-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jan 24 04:57:14 2022 UTC
-- File 20210929_235450_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 13.542227 to 59.881629
-each separately from 13s to 59s and from 292167us to 881629us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.749651 in 1 packets: 18
      25.059298 in 1 packets: 36
      15.529297 in 1 packets: 54
MIU packet time from 2021-09-29T23:55:06.542.227.000 to 2021-09-29T23:55:52.881.629.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.749651 in 1 packets: 18
      25.059298 in 1 packets: 36
      15.529297 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.9273665 to 60.9272134
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      5.9999800 in 1 packets: 18
      24.9999168 in 1 packets: 36
      14.9999501 in 1 packets: 54
Temperature FPGA: 
      04 -> 49.8 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 54 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 54 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210929_235450_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.4695941 to 15.4698831
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.7496310  -- OK for SL --
   MIU time from 2021-09-29T23:55:06.542.227.000 to 2021-09-29T23:55:06.542.516.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.749651

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.2195141 to 21.2198031
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.0592150  -- OK for SL --
   MIU time from 2021-09-29T23:55:12.292.167.000 to 2021-09-29T23:55:12.292.456.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.059298

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.2790181 to 46.2793071
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.5292470  -- OK for SL --
   MIU time from 2021-09-29T23:55:37.351.754.000 to 2021-09-29T23:55:37.352.043.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.529297

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 61.8085541 to 61.8088431
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-29T23:55:52.881.340.000 to 2021-09-29T23:55:52.881.629.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 13, 32bit 10MHz counter = 15.469594100 corrected by 40bit 10MHz counter of 14.927366500 obtained at 1Hz counter of 13. Result: 13.542229389
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-29T23:55:06.542.086.725
   Max 1st sample MIU time among all SL packets: 2021-09-29T23:55:52.881.488.725
   Min 1st sample XO time among all SL packets: 13.542089114
   Max 1st sample XO time among all SL packets: 59.881492334
   TM values from -282 to 246 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-29T23:55:06.542.086.725 (XO: 13.542089114 - 13.542395115, t=37.2 degC)  Rank: 00081
   TM values: Max =   221  Min =  -270  Ave =    -4.52  Std =    21.94
   Min TM value:-270 at an offset of 123.688 usec -> 2021-09-29T23:55:06.542.210.412 (XO: 13.542212802)
   Max TM value:221 at an offset of 123.375 usec -> 2021-09-29T23:55:06.542.210.100 (XO: 13.542212489)
   Max dTM/dt [1/us]:2.16e+03 at an offset of 126.050 usec -> 2021-09-29T23:55:06.542.212.775 (XO: 13.542215164)
   Min dTM/dt [1/us]:-2.34e+03 at an offset of 123.500 usec -> 2021-09-29T23:55:06.542.210.225 (XO: 13.542212614)

 - Block number: 2 starting at 2021-09-29T23:55:12.292.026.725 (XO: 19.292028289 - 19.292334290, t=37.2 degC)  Rank: 00078
   TM values: Max =   246  Min =  -269  Ave =    -4.40  Std =    28.03
   Min TM value:-269 at an offset of 125.513 usec -> 2021-09-29T23:55:12.292.152.237 (XO: 19.292153802)
   Max TM value:246 at an offset of 166.488 usec -> 2021-09-29T23:55:12.292.193.213 (XO: 19.292194777)
   Max dTM/dt [1/us]:2.26e+03 at an offset of 166.363 usec -> 2021-09-29T23:55:12.292.193.088 (XO: 19.292194652)
   Min dTM/dt [1/us]:-2.21e+03 at an offset of 168.726 usec -> 2021-09-29T23:55:12.292.195.450 (XO: 19.292197014)

 - Block number: 3 starting at 2021-09-29T23:55:37.351.613.725 (XO: 44.351615720 - 44.351921722, t=37.2 degC)  Rank: 00080
   TM values: Max =   235  Min =  -282  Ave =    -4.45  Std =    24.96
   Min TM value:-282 at an offset of 119.525 usec -> 2021-09-29T23:55:37.351.733.250 (XO: 44.351735246)
   Max TM value:235 at an offset of 117.538 usec -> 2021-09-29T23:55:37.351.731.262 (XO: 44.351733258)
   Max dTM/dt [1/us]:2.08e+03 at an offset of 122.075 usec -> 2021-09-29T23:55:37.351.735.800 (XO: 44.351737796)
   Min dTM/dt [1/us]:-2.05e+03 at an offset of 119.413 usec -> 2021-09-29T23:55:37.351.733.137 (XO: 44.351735133)

 - Block number: 4 starting at 2021-09-29T23:55:52.881.199.725 (XO: 59.881203333 - 59.881509334, t=37.2 degC)  Rank: 00082
   TM values: Max =   231  Min =  -261  Ave =    -4.39  Std =    20.89
   Min TM value:-261 at an offset of 110.638 usec -> 2021-09-29T23:55:52.881.310.362 (XO: 59.881313971)
   Max TM value:231 at an offset of 113.200 usec -> 2021-09-29T23:55:52.881.312.925 (XO: 59.881316533)
   Max dTM/dt [1/us]:2.11e+03 at an offset of 113.038 usec -> 2021-09-29T23:55:52.881.312.762 (XO: 59.881316371)
   Min dTM/dt [1/us]:-2.12e+03 at an offset of 110.475 usec -> 2021-09-29T23:55:52.881.310.200 (XO: 59.881313808)
