-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jan 24 23:46:08 2022 UTC
-- File 20210930_131655_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.184194 to 21.664483
-each separately from 2s to 21s and from 184194us to 664483us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.479711 in 1 packets: 18
MIU packet time from 2021-09-30T13:17:00.184.194.000 to 2021-09-30T13:17:19.664.483.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.479711 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.8825657 to 22.8825019
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.9999362 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210930_131655_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.0667604 to 4.0670494
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.4796470  -- OK for SL --
   MIU time from 2021-09-30T13:17:00.184.194.000 to 2021-09-30T13:17:00.184.483.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.479711

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.5466964 to 23.5469854
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-30T13:17:19.664.194.000 to 2021-09-30T13:17:19.664.483.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.066760400 corrected by 40bit 10MHz counter of 3.882565700 obtained at 1Hz counter of 2. Result: 2.184195308
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-30T13:17:00.184.053.725
   Max 1st sample MIU time among all SL packets: 2021-09-30T13:17:19.664.342.725
   Min 1st sample XO time among all SL packets: 2.184055032
   Max 1st sample XO time among all SL packets: 21.664345484
   TM values from -234 to 207 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-30T13:17:00.184.053.725 (XO: 2.184055032 - 2.184361033, t=37.2 degC)  Rank: 00074
   TM values: Max =   203  Min =  -230  Ave =    -4.57  Std =    18.17
   Min TM value:-230 at an offset of 123.825 usec -> 2021-09-30T13:17:00.184.177.550 (XO: 2.184178858)
   Max TM value:203 at an offset of 126.438 usec -> 2021-09-30T13:17:00.184.180.162 (XO: 2.184181470)
   Max dTM/dt [1/us]:1.49e+03 at an offset of 126.275 usec -> 2021-09-30T13:17:00.184.180.000 (XO: 2.184181308)
   Min dTM/dt [1/us]:-1.78e+03 at an offset of 123.663 usec -> 2021-09-30T13:17:00.184.177.387 (XO: 2.184178695)

 - Block number: 2 starting at 2021-09-30T13:17:19.664.053.725 (XO: 21.664056483 - 21.664362484, t=37.2 degC)  Rank: 00083
   TM values: Max =   207  Min =  -234  Ave =    -4.43  Std =    26.84
   Min TM value:-234 at an offset of 98.675 usec -> 2021-09-30T13:17:19.664.152.400 (XO: 21.664155158)
   Max TM value:207 at an offset of 144.613 usec -> 2021-09-30T13:17:19.664.198.337 (XO: 21.664201096)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 144.488 usec -> 2021-09-30T13:17:19.664.198.212 (XO: 21.664200971)
   Min dTM/dt [1/us]:-1.86e+03 at an offset of 98.513 usec -> 2021-09-30T13:17:19.664.152.237 (XO: 21.664154996)
