-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jan 30 09:03:29 2022 UTC
-- File 20211005_150729_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 13.223748 to 84.381586
-each separately from 13s to 84s and from 43111us to 913524us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.689198 in 1 packets: 18
      5.129587 in 1 packets: 36
      23.578780 in 1 packets: 54
      4.019532 in 1 packets: 72
      23.739007 in 1 packets: 90
MIU packet time from 2021-10-05T15:07:45.223.748.000 to 2021-10-05T15:08:56.381.586.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.689198 in 1 packets: 18
      5.129587 in 1 packets: 36
      23.578780 in 1 packets: 54
      4.019532 in 1 packets: 72
      23.739007 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.9049490 to 85.9047120
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999533 in 1 packets: 18
      5.9999800 in 1 packets: 36
      22.9999232 in 1 packets: 54
      3.9999867 in 1 packets: 72
      23.9999198 in 1 packets: 90
Temperature FPGA: 
      05 -> 48.5 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 54 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, ...
Temperature XO: 
      C1 -> 37.2 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55
      2 in 53 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55
      2 in 53 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20211005_150729_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.1286980 to 15.1289870
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.6891510  -- OK for SL --
   MIU time from 2021-10-05T15:07:45.223.748.000 to 2021-10-05T15:07:45.224.037.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.689198

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.8181380 to 29.8184270
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.1295670  -- OK for SL --
   MIU time from 2021-10-05T15:07:59.913.235.000 to 2021-10-05T15:07:59.913.524.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.129587

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.9479940 to 34.9482830
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.5787030  -- OK for SL --
   MIU time from 2021-10-05T15:08:05.043.111.000 to 2021-10-05T15:08:05.043.400.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.578780

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 58.5269860 to 58.5272750
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.0195190  -- OK for SL --
   MIU time from 2021-10-05T15:08:28.622.180.000 to 2021-10-05T15:08:28.622.469.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.019532

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 62.5467940 to 62.5470830
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.7389270  -- OK for SL --
   MIU time from 2021-10-05T15:08:32.642.001.000 to 2021-10-05T15:08:32.642.290.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.739007

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 86.2860100 to 86.2862990
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-10-05T15:08:56.381.297.000 to 2021-10-05T15:08:56.381.586.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 13, 32bit 10MHz counter = 15.128698000 corrected by 40bit 10MHz counter of 14.904949000 obtained at 1Hz counter of 13. Result: 13.223749738
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-10-05T15:07:45.223.607.725
   Max 1st sample MIU time among all SL packets: 2021-10-05T15:08:56.381.445.725
   Min 1st sample XO time among all SL packets: 13.223609463
   Max 1st sample XO time among all SL packets: 84.381448022
   TM values from -277 to 252 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-10-05T15:07:45.223.607.725 (XO: 13.223609463 - 13.223915464, t=37.2 degC)  Rank: 00075
   TM values: Max =   204  Min =  -260  Ave =    -4.53  Std =    19.64
   Min TM value:-260 at an offset of 121.238 usec -> 2021-10-05T15:07:45.223.728.962 (XO: 13.223730701)
   Max TM value:204 at an offset of 120.938 usec -> 2021-10-05T15:07:45.223.728.662 (XO: 13.223730401)
   Max dTM/dt [1/us]:1.81e+03 at an offset of 123.588 usec -> 2021-10-05T15:07:45.223.731.312 (XO: 13.223733051)
   Min dTM/dt [1/us]:-1.82e+03 at an offset of 121.025 usec -> 2021-10-05T15:07:45.223.728.750 (XO: 13.223730488)

