-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Feb 12 21:14:19 2022 UTC
-- File 20211029_201603_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.538854 to 40.498790
-each separately from 3s to 40s and from 498501us to 839063us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.299631 in 1 packets: 18
      22.659438 in 1 packets: 36
MIU packet time from 2021-10-29T20:16:09.538.854.000 to 2021-10-29T20:16:46.498.790.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.299631 in 1 packets: 18
      22.659438 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.4669475 to 41.4668205
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999520 in 1 packets: 18
      22.9999210 in 1 packets: 36
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C0 -> 35.6 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      BF -> 34.0 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20211029_201603_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.0058018 to 5.0060908
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.2995830  -- OK for SL --
   MIU time from 2021-10-29T20:16:09.538.854.000 to 2021-10-29T20:16:09.539.143.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.299631

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.3056738 to 19.3059628
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.6593590  -- OK for SL --
   MIU time from 2021-10-29T20:16:23.838.774.000 to 2021-10-29T20:16:23.839.063.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.659438

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.9653218 to 41.9656108
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-10-29T20:16:46.498.501.000 to 2021-10-29T20:16:46.498.790.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.005801800 corrected by 40bit 10MHz counter of 4.466947500 obtained at 1Hz counter of 3. Result: 3.538856186
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-10-29T20:16:09.538.713.725
   Max 1st sample MIU time among all SL packets: 2021-10-29T20:16:46.498.649.725
   Min 1st sample XO time among all SL packets: 3.538715911
   Max 1st sample XO time among all SL packets: 40.498651720
   TM values from -265 to 222 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-10-29T20:16:09.538.713.725 (XO: 3.538715911 - 3.539021912, t=35.6 degC)  Rank: 00080
   TM values: Max =   220  Min =  -265  Ave =    -4.47  Std =    25.55
   Min TM value:-265 at an offset of 117.575 usec -> 2021-10-29T20:16:09.538.831.300 (XO: 3.538833486)
   Max TM value:220 at an offset of 120.188 usec -> 2021-10-29T20:16:09.538.833.912 (XO: 3.538836098)
   Max dTM/dt [1/us]:2.01e+03 at an offset of 119.975 usec -> 2021-10-29T20:16:09.538.833.700 (XO: 3.538835886)
   Min dTM/dt [1/us]:-2.45e+03 at an offset of 117.413 usec -> 2021-10-29T20:16:09.538.831.137 (XO: 3.538833323)

 - Block number: 2 starting at 2021-10-29T20:16:23.838.633.725 (XO: 17.838636876 - 17.838942877, t=34.0 degC)  Rank: 00085
   TM values: Max =   222  Min =  -254  Ave =    -4.39  Std =    23.21
   Min TM value:-254 at an offset of 105.500 usec -> 2021-10-29T20:16:23.838.739.225 (XO: 17.838742377)
   Max TM value:222 at an offset of 108.050 usec -> 2021-10-29T20:16:23.838.741.775 (XO: 17.838744927)
   Max dTM/dt [1/us]:2.74e+03 at an offset of 107.913 usec -> 2021-10-29T20:16:23.838.741.637 (XO: 17.838744789)
   Min dTM/dt [1/us]:-2.28e+03 at an offset of 105.363 usec -> 2021-10-29T20:16:23.838.739.087 (XO: 17.838742239)

 - Block number: 3 starting at 2021-10-29T20:16:46.498.360.725 (XO: 40.498362719 - 40.498668720, t=35.6 degC)  Rank: 00082
   TM values: Max =   214  Min =  -255  Ave =    -4.44  Std =    25.97
   Min TM value:-255 at an offset of 118.713 usec -> 2021-10-29T20:16:46.498.479.437 (XO: 40.498481432)
   Max TM value:214 at an offset of 121.263 usec -> 2021-10-29T20:16:46.498.481.987 (XO: 40.498483982)
   Max dTM/dt [1/us]:2.09e+03 at an offset of 121.063 usec -> 2021-10-29T20:16:46.498.481.787 (XO: 40.498483782)
   Min dTM/dt [1/us]:-2.51e+03 at an offset of 118.513 usec -> 2021-10-29T20:16:46.498.479.237 (XO: 40.498481232)
