-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jan 24 17:12:23 2022 UTC
-- File 20210930_115524_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.835315 to 77.773268
-each separately from 10s to 77s and from 64822us to 904252us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.889630 in 1 packets: 18
      24.339299 in 1 packets: 36
      11.838852 in 1 packets: 54
      9.989217 in 1 packets: 72
      15.879221 in 1 packets: 90
MIU packet time from 2021-09-30T11:55:37.835.315.000 to 2021-09-30T11:56:44.773.268.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.889630 in 1 packets: 18
      24.339299 in 1 packets: 36
      11.838852 in 1 packets: 54
      9.989217 in 1 packets: 72
      15.879221 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.8689589 to 78.8687340
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.9999833 in 1 packets: 18
      24.9999162 in 1 packets: 36
      10.9999631 in 1 packets: 54
      9.9999665 in 1 packets: 72
      15.9999460 in 1 packets: 90
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 90 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 1 packets: 55
      2 in 53 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Number of generated alerts: 
      0 in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 1 packets: 55
      2 in 53 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20210930_115524_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.7042733 to 12.7045623
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.8896150  -- OK for SL --
   MIU time from 2021-09-30T11:55:37.835.315.000 to 2021-09-30T11:55:37.835.604.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.889630

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.5941773 to 17.5944663
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.3392150  -- OK for SL --
   MIU time from 2021-09-30T11:55:42.725.234.000 to 2021-09-30T11:55:42.725.523.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.339299

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.9336813 to 41.9339703
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.8388150  -- OK for SL --
   MIU time from 2021-09-30T11:56:07.064.822.000 to 2021-09-30T11:56:07.065.111.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.838852

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 53.7727853 to 53.7730743
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.9891830  -- OK for SL --
   MIU time from 2021-09-30T11:56:18.903.963.000 to 2021-09-30T11:56:18.904.252.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.989217

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 63.7622573 to 63.7625463
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.8791670  -- OK for SL --
   MIU time from 2021-09-30T11:56:28.893.469.000 to 2021-09-30T11:56:28.893.758.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.879221

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 79.6417133 to 79.6420023
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-30T11:56:44.772.979.000 to 2021-09-30T11:56:44.773.268.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.704273300 corrected by 40bit 10MHz counter of 11.868958900 obtained at 1Hz counter of 10. Result: 10.835317157
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-30T11:55:37.835.174.725
   Max 1st sample MIU time among all SL packets: 2021-09-30T11:56:44.773.127.725
   Min 1st sample XO time among all SL packets: 10.835176881
   Max 1st sample XO time among all SL packets: 77.773130576
   TM values from -250 to 205 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-30T11:55:37.835.174.725 (XO: 10.835176881 - 10.835482882, t=37.2 degC)  Rank: 00078
   TM values: Max =   172  Min =  -223  Ave =    -4.43  Std =    20.04
   Min TM value:-223 at an offset of 128.375 usec -> 2021-09-30T11:55:37.835.303.100 (XO: 10.835305257)
   Max TM value:172 at an offset of 130.988 usec -> 2021-09-30T11:55:37.835.305.712 (XO: 10.835307869)
   Max dTM/dt [1/us]:1.40e+03 at an offset of 130.825 usec -> 2021-09-30T11:55:37.835.305.550 (XO: 10.835307707)
   Min dTM/dt [1/us]:-1.59e+03 at an offset of 157.338 usec -> 2021-09-30T11:55:37.835.332.063 (XO: 10.835334219)

 - Block number: 2 starting at 2021-09-30T11:55:42.725.093.725 (XO: 15.725097218 - 15.725403219, t=37.2 degC)  Rank: 00078
   TM values: Max =   172  Min =  -222  Ave =    -4.39  Std =    18.17
   Min TM value:-222 at an offset of 126.800 usec -> 2021-09-30T11:55:42.725.220.525 (XO: 15.725224018)
   Max TM value:172 at an offset of 129.475 usec -> 2021-09-30T11:55:42.725.223.200 (XO: 15.725226693)
   Max dTM/dt [1/us]:1.40e+03 at an offset of 129.350 usec -> 2021-09-30T11:55:42.725.223.075 (XO: 15.725226568)
   Min dTM/dt [1/us]:-1.57e+03 at an offset of 113.838 usec -> 2021-09-30T11:55:42.725.207.562 (XO: 15.725211056)

