-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jan 24 12:47:30 2022 UTC
-- File 20210930_083350_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.718110 to 43.425456
-each separately from 9s to 43s and from 425167us to 718399us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.893127 in 1 packets: 18
      16.813352 in 1 packets: 36
MIU packet time from 2021-09-30T08:34:02.718.110.000 to 2021-09-30T08:34:36.425.456.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.893127 in 1 packets: 18
      16.813352 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.8936996 to 44.8935862
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999433 in 2 packets: 18, 36
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210930_083350_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.6118100 to 11.6120990
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.8930710  -- OK for SL --
   MIU time from 2021-09-30T08:34:02.718.110.000 to 2021-09-30T08:34:02.718.399.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.893127

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.5051700 to 28.5054590
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.8132950  -- OK for SL --
   MIU time from 2021-09-30T08:34:19.611.526.000 to 2021-09-30T08:34:19.611.815.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.813352

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 45.3187540 to 45.3190430
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-30T08:34:36.425.167.000 to 2021-09-30T08:34:36.425.456.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.611810000 corrected by 40bit 10MHz counter of 10.893699600 obtained at 1Hz counter of 9. Result: 9.718112770
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-30T08:34:02.717.969.725
   Max 1st sample MIU time among all SL packets: 2021-09-30T08:34:36.425.315.725
   Min 1st sample XO time among all SL packets: 9.717972494
   Max 1st sample XO time among all SL packets: 43.425317929
   TM values from -231 to 210 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-30T08:34:02.717.969.725 (XO: 9.717972494 - 9.718278495, t=37.2 degC)  Rank: 00077
   TM values: Max =   210  Min =  -231  Ave =    -4.50  Std =    22.75
   Min TM value:-231 at an offset of 123.025 usec -> 2021-09-30T08:34:02.718.092.750 (XO: 9.718095520)
   Max TM value:210 at an offset of 125.538 usec -> 2021-09-30T08:34:02.718.095.262 (XO: 9.718098032)
   Max dTM/dt [1/us]:1.98e+03 at an offset of 125.338 usec -> 2021-09-30T08:34:02.718.095.062 (XO: 9.718097832)
   Min dTM/dt [1/us]:-2.11e+03 at an offset of 122.825 usec -> 2021-09-30T08:34:02.718.092.550 (XO: 9.718095320)

 - Block number: 2 starting at 2021-09-30T08:34:19.611.385.725 (XO: 26.611388843 - 26.611694844, t=37.2 degC)  Rank: 00086
   TM values: Max =   176  Min =  -227  Ave =    -4.34  Std =    18.48
   Min TM value:-227 at an offset of 121.025 usec -> 2021-09-30T08:34:19.611.506.750 (XO: 26.611509868)
   Max TM value:176 at an offset of 123.700 usec -> 2021-09-30T08:34:19.611.509.425 (XO: 26.611512543)
   Max dTM/dt [1/us]:1.41e+03 at an offset of 123.500 usec -> 2021-09-30T08:34:19.611.509.225 (XO: 26.611512343)
   Min dTM/dt [1/us]:-1.71e+03 at an offset of 120.825 usec -> 2021-09-30T08:34:19.611.506.550 (XO: 26.611509668)

 - Block number: 3 starting at 2021-09-30T08:34:36.425.026.725 (XO: 43.425028928 - 43.425334929, t=37.2 degC)  Rank: 00080
   TM values: Max =   167  Min =  -217  Ave =    -4.50  Std =    19.59
   Min TM value:-217 at an offset of 97.050 usec -> 2021-09-30T08:34:36.425.123.775 (XO: 43.425125978)
   Max TM value:167 at an offset of 112.438 usec -> 2021-09-30T08:34:36.425.139.162 (XO: 43.425141365)
   Max dTM/dt [1/us]:1.30e+03 at an offset of 112.338 usec -> 2021-09-30T08:34:36.425.139.062 (XO: 43.425141265)
   Min dTM/dt [1/us]:-1.47e+03 at an offset of 96.913 usec -> 2021-09-30T08:34:36.425.123.637 (XO: 43.425125840)
