-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jan 23 12:41:31 2022 UTC
-- File 20210929_114205_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.296155 to 30.925685
-each separately from 8s to 30s and from 296155us to 925685us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.459386 in 1 packets: 18
      9.169277 in 1 packets: 36
MIU packet time from 2021-09-29T11:42:16.296.155.000 to 2021-09-29T11:42:38.925.685.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.459386 in 1 packets: 18
      9.169277 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.9263264 to 31.9262531
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.9999567 in 1 packets: 18
      8.9999700 in 1 packets: 36
Temperature FPGA: 
      03 -> 51.1 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 53 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 53 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210929_114205_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.2224817 to 10.2227707
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.4593430  -- OK for SL --
   MIU time from 2021-09-29T11:42:16.296.155.000 to 2021-09-29T11:42:16.296.444.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.459386

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.6821137 to 23.6824027
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.1692470  -- OK for SL --
   MIU time from 2021-09-29T11:42:29.755.830.000 to 2021-09-29T11:42:29.756.119.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.169277

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.8516497 to 32.8519387
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-29T11:42:38.925.396.000 to 2021-09-29T11:42:38.925.685.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 10.222481700 corrected by 40bit 10MHz counter of 9.926326400 obtained at 1Hz counter of 8. Result: 8.296156277
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-29T11:42:16.296.014.725
   Max 1st sample MIU time among all SL packets: 2021-09-29T11:42:38.925.544.725
   Min 1st sample XO time among all SL packets: 8.296016002
   Max 1st sample XO time among all SL packets: 30.925548379
   TM values from -247 to 205 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-29T11:42:16.296.014.725 (XO: 8.296016002 - 8.296322003, t=37.2 degC)  Rank: 00083
   TM values: Max =   205  Min =  -247  Ave =    -4.50  Std =    19.03
   Min TM value:-247 at an offset of 118.463 usec -> 2021-09-29T11:42:16.296.133.187 (XO: 8.296134465)
   Max TM value:205 at an offset of 118.188 usec -> 2021-09-29T11:42:16.296.132.912 (XO: 8.296134190)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 120.825 usec -> 2021-09-29T11:42:16.296.135.550 (XO: 8.296136827)
   Min dTM/dt [1/us]:-2.32e+03 at an offset of 118.300 usec -> 2021-09-29T11:42:16.296.133.025 (XO: 8.296134302)

 - Block number: 2 starting at 2021-09-29T11:42:29.755.689.725 (XO: 21.755692819 - 21.755998820, t=37.2 degC)  Rank: 00080
   TM values: Max =   178  Min =  -216  Ave =    -4.46  Std =    25.42
   Min TM value:-216 at an offset of 223.163 usec -> 2021-09-29T11:42:29.755.912.888 (XO: 21.755915982)
   Max TM value:178 at an offset of 127.213 usec -> 2021-09-29T11:42:29.755.816.937 (XO: 21.755820032)
   Max dTM/dt [1/us]:1.49e+03 at an offset of 220.676 usec -> 2021-09-29T11:42:29.755.910.400 (XO: 21.755913495)
   Min dTM/dt [1/us]:-1.74e+03 at an offset of 124.338 usec -> 2021-09-29T11:42:29.755.814.062 (XO: 21.755817157)

 - Block number: 3 starting at 2021-09-29T11:42:38.925.255.725 (XO: 30.925259378 - 30.925565379, t=37.2 degC)  Rank: 00079
   TM values: Max =   194  Min =  -243  Ave =    -4.54  Std =    23.80
   Min TM value:-243 at an offset of 167.863 usec -> 2021-09-29T11:42:38.925.423.588 (XO: 30.925427241)
   Max TM value:194 at an offset of 129.375 usec -> 2021-09-29T11:42:38.925.385.100 (XO: 30.925388754)
   Max dTM/dt [1/us]:1.61e+03 at an offset of 124.488 usec -> 2021-09-29T11:42:38.925.380.212 (XO: 30.925383866)
   Min dTM/dt [1/us]:-1.77e+03 at an offset of 126.675 usec -> 2021-09-29T11:42:38.925.382.400 (XO: 30.925386054)