 - Block number: 2 starting at 2021-10-05T15:07:59.913.094.725 (XO: 27.913098530 - 27.913404531, t=37.2 degC)  Rank: 00080
   TM values: Max =   234  Min =  -249  Ave =    -4.57  Std =    23.72
   Min TM value:-249 at an offset of 123.675 usec -> 2021-10-05T15:07:59.913.218.400 (XO: 27.913222205)
   Max TM value:234 at an offset of 110.663 usec -> 2021-10-05T15:07:59.913.205.387 (XO: 27.913209192)
   Max dTM/dt [1/us]:2.15e+03 at an offset of 108.700 usec -> 2021-10-05T15:07:59.913.203.425 (XO: 27.913207230)
   Min dTM/dt [1/us]:-2.29e+03 at an offset of 110.800 usec -> 2021-10-05T15:07:59.913.205.525 (XO: 27.913209330)

 - Block number: 3 starting at 2021-10-05T15:08:05.042.970.725 (XO: 33.042971571 - 33.043277572, t=37.2 degC)  Rank: 00087
   TM values: Max =   252  Min =  -277  Ave =    -4.51  Std =    24.74
   Min TM value:-277 at an offset of 112.713 usec -> 2021-10-05T15:08:05.043.083.437 (XO: 33.043084284)
   Max TM value:252 at an offset of 112.463 usec -> 2021-10-05T15:08:05.043.083.187 (XO: 33.043084034)
   Max dTM/dt [1/us]:2.07e+03 at an offset of 110.525 usec -> 2021-10-05T15:08:05.043.081.250 (XO: 33.043082096)
   Min dTM/dt [1/us]:-2.78e+03 at an offset of 112.588 usec -> 2021-10-05T15:08:05.043.083.312 (XO: 33.043084159)

 - Block number: 4 starting at 2021-10-05T15:08:28.622.039.725 (XO: 56.622042278 - 56.622348279, t=37.2 degC)  Rank: 00087
   TM values: Max =   232  Min =  -258  Ave =    -4.56  Std =    23.05
   Min TM value:-258 at an offset of 123.463 usec -> 2021-10-05T15:08:28.622.163.187 (XO: 56.622165741)
   Max TM value:232 at an offset of 123.138 usec -> 2021-10-05T15:08:28.622.162.862 (XO: 56.622165416)
   Max dTM/dt [1/us]:2.46e+03 at an offset of 125.775 usec -> 2021-10-05T15:08:28.622.165.500 (XO: 56.622168053)
   Min dTM/dt [1/us]:-2.39e+03 at an offset of 123.250 usec -> 2021-10-05T15:08:28.622.162.975 (XO: 56.622165528)

 - Block number: 5 starting at 2021-10-05T15:08:32.641.860.725 (XO: 60.641863643 - 60.642169644, t=37.2 degC)  Rank: 00077
   TM values: Max =   213  Min =  -246  Ave =    -4.53  Std =    22.50
   Min TM value:-246 at an offset of 150.238 usec -> 2021-10-05T15:08:32.642.010.962 (XO: 60.642013881)
   Max TM value:213 at an offset of 122.888 usec -> 2021-10-05T15:08:32.641.983.612 (XO: 60.641986531)
   Max dTM/dt [1/us]:1.93e+03 at an offset of 125.563 usec -> 2021-10-05T15:08:32.641.986.287 (XO: 60.641989206)
   Min dTM/dt [1/us]:-2.06e+03 at an offset of 122.975 usec -> 2021-10-05T15:08:32.641.983.700 (XO: 60.641986619)

 - Block number: 6 starting at 2021-10-05T15:08:56.381.156.725 (XO: 84.381159021 - 84.381465022, t=37.2 degC)  Rank: 00079
   TM values: Max =   227  Min =  -244  Ave =    -4.45  Std =    19.22
   Min TM value:-244 at an offset of 124.075 usec -> 2021-10-05T15:08:56.381.280.800 (XO: 84.381283096)
   Max TM value:227 at an offset of 123.813 usec -> 2021-10-05T15:08:56.381.280.537 (XO: 84.381282834)
   Max dTM/dt [1/us]:2.04e+03 at an offset of 126.438 usec -> 2021-10-05T15:08:56.381.283.162 (XO: 84.381285459)
   Min dTM/dt [1/us]:-2.37e+03 at an offset of 123.913 usec -> 2021-10-05T15:08:56.381.280.637 (XO: 84.381282934)
