-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jan 23 12:26:18 2022 UTC
-- File 20210929_111920_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.749941 to 45.038171
-each separately from 5s to 45s and from 37882us to 750230us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.549185 in 1 packets: 18
      24.738178 in 1 packets: 36
MIU packet time from 2021-09-29T11:19:28.749.941.000 to 2021-09-29T11:20:08.038.171.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.549185 in 1 packets: 18
      24.738178 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.8135444 to 45.8134110
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999500 in 1 packets: 18
      24.9999166 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      03 -> 51.1 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      2 in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      2 in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210929_111920_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.5634861 to 6.5637751
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.5491350  -- OK for SL --
   MIU time from 2021-09-29T11:19:28.749.941.000 to 2021-09-29T11:19:28.750.230.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.549185

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      94 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.1129101 to 21.1131991
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.7380950  -- OK for SL --
   MIU time from 2021-09-29T11:19:43.299.415.000 to 2021-09-29T11:19:43.299.704.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.738178

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 45.8512941 to 45.8515831
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-29T11:20:08.037.882.000 to 2021-09-29T11:20:08.038.171.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 45 XO calibration points at 1Hz counter from 0 to 44
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 6.563486100 corrected by 40bit 10MHz counter of 5.813544400 obtained at 1Hz counter of 5. Result: 5.749944175
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-29T11:19:28.749.800.725
   Max 1st sample MIU time among all SL packets: 2021-09-29T11:20:08.038.030.725
   Min 1st sample XO time among all SL packets: 5.749803899
   Max 1st sample XO time among all SL packets: 45.038031951
   TM values from -249 to 217 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-29T11:19:28.749.800.725 (XO: 5.749803899 - 5.750109900, t=37.2 degC)  Rank: 00080
   TM values: Max =   217  Min =  -249  Ave =    -4.44  Std =    23.02
   Min TM value:-249 at an offset of 108.538 usec -> 2021-09-29T11:19:28.749.909.262 (XO: 5.749912437)
   Max TM value:217 at an offset of 108.238 usec -> 2021-09-29T11:19:28.749.908.962 (XO: 5.749912137)
   Max dTM/dt [1/us]:1.93e+03 at an offset of 106.200 usec -> 2021-09-29T11:19:28.749.906.925 (XO: 5.749910100)
   Min dTM/dt [1/us]:-2.15e+03 at an offset of 108.350 usec -> 2021-09-29T11:19:28.749.909.075 (XO: 5.749912250)

 - Block number: 2 starting at 2021-09-29T11:19:43.299.274.725 (XO: 20.299276413 - 20.299582414, t=37.2 degC)  Rank: 00094
   TM values: Max =   194  Min =  -240  Ave =    -4.59  Std =    25.32
   Min TM value:-240 at an offset of 147.975 usec -> 2021-09-29T11:19:43.299.422.700 (XO: 20.299424388)
   Max TM value:194 at an offset of 119.488 usec -> 2021-09-29T11:19:43.299.394.212 (XO: 20.299395901)
   Max dTM/dt [1/us]:1.64e+03 at an offset of 150.363 usec -> 2021-09-29T11:19:43.299.425.087 (XO: 20.299426776)
   Min dTM/dt [1/us]:-1.88e+03 at an offset of 116.688 usec -> 2021-09-29T11:19:43.299.391.412 (XO: 20.299393100)

 - Block number: 3 starting at 2021-09-29T11:20:08.037.741.725 (XO: 45.037742950 - 45.038048951, t=37.2 degC)  Rank: 00080
   TM values: Max =   174  Min =  -225  Ave =    -4.56  Std =    20.76
   Min TM value:-225 at an offset of 105.900 usec -> 2021-09-29T11:20:08.037.847.625 (XO: 45.037848850)
   Max TM value:174 at an offset of 108.550 usec -> 2021-09-29T11:20:08.037.850.275 (XO: 45.037851500)
   Max dTM/dt [1/us]:1.35e+03 at an offset of 108.375 usec -> 2021-09-29T11:20:08.037.850.100 (XO: 45.037851325)
   Min dTM/dt [1/us]:-1.56e+03 at an offset of 105.688 usec -> 2021-09-29T11:20:08.037.847.412 (XO: 45.037848637)
