-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Feb 14 20:10:35 2022 UTC
-- File 20211113_142614_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.200572 to 24.488049
-each separately from 9s to 24s and from 200572us to 488049us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.286899 in 1 packets: 18
MIU packet time from 2021-11-13T14:26:26.200.572.000 to 2021-11-13T14:26:41.488.049.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.286899 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.9092767 to 25.9092244
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999477 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      07 -> 45.9 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20211113_142614_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      92 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.1098489 to 11.1101379
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.2868470  -- OK for SL --
   MIU time from 2021-11-13T14:26:26.200.572.000 to 2021-11-13T14:26:26.200.861.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.286899

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      58 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.3969849 to 26.3972739
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-11-13T14:26:41.487.760.000 to 2021-11-13T14:26:41.488.049.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.109848900 corrected by 40bit 10MHz counter of 10.909276700 obtained at 1Hz counter of 9. Result: 9.200572902
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-11-13T14:26:26.200.431.725
   Max 1st sample MIU time among all SL packets: 2021-11-13T14:26:41.487.908.725
   Min 1st sample XO time among all SL packets: 9.200432627
   Max 1st sample XO time among all SL packets: 24.487910884
   TM values from -311 to 263 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-11-13T14:26:26.200.431.725 (XO: 9.200432627 - 9.200738628, t=35.6 degC)  Rank: 00092
   TM values: Max =   263  Min =  -311  Ave =    -4.45  Std =    33.24
   Min TM value:-311 at an offset of 112.163 usec -> 2021-11-13T14:26:26.200.543.887 (XO: 9.200544789)
   Max TM value:263 at an offset of 128.613 usec -> 2021-11-13T14:26:26.200.560.337 (XO: 9.200561239)
   Max dTM/dt [1/us]:2.98e+03 at an offset of 167.926 usec -> 2021-11-13T14:26:26.200.599.650 (XO: 9.200600552)
   Min dTM/dt [1/us]:-3.17e+03 at an offset of 168.013 usec -> 2021-11-13T14:26:26.200.599.738 (XO: 9.200600640)

 - Block number: 2 starting at 2021-11-13T14:26:41.487.619.725 (XO: 24.487621883 - 24.487927884, t=35.6 degC)  Rank: 00058
   TM values: Max =   190  Min =  -184  Ave =    -4.55  Std =    14.09
   Min TM value:-184 at an offset of 124.925 usec -> 2021-11-13T14:26:41.487.744.650 (XO: 24.487746808)
   Max TM value:190 at an offset of 127.188 usec -> 2021-11-13T14:26:41.487.746.912 (XO: 24.487749071)
   Max dTM/dt [1/us]:1.65e+03 at an offset of 127.088 usec -> 2021-11-13T14:26:41.487.746.812 (XO: 24.487748971)
   Min dTM/dt [1/us]:-1.36e+03 at an offset of 127.275 usec -> 2021-11-13T14:26:41.487.747.000 (XO: 24.487749158)
