-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Feb 12 22:46:05 2022 UTC
-- File 20211030_085300_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.893266 to 27.335226
-each separately from 1s to 27s and from 334937us to 893555us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.441382 in 1 packets: 18
MIU packet time from 2021-10-30T08:53:04.893.266.000 to 2021-10-30T08:53:30.335.226.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.441382 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.8217772 to 28.8216893
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.9999121 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 36 packets (all packets)
Temperature XO: 
      C0 -> 35.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20211030_085300_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.7150430 to 3.7153320
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.4412950  -- OK for SL --
   MIU time from 2021-10-30T08:53:04.893.266.000 to 2021-10-30T08:53:04.893.555.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.441382

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.1566270 to 29.1569160
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-10-30T08:53:30.334.937.000 to 2021-10-30T08:53:30.335.226.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.715043000 corrected by 40bit 10MHz counter of 2.821777200 obtained at 1Hz counter of 1. Result: 1.893268837
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-10-30T08:53:04.893.125.725
   Max 1st sample MIU time among all SL packets: 2021-10-30T08:53:30.335.085.725
   Min 1st sample XO time among all SL packets: 1.893128562
   Max 1st sample XO time among all SL packets: 27.335087531
   TM values from -270 to 235 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-10-30T08:53:04.893.125.725 (XO: 1.893128562 - 1.893434563, t=35.6 degC)  Rank: 00078
   TM values: Max =   220  Min =  -270  Ave =    -4.57  Std =    23.07
   Min TM value:-270 at an offset of 91.425 usec -> 2021-10-30T08:53:04.893.217.150 (XO: 1.893219987)
   Max TM value:220 at an offset of 123.188 usec -> 2021-10-30T08:53:04.893.248.912 (XO: 1.893251750)
   Max dTM/dt [1/us]:1.96e+03 at an offset of 123.025 usec -> 2021-10-30T08:53:04.893.248.750 (XO: 1.893251587)
   Min dTM/dt [1/us]:-2.00e+03 at an offset of 91.263 usec -> 2021-10-30T08:53:04.893.216.987 (XO: 1.893219824)

 - Block number: 2 starting at 2021-10-30T08:53:30.334.796.725 (XO: 27.334798530 - 27.335104531, t=35.6 degC)  Rank: 00088
   TM values: Max =   235  Min =  -234  Ave =    -4.45  Std =    19.11
   Min TM value:-234 at an offset of 108.388 usec -> 2021-10-30T08:53:30.334.905.112 (XO: 27.334906918)
   Max TM value:235 at an offset of 111.063 usec -> 2021-10-30T08:53:30.334.907.787 (XO: 27.334909593)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 110.963 usec -> 2021-10-30T08:53:30.334.907.687 (XO: 27.334909493)
   Min dTM/dt [1/us]:-2.16e+03 at an offset of 108.213 usec -> 2021-10-30T08:53:30.334.904.937 (XO: 27.334906743)
