-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jan 31 23:21:42 2022 UTC
-- File 20211007_041409_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.666147 to 60.778760
-each separately from 11s to 60s and from 206764us to 778760us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.540328 in 1 packets: 18
      4.419709 in 1 packets: 36
      19.610993 in 1 packets: 54
      8.540138 in 1 packets: 72
MIU packet time from 2021-10-07T04:14:23.666.147.000 to 2021-10-07T04:15:12.778.760.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.540328 in 1 packets: 18
      4.419709 in 1 packets: 36
      19.610993 in 1 packets: 54
      8.540138 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.8347840 to 61.8346202
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999432 in 1 packets: 18
      3.9999866 in 1 packets: 36
      19.9999331 in 1 packets: 54
      7.9999733 in 1 packets: 72
Temperature FPGA: 
      04 -> 49.8 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      03 -> 51.1 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 53 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 18 packets: 73, 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 53 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 18 packets: 73, 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20211007_041409_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.5009322 to 13.5012212
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.5402710  -- OK for SL --
   MIU time from 2021-10-07T04:14:23.666.147.000 to 2021-10-07T04:14:23.666.436.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.540328

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.0414922 to 30.0417812
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.4196950  -- OK for SL --
   MIU time from 2021-10-07T04:14:40.206.764.000 to 2021-10-07T04:14:40.207.053.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.419709

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.4614762 to 34.4617652
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.6109270  -- OK for SL --
   MIU time from 2021-10-07T04:14:44.626.762.000 to 2021-10-07T04:14:44.627.051.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.610993

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 54.0726922 to 54.0729812
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.5401110  -- OK for SL --
   MIU time from 2021-10-07T04:15:04.238.044.000 to 2021-10-07T04:15:04.238.333.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.540138

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 62.6130922 to 62.6133812
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-10-07T04:15:12.778.471.000 to 2021-10-07T04:15:12.778.760.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 13.500932200 corrected by 40bit 10MHz counter of 12.834784000 obtained at 1Hz counter of 11. Result: 11.666150398
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-10-07T04:14:23.666.006.725
   Max 1st sample MIU time among all SL packets: 2021-10-07T04:15:12.778.619.725
   Min 1st sample XO time among all SL packets: 11.666010123
   Max 1st sample XO time among all SL packets: 60.778623372
   TM values from -270 to 272 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-10-07T04:14:23.666.006.725 (XO: 11.666010123 - 11.666316124, t=37.2 degC)  Rank: 00080
   TM values: Max =   206  Min =  -264  Ave =    -4.49  Std =    20.90
   Min TM value:-264 at an offset of 113.000 usec -> 2021-10-07T04:14:23.666.119.725 (XO: 11.666123123)
   Max TM value:206 at an offset of 112.675 usec -> 2021-10-07T04:14:23.666.119.400 (XO: 11.666122798)
   Max dTM/dt [1/us]:2.47e+03 at an offset of 115.338 usec -> 2021-10-07T04:14:23.666.122.062 (XO: 11.666125461)
   Min dTM/dt [1/us]:-2.06e+03 at an offset of 112.800 usec -> 2021-10-07T04:14:23.666.119.525 (XO: 11.666122923)

 - Block number: 2 starting at 2021-10-07T04:14:40.206.623.725 (XO: 28.206625428 - 28.206931429, t=37.2 degC)  Rank: 00082
   TM values: Max =   226  Min =  -267  Ave =    -4.42  Std =    23.40
   Min TM value:-267 at an offset of 126.513 usec -> 2021-10-07T04:14:40.206.750.237 (XO: 28.206751940)
   Max TM value:226 at an offset of 123.963 usec -> 2021-10-07T04:14:40.206.747.687 (XO: 28.206749390)
   Max dTM/dt [1/us]:1.94e+03 at an offset of 126.638 usec -> 2021-10-07T04:14:40.206.750.362 (XO: 28.206752065)
   Min dTM/dt [1/us]:-2.43e+03 at an offset of 124.050 usec -> 2021-10-07T04:14:40.206.747.775 (XO: 28.206749478)

 - Block number: 3 starting at 2021-10-07T04:14:44.626.621.725 (XO: 32.626624193 - 32.626930194, t=37.2 degC)  Rank: 00080
   TM values: Max =   218  Min =  -235  Ave =    -4.43  Std =    24.72
   Min TM value:-235 at an offset of 124.088 usec -> 2021-10-07T04:14:44.626.745.812 (XO: 32.626748281)
   Max TM value:218 at an offset of 126.713 usec -> 2021-10-07T04:14:44.626.748.437 (XO: 32.626750906)
   Max dTM/dt [1/us]:2.24e+03 at an offset of 126.588 usec -> 2021-10-07T04:14:44.626.748.312 (XO: 32.626750781)
   Min dTM/dt [1/us]:-2.28e+03 at an offset of 123.825 usec -> 2021-10-07T04:14:44.626.745.550 (XO: 32.626748018)

 - Block number: 4 starting at 2021-10-07T04:15:04.237.903.725 (XO: 52.237905834 - 52.238211835, t=37.2 degC)  Rank: 00076
   TM values: Max =   228  Min =  -247  Ave =    -4.50  Std =    25.88
   Min TM value:-247 at an offset of 182.813 usec -> 2021-10-07T04:15:04.238.086.538 (XO: 52.238088647)
   Max TM value:228 at an offset of 185.451 usec -> 2021-10-07T04:15:04.238.089.175 (XO: 52.238091285)
   Max dTM/dt [1/us]:1.80e+03 at an offset of 185.301 usec -> 2021-10-07T04:15:04.238.089.025 (XO: 52.238091135)
   Min dTM/dt [1/us]:-1.96e+03 at an offset of 182.638 usec -> 2021-10-07T04:15:04.238.086.363 (XO: 52.238088472)

 - Block number: 5 starting at 2021-10-07T04:15:12.778.330.725 (XO: 60.778334371 - 60.778640372, t=37.2 degC)  Rank: 00083
   TM values: Max =   272  Min =  -270  Ave =    -4.45  Std =    20.98
   Min TM value:-270 at an offset of 39.513 usec -> 2021-10-07T04:15:12.778.370.237 (XO: 60.778373884)
   Max TM value:272 at an offset of 37.400 usec -> 2021-10-07T04:15:12.778.368.125 (XO: 60.778371771)
   Max dTM/dt [1/us]:2.49e+03 at an offset of 37.250 usec -> 2021-10-07T04:15:12.778.367.975 (XO: 60.778371621)
   Min dTM/dt [1/us]:-2.76e+03 at an offset of 39.350 usec -> 2021-10-07T04:15:12.778.370.075 (XO: 60.778373721)
