-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan 29 15:44:26 2022 UTC
-- File 20211005_072700_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.272914 to 22.640378
-each separately from 1s to 22s and from 272914us to 640378us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.366886 in 1 packets: 18
MIU packet time from 2021-10-05T07:27:04.272.914.000 to 2021-10-05T07:27:25.640.378.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.366886 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.7562939 to 23.7562238
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.9999299 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20211005_072700_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.0292092 to 3.0294982
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.3668150  -- OK for SL --
   MIU time from 2021-10-05T07:27:04.272.914.000 to 2021-10-05T07:27:04.273.203.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.366886

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.3963132 to 24.3966022
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-10-05T07:27:25.640.089.000 to 2021-10-05T07:27:25.640.378.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.029209200 corrected by 40bit 10MHz counter of 2.756293900 obtained at 1Hz counter of 1. Result: 1.272916228
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2021-10-05T07:27:04.272.773.725
   Max 1st sample MIU time among all SL packets: 2021-10-05T07:27:25.640.237.725
   Min 1st sample XO time among all SL packets: 1.272775952
   Max 1st sample XO time among all SL packets: 22.640240302
   TM values from -246 to 231 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-10-05T07:27:04.272.773.725 (XO: 1.272775952 - 1.273081953, t=37.2 degC)  Rank: 00083
   TM values: Max =   231  Min =  -231  Ave =    -4.41  Std =    18.63
   Min TM value:-231 at an offset of 121.388 usec -> 2021-10-05T07:27:04.272.895.112 (XO: 1.272897340)
   Max TM value:231 at an offset of 121.063 usec -> 2021-10-05T07:27:04.272.894.787 (XO: 1.272897015)
   Max dTM/dt [1/us]:1.90e+03 at an offset of 119.038 usec -> 2021-10-05T07:27:04.272.892.762 (XO: 1.272894990)
   Min dTM/dt [1/us]:-2.16e+03 at an offset of 121.163 usec -> 2021-10-05T07:27:04.272.894.887 (XO: 1.272897115)

 - Block number: 2 starting at 2021-10-05T07:27:25.639.948.725 (XO: 22.639951301 - 22.640257302, t=37.2 degC)  Rank: 00080
   TM values: Max =   222  Min =  -246  Ave =    -4.33  Std =    26.63
   Min TM value:-246 at an offset of 119.100 usec -> 2021-10-05T07:27:25.640.067.825 (XO: 22.640070401)
   Max TM value:222 at an offset of 118.763 usec -> 2021-10-05T07:27:25.640.067.487 (XO: 22.640070064)
   Max dTM/dt [1/us]:2.17e+03 at an offset of 121.450 usec -> 2021-10-05T07:27:25.640.070.175 (XO: 22.640072751)
   Min dTM/dt [1/us]:-2.17e+03 at an offset of 118.875 usec -> 2021-10-05T07:27:25.640.067.600 (XO: 22.640070176)
