-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jan 24 05:17:02 2022 UTC
-- File 20210930_002053_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.297555 to 16.587888
-each separately from 3s to 16s and from 297555us to 587888us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.289755 in 1 packets: 18
MIU packet time from 2021-09-30T00:20:59.297.555.000 to 2021-09-30T00:21:12.587.888.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.289755 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.5877735 to 17.5877302
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.9999567 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210930_002053_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.8853298 to 4.8856188
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.2897110  -- OK for SL --
   MIU time from 2021-09-30T00:20:59.297.555.000 to 2021-09-30T00:20:59.297.844.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.289755

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.1753298 to 18.1756188
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-30T00:21:12.587.599.000 to 2021-09-30T00:21:12.587.888.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 4.885329800 corrected by 40bit 10MHz counter of 4.587773500 obtained at 1Hz counter of 3. Result: 3.297557282
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-30T00:20:59.297.414.725
   Max 1st sample MIU time among all SL packets: 2021-09-30T00:21:12.587.747.725
   Min 1st sample XO time among all SL packets: 3.297417006
   Max 1st sample XO time among all SL packets: 16.587750265
   TM values from -266 to 228 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-30T00:20:59.297.414.725 (XO: 3.297417006 - 3.297723007, t=37.2 degC)  Rank: 00078
   TM values: Max =   209  Min =  -266  Ave =    -4.49  Std =    19.74
   Min TM value:-266 at an offset of 123.838 usec -> 2021-09-30T00:20:59.297.538.562 (XO: 3.297540844)
   Max TM value:209 at an offset of 121.525 usec -> 2021-09-30T00:20:59.297.536.250 (XO: 3.297538532)
   Max dTM/dt [1/us]:1.91e+03 at an offset of 121.413 usec -> 2021-09-30T00:20:59.297.536.137 (XO: 3.297538419)
   Min dTM/dt [1/us]:-1.96e+03 at an offset of 123.675 usec -> 2021-09-30T00:20:59.297.538.400 (XO: 3.297540682)

 - Block number: 2 starting at 2021-09-30T00:21:12.587.458.725 (XO: 16.587461264 - 16.587767265, t=37.2 degC)  Rank: 00085
   TM values: Max =   228  Min =  -228  Ave =    -4.54  Std =    20.63
   Min TM value:-228 at an offset of 121.213 usec -> 2021-09-30T00:21:12.587.579.937 (XO: 16.587582477)
   Max TM value:228 at an offset of 123.813 usec -> 2021-09-30T00:21:12.587.582.537 (XO: 16.587585077)
   Max dTM/dt [1/us]:2.04e+03 at an offset of 123.675 usec -> 2021-09-30T00:21:12.587.582.400 (XO: 16.587584939)
   Min dTM/dt [1/us]:-1.88e+03 at an offset of 120.988 usec -> 2021-09-30T00:21:12.587.579.712 (XO: 16.587582252)
