.913.704.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      29.523173 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.5959904 to 30.5958896
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      28.9998992 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20211113_193950_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      105 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.9859444 to 1.9862334
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 29.5230710  -- OK for SL --
   MIU time from 2021-11-13T19:39:53.389.953.000 to 2021-11-13T19:39:53.390.242.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 29.523173

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      57 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.5093044 to 31.5095934
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-11-13T19:40:22.913.415.000 to 2021-11-13T19:40:22.913.704.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.985944400 corrected by 40bit 10MHz counter of 1.595990400 obtained at 1Hz counter of 0. Result: 0.389955365
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2021-11-13T19:39:53.389.812.725
   Max 1st sample MIU time among all SL packets: 2021-11-13T19:40:22.913.563.725
   Min 1st sample XO time among all SL packets: 0.389815089
   Max 1st sample XO time among all SL packets: 29.913566722
   TM values from -225 to 465 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-11-13T19:39:53.389.812.725 (XO: 0.389815089 - 0.390121090, t=35.6 degC)  Rank: 00105
   TM values: Max =   414  Min =  -225  Ave =    -4.40  Std =    26.62
   Min TM value:-225 at an offset of 134.075 usec -> 2021-11-13T19:39:53.389.946.800 (XO: 0.389949165)
   Max TM value:414 at an offset of 131.625 usec -> 2021-11-13T19:39:53.389.944.350 (XO: 0.389946715)
   Max dTM/dt [1/us]:3.35e+03 at an offset of 133.088 usec -> 2021-11-13T19:39:53.389.945.812 (XO: 0.389948177)
   Min dTM/dt [1/us]:-3.72e+03 at an offset of 113.263 usec -> 2021-11-13T19:39:53.389.925.987 (XO: 0.389928352)

 - Block number: 2 starting at 2021-11-13T19:40:22.913.274.725 (XO: 29.913277721 - 29.913583723, t=35.6 degC)  Rank: 00057
   TM values: Max =   465  Min =  -200  Ave =    -4.35  Std =    25.52
   Min TM value:-200 at an offset of 118.300 usec -> 2021-11-13T19:40:22.913.393.025 (XO: 29.913396022)
   Max TM value:465 at an offset of 125.513 usec -> 2021-11-13T19:40:22.913.400.237 (XO: 29.913403234)
   Max dTM/dt [1/us]:3.10e+03 at an offset of 125.425 usec -> 2021-11-13T19:40:22.913.400.150 (XO: 29.913403147)
   Min dTM/dt [1/us]:-2.85e+03 at an offset of 118.175 usec -> 2021-11-13T19:40:22.913.392.900 (XO: 29.913395897)
