-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan 29 06:28:12 2022 UTC
-- File 20211004_080225_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.447481 to 27.035845
-each separately from 9s to 27s and from 35556us to 447770us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.587786 in 1 packets: 18
MIU packet time from 2021-10-04T08:02:37.447.481.000 to 2021-10-04T08:02:55.035.845.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.587786 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.6935483 to 28.6934883
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.9999400 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20211004_080225_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      94 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.1410298 to 11.1413188
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.5877270  -- OK for SL --
   MIU time from 2021-10-04T08:02:37.447.481.000 to 2021-10-04T08:02:37.447.770.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.587786

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.7290458 to 28.7293348
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-10-04T08:02:55.035.556.000 to 2021-10-04T08:02:55.035.845.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.141029800 corrected by 40bit 10MHz counter of 10.693548300 obtained at 1Hz counter of 9. Result: 9.447482977
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-10-04T08:02:37.447.340.725
   Max 1st sample MIU time among all SL packets: 2021-10-04T08:02:55.035.704.725
   Min 1st sample XO time among all SL packets: 9.447342701
   Max 1st sample XO time among all SL packets: 27.035706343
   TM values from -474 to 224 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-10-04T08:02:37.447.340.725 (XO: 9.447342701 - 9.447648702, t=37.2 degC)  Rank: 00094
   TM values: Max =   224  Min =  -474  Ave =    -4.45  Std =    36.76
   Min TM value:-474 at an offset of 127.200 usec -> 2021-10-04T08:02:37.447.467.925 (XO: 9.447469902)
   Max TM value:224 at an offset of 91.663 usec -> 2021-10-04T08:02:37.447.432.387 (XO: 9.447434364)
   Max dTM/dt [1/us]:3.82e+03 at an offset of 127.250 usec -> 2021-10-04T08:02:37.447.467.975 (XO: 9.447469952)
   Min dTM/dt [1/us]:-4.05e+03 at an offset of 127.150 usec -> 2021-10-04T08:02:37.447.467.875 (XO: 9.447469852)

 - Block number: 2 starting at 2021-10-04T08:02:55.035.415.725 (XO: 27.035417342 - 27.035723343, t=37.2 degC)  Rank: 00081
   TM values: Max =   171  Min =  -226  Ave =    -4.38  Std =    17.35
   Min TM value:-226 at an offset of 118.513 usec -> 2021-10-04T08:02:55.035.534.237 (XO: 27.035535855)
   Max TM value:171 at an offset of 121.100 usec -> 2021-10-04T08:02:55.035.536.825 (XO: 27.035538442)
   Max dTM/dt [1/us]:1.40e+03 at an offset of 120.925 usec -> 2021-10-04T08:02:55.035.536.650 (XO: 27.035538267)
   Min dTM/dt [1/us]:-1.63e+03 at an offset of 118.313 usec -> 2021-10-04T08:02:55.035.534.037 (XO: 27.035535655)
