-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Feb  1 02:43:16 2022 UTC
-- File 20211007_073416_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.517732 to 55.238056
-each separately from 11s to 55s and from 237767us to 547821us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.029511 in 1 packets: 18
      8.769677 in 1 packets: 36
      22.919980 in 1 packets: 54
MIU packet time from 2021-10-07T07:34:30.517.732.000 to 2021-10-07T07:35:14.238.056.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.029511 in 1 packets: 18
      8.769677 in 1 packets: 36
      22.919980 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.8981710 to 56.8980238
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999598 in 1 packets: 18
      8.9999699 in 1 packets: 36
      22.9999231 in 1 packets: 54
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 71 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 71 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20211007_073416_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.4159034 to 13.4161924
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.0294710  -- OK for SL --
   MIU time from 2021-10-07T07:34:30.517.732.000 to 2021-10-07T07:34:30.518.021.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.029511

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.4456634 to 25.4459524
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.7696470  -- OK for SL --
   MIU time from 2021-10-07T07:34:42.547.532.000 to 2021-10-07T07:34:42.547.821.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.769677

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.2155994 to 34.2158884
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.9199030  -- OK for SL --
   MIU time from 2021-10-07T07:34:51.317.498.000 to 2021-10-07T07:34:51.317.787.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.919980

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 57.1357914 to 57.1360804
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-10-07T07:35:14.237.767.000 to 2021-10-07T07:35:14.238.056.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 13.415903400 corrected by 40bit 10MHz counter of 12.898171000 obtained at 1Hz counter of 11. Result: 11.517734160
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-10-07T07:34:30.517.591.725
   Max 1st sample MIU time among all SL packets: 2021-10-07T07:35:14.237.915.725
   Min 1st sample XO time among all SL packets: 11.517593885
   Max 1st sample XO time among all SL packets: 55.237917110
   TM values from -275 to 254 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-10-07T07:34:30.517.591.725 (XO: 11.517593885 - 11.517899886, t=37.2 degC)  Rank: 00085
   TM values: Max =   234  Min =  -269  Ave =    -4.47  Std =    23.14
   Min TM value:-269 at an offset of 122.425 usec -> 2021-10-07T07:34:30.517.714.150 (XO: 11.517716310)
   Max TM value:234 at an offset of 120.150 usec -> 2021-10-07T07:34:30.517.711.875 (XO: 11.517714035)
   Max dTM/dt [1/us]:2.04e+03 at an offset of 120.013 usec -> 2021-10-07T07:34:30.517.711.737 (XO: 11.517713898)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 122.200 usec -> 2021-10-07T07:34:30.517.713.925 (XO: 11.517716085)

 - Block number: 2 starting at 2021-10-07T07:34:42.547.391.725 (XO: 23.547394131 - 23.547700132, t=37.2 degC)  Rank: 00080
   TM values: Max =   253  Min =  -275  Ave =    -4.46  Std =    26.85
   Min TM value:-275 at an offset of 109.788 usec -> 2021-10-07T07:34:42.547.501.512 (XO: 23.547503919)
   Max TM value:253 at an offset of 109.538 usec -> 2021-10-07T07:34:42.547.501.262 (XO: 23.547503669)
   Max dTM/dt [1/us]:2.29e+03 at an offset of 112.238 usec -> 2021-10-07T07:34:42.547.503.962 (XO: 23.547506369)
   Min dTM/dt [1/us]:-2.81e+03 at an offset of 109.663 usec -> 2021-10-07T07:34:42.547.501.387 (XO: 23.547503794)

 - Block number: 3 starting at 2021-10-07T07:34:51.317.357.725 (XO: 32.317359472 - 32.317665473, t=37.2 degC)  Rank: 00078
   TM values: Max =   221  Min =  -241  Ave =    -4.55  Std =    19.40
   Min TM value:-241 at an offset of 122.175 usec -> 2021-10-07T07:34:51.317.479.900 (XO: 32.317481648)
   Max TM value:221 at an offset of 124.675 usec -> 2021-10-07T07:34:51.317.482.400 (XO: 32.317484148)
   Max dTM/dt [1/us]:2.02e+03 at an offset of 124.538 usec -> 2021-10-07T07:34:51.317.482.262 (XO: 32.317484010)
   Min dTM/dt [1/us]:-2.04e+03 at an offset of 121.975 usec -> 2021-10-07T07:34:51.317.479.700 (XO: 32.317481448)

 - Block number: 4 starting at 2021-10-07T07:35:14.237.626.725 (XO: 55.237628109 - 55.237934110, t=37.2 degC)  Rank: 00079
   TM values: Max =   254  Min =  -269  Ave =    -4.52  Std =    24.28
   Min TM value:-269 at an offset of 47.275 usec -> 2021-10-07T07:35:14.237.674.000 (XO: 55.237675384)
   Max TM value:254 at an offset of 34.350 usec -> 2021-10-07T07:35:14.237.661.075 (XO: 55.237662459)
   Max dTM/dt [1/us]:2.21e+03 at an offset of 32.300 usec -> 2021-10-07T07:35:14.237.659.025 (XO: 55.237660409)
   Min dTM/dt [1/us]:-2.23e+03 at an offset of 34.463 usec -> 2021-10-07T07:35:14.237.661.187 (XO: 55.237662572)
