-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Feb 14 22:39:44 2022 UTC
-- File 20211113_182203_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.112599 to 17.527546
-each separately from 12s to 17s and from 112599us to 527546us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.414369 in 1 packets: 18
MIU packet time from 2021-11-13T18:22:18.112.599.000 to 2021-11-13T18:22:23.527.546.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.414369 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.8536850 to 18.8536675
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.9999825 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      4 in 1 packets: 1
      5 in 5 packets: 2, 3, 4, 5, 6
      6 in 13 packets: 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      7 in 5 packets: 20, 21, 22, 23, 24
      8 in 12 packets: 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20211113_182203_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      148 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.9662846 to 13.9665736
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.4143510  -- OK for SL --
   MIU time from 2021-11-13T18:22:18.112.599.000 to 2021-11-13T18:22:18.112.888.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.414369

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      137 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.3809246 to 19.3812136
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-11-13T18:22:23.527.257.000 to 2021-11-13T18:22:23.527.546.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 13.966284600 corrected by 40bit 10MHz counter of 13.853685000 obtained at 1Hz counter of 12. Result: 12.112599994
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-11-13T18:22:18.112.458.725
   Max 1st sample MIU time among all SL packets: 2021-11-13T18:22:23.527.405.725
   Min 1st sample XO time among all SL packets: 12.112459719
   Max 1st sample XO time among all SL packets: 17.527407618
   TM values from -1153 to 424 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-11-13T18:22:18.112.458.725 (XO: 12.112459719 - 12.112765720, t=35.6 degC)  Rank: 00148
   TM values: Max =   335  Min =  -380  Ave =    -4.50  Std =    54.20
   Min TM value:-380 at an offset of 101.025 usec -> 2021-11-13T18:22:18.112.559.750 (XO: 12.112560744)
   Max TM value:335 at an offset of 140.113 usec -> 2021-11-13T18:22:18.112.598.837 (XO: 12.112599832)
   Max dTM/dt [1/us]:4.00e+03 at an offset of 119.663 usec -> 2021-11-13T18:22:18.112.578.387 (XO: 12.112579382)
   Min dTM/dt [1/us]:-4.20e+03 at an offset of 132.613 usec -> 2021-11-13T18:22:18.112.591.337 (XO: 12.112592332)

 - Block number: 2 starting at 2021-11-13T18:22:23.527.116.725 (XO: 17.527118617 - 17.527424618, t=35.6 degC)  Rank: 00137
   TM values: Max =   424  Min = -1153  Ave =    -4.24  Std =    66.77
   Min TM value:-1153 at an offset of 125.800 usec -> 2021-11-13T18:22:23.527.242.525 (XO: 17.527244418)
   Max TM value:424 at an offset of 132.650 usec -> 2021-11-13T18:22:23.527.249.375 (XO: 17.527251268)
   Max dTM/dt [1/us]:6.14e+03 at an offset of 111.575 usec -> 2021-11-13T18:22:23.527.228.300 (XO: 17.527230193)
   Min dTM/dt [1/us]:-6.00e+03 at an offset of 125.625 usec -> 2021-11-13T18:22:23.527.242.350 (XO: 17.527244243)
