-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Mar 23 14:45:01 2023 UTC
-- File 20211101_190848_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.265494 to 50.036569
-each separately from 7s to 50s and from 36280us to 976139us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.710067 in 1 packets: 18
      7.279849 in 1 packets: 36
      19.780003 in 1 packets: 54
MIU packet time from 2021-11-01T19:08:58.265.494.000 to 2021-11-01T19:09:41.036.569.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.710067 in 1 packets: 18
      7.279849 in 1 packets: 36
      19.780003 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.5297218 to 51.5295749
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999488 in 1 packets: 18
      7.9999726 in 1 packets: 36
      19.9999317 in 1 packets: 54
Temperature FPGA: 
      05 -> 48.5 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      06 -> 47.2 degC in 18 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Temperature XO: 
      C0 -> 35.6 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 72 packets (all packets)
Number of generated alerts: 
      0 in 72 packets (all packets)
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20211101_190848_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      68 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.7952165 to 8.7955055
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.7100150  -- OK for SL --
   MIU time from 2021-11-01T19:08:58.265.494.000 to 2021-11-01T19:08:58.265.783.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.710067

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      72 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.5055205 to 24.5058095
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.2798230  -- OK for SL --
   MIU time from 2021-11-01T19:09:13.975.850.000 to 2021-11-01T19:09:13.976.139.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.279849

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      69 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.7856325 to 31.7859215
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.7799350  -- OK for SL --
   MIU time from 2021-11-01T19:09:21.255.988.000 to 2021-11-01T19:09:21.256.277.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.780003

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      64 in 18 packets (all packets)  -- OK --
   10 MHz counter from 51.5658565 to 51.5661455
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-11-01T19:09:41.036.280.000 to 2021-11-01T19:09:41.036.569.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 8.795216500 corrected by 40bit 10MHz counter of 8.529721800 obtained at 1Hz counter of 7. Result: 7.265495603
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-11-01T19:08:58.265.353.725
   Max 1st sample MIU time among all SL packets: 2021-11-01T19:09:41.036.428.725
   Min 1st sample XO time among all SL packets: 7.265355327
   Max 1st sample XO time among all SL packets: 50.036430449
   TM values from -219 to 224 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-11-01T19:08:58.265.353.725 (XO: 7.265355327 - 7.265661328, t=35.6 degC)  Rank: 00068
   TM values: Max =   191  Min =  -219  Ave =    -4.66  Std =    24.90
   Min TM value:-219 at an offset of 110.913 usec -> 2021-11-01T19:08:58.265.464.637 (XO: 7.265466240)
   Max TM value:191 at an offset of 97.750 usec -> 2021-11-01T19:08:58.265.451.475 (XO: 7.265453078)
   Max dTM/dt [1/us]:1.53e+03 at an offset of 68.675 usec -> 2021-11-01T19:08:58.265.422.400 (XO: 7.265424002)
   Min dTM/dt [1/us]:-1.27e+03 at an offset of 68.888 usec -> 2021-11-01T19:08:58.265.422.612 (XO: 7.265424215)

 - Block number: 2 starting at 2021-11-01T19:09:13.975.709.725 (XO: 22.975712942 - 22.976018943, t=35.6 degC)  Rank: 00072
   TM values: Max =   182  Min =  -174  Ave =    -4.61  Std =    24.25
   Min TM value:-174 at an offset of 126.013 usec -> 2021-11-01T19:09:13.975.835.737 (XO: 22.975838955)
   Max TM value:182 at an offset of 113.000 usec -> 2021-11-01T19:09:13.975.822.725 (XO: 22.975825943)
   Max dTM/dt [1/us]:1.61e+03 at an offset of 128.100 usec -> 2021-11-01T19:09:13.975.837.825 (XO: 22.975841043)
   Min dTM/dt [1/us]:-1.60e+03 at an offset of 125.600 usec -> 2021-11-01T19:09:13.975.835.325 (XO: 22.975838543)

 - Block number: 3 starting at 2021-11-01T19:09:21.255.847.725 (XO: 30.255849895 - 30.256155896, t=35.6 degC)  Rank: 00069
   TM values: Max =   181  Min =  -199  Ave =    -4.67  Std =    20.89
   Min TM value:-199 at an offset of 125.925 usec -> 2021-11-01T19:09:21.255.973.650 (XO: 30.255975820)
   Max TM value:181 at an offset of 126.538 usec -> 2021-11-01T19:09:21.255.974.262 (XO: 30.255976433)
   Max dTM/dt [1/us]:1.31e+03 at an offset of 126.000 usec -> 2021-11-01T19:09:21.255.973.725 (XO: 30.255975895)
   Min dTM/dt [1/us]:-1.58e+03 at an offset of 125.775 usec -> 2021-11-01T19:09:21.255.973.500 (XO: 30.255975670)

 - Block number: 4 starting at 2021-11-01T19:09:41.036.139.725 (XO: 50.036141448 - 50.036447449, t=35.6 degC)  Rank: 00064
   TM values: Max =   224  Min =  -200  Ave =    -4.62  Std =    22.95
   Min TM value:-200 at an offset of 125.075 usec -> 2021-11-01T19:09:41.036.264.800 (XO: 50.036266523)
   Max TM value:224 at an offset of 124.488 usec -> 2021-11-01T19:09:41.036.264.212 (XO: 50.036265936)
   Max dTM/dt [1/us]:1.66e+03 at an offset of 124.438 usec -> 2021-11-01T19:09:41.036.264.162 (XO: 50.036265886)
   Min dTM/dt [1/us]:-1.51e+03 at an offset of 126.988 usec -> 2021-11-01T19:09:41.036.266.712 (XO: 50.036268436)