 - Block number: 3 starting at 2021-09-30T11:56:07.064.681.725 (XO: 40.064682838 - 40.064988839, t=37.2 degC)  Rank: 00077
   TM values: Max =   199  Min =  -250  Ave =    -4.38  Std =    19.65
   Min TM value:-250 at an offset of 127.338 usec -> 2021-09-30T11:56:07.064.809.062 (XO: 40.064810176)
   Max TM value:199 at an offset of 125.075 usec -> 2021-09-30T11:56:07.064.806.800 (XO: 40.064807914)
   Max dTM/dt [1/us]:1.62e+03 at an offset of 124.913 usec -> 2021-09-30T11:56:07.064.806.637 (XO: 40.064807751)
   Min dTM/dt [1/us]:-1.68e+03 at an offset of 127.125 usec -> 2021-09-30T11:56:07.064.808.850 (XO: 40.064809964)

 - Block number: 4 starting at 2021-09-30T11:56:18.903.822.725 (XO: 51.903826508 - 51.904132509, t=37.2 degC)  Rank: 00082
   TM values: Max =   182  Min =  -248  Ave =    -4.46  Std =    17.81
   Min TM value:-248 at an offset of 128.288 usec -> 2021-09-30T11:56:18.903.951.012 (XO: 51.903954796)
   Max TM value:182 at an offset of 126.013 usec -> 2021-09-30T11:56:18.903.948.737 (XO: 51.903952521)
   Max dTM/dt [1/us]:1.55e+03 at an offset of 130.688 usec -> 2021-09-30T11:56:18.903.953.412 (XO: 51.903957196)
   Min dTM/dt [1/us]:-1.63e+03 at an offset of 128.088 usec -> 2021-09-30T11:56:18.903.950.812 (XO: 51.903954596)

 - Block number: 5 starting at 2021-09-30T11:56:28.893.328.725 (XO: 61.893332062 - 61.893638063, t=37.2 degC)  Rank: 00076
   TM values: Max =   205  Min =  -231  Ave =    -4.47  Std =    17.65
   Min TM value:-231 at an offset of 116.163 usec -> 2021-09-30T11:56:28.893.444.887 (XO: 61.893448225)
   Max TM value:205 at an offset of 113.825 usec -> 2021-09-30T11:56:28.893.442.550 (XO: 61.893445888)
   Max dTM/dt [1/us]:1.67e+03 at an offset of 113.713 usec -> 2021-09-30T11:56:28.893.442.437 (XO: 61.893445775)
   Min dTM/dt [1/us]:-1.74e+03 at an offset of 115.925 usec -> 2021-09-30T11:56:28.893.444.650 (XO: 61.893447988)

 - Block number: 6 starting at 2021-09-30T11:56:44.772.838.725 (XO: 77.772841575 - 77.773147576, t=37.2 degC)  Rank: 00075
   TM values: Max =   179  Min =  -226  Ave =    -4.47  Std =    19.54
   Min TM value:-226 at an offset of 115.925 usec -> 2021-09-30T11:56:44.772.954.650 (XO: 77.772957501)
   Max TM value:179 at an offset of 118.638 usec -> 2021-09-30T11:56:44.772.957.362 (XO: 77.772960213)
   Max dTM/dt [1/us]:1.49e+03 at an offset of 118.488 usec -> 2021-09-30T11:56:44.772.957.212 (XO: 77.772960063)
   Min dTM/dt [1/us]:-1.45e+03 at an offset of 103.313 usec -> 2021-09-30T11:56:44.772.942.037 (XO: 77.772944888)
