-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jan 24 08:19:53 2022 UTC
-- File 20210930_024224_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 13.645954 to 30.506251
-each separately from 13s to 30s and from 505962us to 826225us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.179693 in 1 packets: 18
      7.679737 in 1 packets: 36
MIU packet time from 2021-09-30T02:42:40.645.954.000 to 2021-09-30T02:42:57.506.251.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.179693 in 1 packets: 18
      7.679737 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.9039483 to 31.9038918
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999702 in 1 packets: 18
      7.9999733 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210930_024224_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.5499028 to 15.5501918
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.1796630  -- OK for SL --
   MIU time from 2021-09-30T02:42:40.645.954.000 to 2021-09-30T02:42:40.646.243.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.179693

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.7298548 to 24.7301438
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.6797110  -- OK for SL --
   MIU time from 2021-09-30T02:42:49.825.936.000 to 2021-09-30T02:42:49.826.225.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.679737

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.4098548 to 32.4101438
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-30T02:42:57.505.962.000 to 2021-09-30T02:42:57.506.251.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 13, 32bit 10MHz counter = 15.549902800 corrected by 40bit 10MHz counter of 14.903948300 obtained at 1Hz counter of 13. Result: 13.645956632
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996800
   Min 1st sample MIU time among all SL packets: 2021-09-30T02:42:40.645.813.725
   Max 1st sample MIU time among all SL packets: 2021-09-30T02:42:57.506.110.725
   Min 1st sample XO time among all SL packets: 13.645816356
   Max 1st sample XO time among all SL packets: 30.506113345
   TM values from -249 to 201 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-30T02:42:40.645.813.725 (XO: 13.645816356 - 13.646122357, t=37.2 degC)  Rank: 00085
   TM values: Max =   199  Min =  -245  Ave =    -4.40  Std =    25.95
   Min TM value:-245 at an offset of 116.100 usec -> 2021-09-30T02:42:40.645.929.825 (XO: 13.645932457)
   Max TM value:199 at an offset of 118.788 usec -> 2021-09-30T02:42:40.645.932.512 (XO: 13.645935144)
   Max dTM/dt [1/us]:1.58e+03 at an offset of 118.613 usec -> 2021-09-30T02:42:40.645.932.337 (XO: 13.645934969)
   Min dTM/dt [1/us]:-1.79e+03 at an offset of 115.900 usec -> 2021-09-30T02:42:40.645.929.625 (XO: 13.645932257)

 - Block number: 2 starting at 2021-09-30T02:42:49.825.795.725 (XO: 22.825798833 - 22.826104834, t=37.2 degC)  Rank: 00084
   TM values: Max =   184  Min =  -240  Ave =    -4.54  Std =    19.56
   Min TM value:-240 at an offset of 115.375 usec -> 2021-09-30T02:42:49.825.911.100 (XO: 22.825914208)
   Max TM value:184 at an offset of 130.488 usec -> 2021-09-30T02:42:49.825.926.212 (XO: 22.825929321)
   Max dTM/dt [1/us]:1.58e+03 at an offset of 130.325 usec -> 2021-09-30T02:42:49.825.926.050 (XO: 22.825929158)
   Min dTM/dt [1/us]:-2.11e+03 at an offset of 127.663 usec -> 2021-09-30T02:42:49.825.923.387 (XO: 22.825926496)

 - Block number: 3 starting at 2021-09-30T02:42:57.505.821.725 (XO: 30.505824344 - 30.506130345, t=37.2 degC)  Rank: 00080
   TM values: Max =   201  Min =  -249  Ave =    -4.58  Std =    24.52
   Min TM value:-249 at an offset of 99.388 usec -> 2021-09-30T02:42:57.505.921.112 (XO: 30.505923731)
   Max TM value:201 at an offset of 127.750 usec -> 2021-09-30T02:42:57.505.949.475 (XO: 30.505952094)
   Max dTM/dt [1/us]:1.69e+03 at an offset of 96.913 usec -> 2021-09-30T02:42:57.505.918.637 (XO: 30.505921256)
   Min dTM/dt [1/us]:-2.16e+03 at an offset of 127.875 usec -> 2021-09-30T02:42:57.505.949.600 (XO: 30.505952219)
