-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan 29 22:55:34 2022 UTC
-- File 20211005_082721_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.541394 to 22.856418
-each separately from 3s to 22s and from 541394us to 856418us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.314446 in 1 packets: 18
MIU packet time from 2021-10-05T08:27:27.541.394.000 to 2021-10-05T08:27:46.856.418.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.314446 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.7896683 to 23.7896050
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.9999367 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20211005_082721_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.3310627 to 5.3313517
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.3143830  -- OK for SL --
   MIU time from 2021-10-05T08:27:27.541.394.000 to 2021-10-05T08:27:27.541.683.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.314446

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      90 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.6457347 to 24.6460237
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-10-05T08:27:46.856.129.000 to 2021-10-05T08:27:46.856.418.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.331062700 corrected by 40bit 10MHz counter of 4.789668300 obtained at 1Hz counter of 3. Result: 3.541396241
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-10-05T08:27:27.541.253.725
   Max 1st sample MIU time among all SL packets: 2021-10-05T08:27:46.856.277.725
   Min 1st sample XO time among all SL packets: 3.541255965
   Max 1st sample XO time among all SL packets: 22.856281336
   TM values from -234 to 218 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-10-05T08:27:27.541.253.725 (XO: 3.541255965 - 3.541561966, t=37.2 degC)  Rank: 00080
   TM values: Max =   218  Min =  -234  Ave =    -4.39  Std =    19.00
   Min TM value:-234 at an offset of 117.713 usec -> 2021-10-05T08:27:27.541.371.437 (XO: 3.541373678)
   Max TM value:218 at an offset of 117.375 usec -> 2021-10-05T08:27:27.541.371.100 (XO: 3.541373341)
   Max dTM/dt [1/us]:1.90e+03 at an offset of 115.350 usec -> 2021-10-05T08:27:27.541.369.075 (XO: 3.541371316)
   Min dTM/dt [1/us]:-2.02e+03 at an offset of 117.488 usec -> 2021-10-05T08:27:27.541.371.212 (XO: 3.541373453)

 - Block number: 2 starting at 2021-10-05T08:27:46.855.988.725 (XO: 22.855992335 - 22.856298336, t=37.2 degC)  Rank: 00090
   TM values: Max =   203  Min =  -218  Ave =    -4.26  Std =    17.79
   Min TM value:-218 at an offset of 128.038 usec -> 2021-10-05T08:27:46.856.116.762 (XO: 22.856120373)
   Max TM value:203 at an offset of 130.738 usec -> 2021-10-05T08:27:46.856.119.462 (XO: 22.856123073)
   Max dTM/dt [1/us]:1.85e+03 at an offset of 130.613 usec -> 2021-10-05T08:27:46.856.119.337 (XO: 22.856122948)
   Min dTM/dt [1/us]:-1.58e+03 at an offset of 127.863 usec -> 2021-10-05T08:27:46.856.116.587 (XO: 22.856120198)
