-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar 23 15:56:16 2023 UTC
-- File 20211104_152237_1227.dat, 18  packets found.
APID:
      1227 (SL) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.157165 to 10.157454
-each separately from 10s to 10s and from 157165us to 157454us.
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU packet time from 2021-11-04T15:22:50.157.165.000 to 2021-11-04T15:22:50.157.454.000
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.9683524 to 11.9683524
   Differences in neigboring packets: 
      0.0000000 in 17 packets (all packets)
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 18 packets (all packets)
Number of generated alerts: 
      0 in 18 packets (all packets)
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20211104_152237_1227.dat, 1 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      69 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.1255182 to 12.1258072
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-11-04T15:22:50.157.165.000 to 2021-11-04T15:22:50.157.454.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.125518200 corrected by 40bit 10MHz counter of 11.968352400 obtained at 1Hz counter of 10. Result: 10.157166334
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-11-04T15:22:50.157.024.725
   Max 1st sample MIU time among all SL packets: 2021-11-04T15:22:50.157.313.725
   Min 1st sample XO time among all SL packets: 10.157026059
   Max 1st sample XO time among all SL packets: 10.157315060
   TM values from -216 to 202 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-11-04T15:22:50.157.024.725 (XO: 10.157026059 - 10.157332060, t=35.6 degC)  Rank: 00069
   TM values: Max =   202  Min =  -216  Ave =    -4.32  Std =    26.00
   Min TM value:-216 at an offset of 123.700 usec -> 2021-11-04T15:22:50.157.148.425 (XO: 10.157149759)
   Max TM value:202 at an offset of 94.688 usec -> 2021-11-04T15:22:50.157.119.412 (XO: 10.157120747)
   Max dTM/dt [1/us]:1.69e+03 at an offset of 121.275 usec -> 2021-11-04T15:22:50.157.146.000 (XO: 10.157147334)
   Min dTM/dt [1/us]:-1.83e+03 at an offset of 94.775 usec -> 2021-11-04T15:22:50.157.119.500 (XO: 10.157120834)
