-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jan 31 22:58:11 2022 UTC
-- File 20211007_033908_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.303613 to 18.743140
-each separately from 2s to 18s and from 303613us to 743140us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.438949 in 1 packets: 18
MIU packet time from 2021-10-07T03:39:13.303.613.000 to 2021-10-07T03:39:29.743.140.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.438949 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.8915906 to 19.8915370
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999464 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20211007_033908_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.1952043 to 4.1954933
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.4388950  -- OK for SL --
   MIU time from 2021-10-07T03:39:13.303.613.000 to 2021-10-07T03:39:13.303.902.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.438949

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.6343883 to 20.6346773
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-10-07T03:39:29.742.851.000 to 2021-10-07T03:39:29.743.140.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.195204300 corrected by 40bit 10MHz counter of 3.891590600 obtained at 1Hz counter of 2. Result: 2.303614702
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-10-07T03:39:13.303.472.725
   Max 1st sample MIU time among all SL packets: 2021-10-07T03:39:29.742.999.725
   Min 1st sample XO time among all SL packets: 2.303474426
   Max 1st sample XO time among all SL packets: 18.743002477
   TM values from -251 to 224 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-10-07T03:39:13.303.472.725 (XO: 2.303474426 - 2.303780427, t=37.2 degC)  Rank: 00076
   TM values: Max =   224  Min =  -251  Ave =    -4.40  Std =    26.93
   Min TM value:-251 at an offset of 130.650 usec -> 2021-10-07T03:39:13.303.603.375 (XO: 2.303605077)
   Max TM value:224 at an offset of 133.275 usec -> 2021-10-07T03:39:13.303.606.000 (XO: 2.303607702)
   Max dTM/dt [1/us]:1.69e+03 at an offset of 133.163 usec -> 2021-10-07T03:39:13.303.605.887 (XO: 2.303607589)
   Min dTM/dt [1/us]:-2.11e+03 at an offset of 188.838 usec -> 2021-10-07T03:39:13.303.661.563 (XO: 2.303663265)

 - Block number: 2 starting at 2021-10-07T03:39:29.742.710.725 (XO: 18.742713476 - 18.743019477, t=37.2 degC)  Rank: 00082
   TM values: Max =   224  Min =  -251  Ave =    -4.48  Std =    19.72
   Min TM value:-251 at an offset of 129.588 usec -> 2021-10-07T03:39:29.742.840.312 (XO: 18.742843064)
   Max TM value:224 at an offset of 129.300 usec -> 2021-10-07T03:39:29.742.840.025 (XO: 18.742842776)
   Max dTM/dt [1/us]:2.33e+03 at an offset of 131.938 usec -> 2021-10-07T03:39:29.742.842.662 (XO: 18.742845414)
   Min dTM/dt [1/us]:-2.35e+03 at an offset of 129.388 usec -> 2021-10-07T03:39:29.742.840.112 (XO: 18.742842864)
