-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Feb 14 00:42:48 2022 UTC
-- File 20211108_131132_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.353355 to 66.975538
-each separately from 5s to 66s and from 352486us to 975538us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.400036 in 1 packets: 18
      19.598517 in 1 packets: 36
      18.472094 in 1 packets: 54
      8.150091 in 1 packets: 72
MIU packet time from 2021-11-08T13:11:40.353.355.000 to 2021-11-08T13:12:41.975.538.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.400036 in 1 packets: 18
      19.598517 in 1 packets: 36
      18.472094 in 1 packets: 54
      8.150091 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.6263525 to 67.6261394
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999477 in 1 packets: 18
      19.9999301 in 1 packets: 36
      17.9999370 in 1 packets: 54
      7.9999721 in 1 packets: 72
Temperature FPGA: 
      05 -> 48.5 degC in 90 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 1 packets: 37
      2 in 53 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 1 packets: 37
      2 in 53 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20211108_131132_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      72 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.9797087 to 6.9799977
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.3999830  -- OK for SL --
   MIU time from 2021-11-08T13:11:40.353.355.000 to 2021-11-08T13:11:40.353.644.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.400036

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      67 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.3799807 to 22.3802697
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.5984470  -- OK for SL --
   MIU time from 2021-11-08T13:11:55.753.680.000 to 2021-11-08T13:11:55.753.969.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.598517

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.9787167 to 41.9790057
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.4720310  -- OK for SL --
   MIU time from 2021-11-08T13:12:15.352.486.000 to 2021-11-08T13:12:15.352.775.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.472094

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      64 in 18 packets (all packets)  -- OK --
   10 MHz counter from 60.4510367 to 60.4513257
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.1500630  -- OK for SL --
   MIU time from 2021-11-08T13:12:33.824.869.000 to 2021-11-08T13:12:33.825.158.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.150091

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      69 in 18 packets (all packets)  -- OK --
   10 MHz counter from 68.6013887 to 68.6016777
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-11-08T13:12:41.975.249.000 to 2021-11-08T13:12:41.975.538.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 6.979708700 corrected by 40bit 10MHz counter of 6.626352500 obtained at 1Hz counter of 5. Result: 5.353357437
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-11-08T13:11:40.353.214.725
   Max 1st sample MIU time among all SL packets: 2021-11-08T13:12:41.975.397.725
   Min 1st sample XO time among all SL packets: 5.353217161
   Max 1st sample XO time among all SL packets: 66.975401439
   TM values from -507 to 774 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-11-08T13:11:40.353.214.725 (XO: 5.353217161 - 5.353523162, t=35.6 degC)  Rank: 00072
   TM values: Max =   173  Min =  -224  Ave =    -4.46  Std =    17.49
   Min TM value:-224 at an offset of 113.725 usec -> 2021-11-08T13:11:40.353.328.450 (XO: 5.353330887)
   Max TM value:173 at an offset of 113.363 usec -> 2021-11-08T13:11:40.353.328.087 (XO: 5.353330524)
   Max dTM/dt [1/us]:1.42e+03 at an offset of 116.100 usec -> 2021-11-08T13:11:40.353.330.825 (XO: 5.353333262)
   Min dTM/dt [1/us]:-1.68e+03 at an offset of 113.475 usec -> 2021-11-08T13:11:40.353.328.200 (XO: 5.353330637)

 - Block number: 2 starting at 2021-11-08T13:11:55.753.539.725 (XO: 20.753542938 - 20.753848939, t=35.6 degC)  Rank: 00067
   TM values: Max =   200  Min =  -215  Ave =    -4.49  Std =    18.37
   Min TM value:-215 at an offset of 116.113 usec -> 2021-11-08T13:11:55.753.655.837 (XO: 20.753659051)
   Max TM value:200 at an offset of 115.750 usec -> 2021-11-08T13:11:55.753.655.475 (XO: 20.753658688)
   Max dTM/dt [1/us]:1.59e+03 at an offset of 118.388 usec -> 2021-11-08T13:11:55.753.658.112 (XO: 20.753661326)
   Min dTM/dt [1/us]:-1.94e+03 at an offset of 115.863 usec -> 2021-11-08T13:11:55.753.655.587 (XO: 20.753658801)

 - Block number: 3 starting at 2021-11-08T13:12:15.352.345.725 (XO: 40.352347358 - 40.352653359, t=35.6 degC)  Rank: 00088
   TM values: Max =   774  Min =  -507  Ave =    -4.40  Std =    64.12
   Min TM value:-507 at an offset of 232.451 usec -> 2021-11-08T13:12:15.352.578.175 (XO: 40.352579809)
   Max TM value:774 at an offset of 212.076 usec -> 2021-11-08T13:12:15.352.557.800 (XO: 40.352559434)
   Max dTM/dt [1/us]:6.06e+03 at an offset of 212.038 usec -> 2021-11-08T13:12:15.352.557.763 (XO: 40.352559396)
   Min dTM/dt [1/us]:-7.49e+03 at an offset of 212.126 usec -> 2021-11-08T13:12:15.352.557.850 (XO: 40.352559484)

 - Block number: 4 starting at 2021-11-08T13:12:33.824.728.725 (XO: 58.824731929 - 58.825037930, t=35.6 degC)  Rank: 00064
   TM values: Max =   184  Min =  -195  Ave =    -4.37  Std =    22.49
   Min TM value:-195 at an offset of 131.550 usec -> 2021-11-08T13:12:33.824.860.275 (XO: 58.824863480)
   Max TM value:184 at an offset of 131.250 usec -> 2021-11-08T13:12:33.824.859.975 (XO: 58.824863180)
   Max dTM/dt [1/us]:1.26e+03 at an offset of 133.988 usec -> 2021-11-08T13:12:33.824.862.712 (XO: 58.824865917)
   Min dTM/dt [1/us]:-1.50e+03 at an offset of 131.375 usec -> 2021-11-08T13:12:33.824.860.100 (XO: 58.824863305)

 - Block number: 5 starting at 2021-11-08T13:12:41.975.108.725 (XO: 66.975112438 - 66.975418439, t=35.6 degC)  Rank: 00069
   TM values: Max =   161  Min =  -220  Ave =    -4.49  Std =    18.18
   Min TM value:-220 at an offset of 126.975 usec -> 2021-11-08T13:12:41.975.235.700 (XO: 66.975239413)
   Max TM value:161 at an offset of 126.625 usec -> 2021-11-08T13:12:41.975.235.350 (XO: 66.975239063)
   Max dTM/dt [1/us]:1.14e+03 at an offset of 129.375 usec -> 2021-11-08T13:12:41.975.238.100 (XO: 66.975241813)
   Min dTM/dt [1/us]:-1.47e+03 at an offset of 114.038 usec -> 2021-11-08T13:12:41.975.222.762 (XO: 66.975226476)
