-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jan 23 11:56:05 2022 UTC
-- File 20210929_104703_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.251510 to 52.373348
-each separately from 0s to 52s and from 251510us to 555289us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      26.044262 in 1 packets: 18
      13.258650 in 1 packets: 36
      12.817770 in 1 packets: 54
MIU packet time from 2021-09-29T10:47:06.251.510.000 to 2021-09-29T10:47:58.373.348.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      26.044262 in 1 packets: 18
      13.258650 in 1 packets: 36
      12.817770 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.9278368 to 53.9276634
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.9999132 in 1 packets: 18
      12.9999567 in 2 packets: 36, 54
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 1 packets: 55
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 1 packets: 55
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210929_104703_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.1793476 to 2.1796366
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 26.0441750  -- OK for SL --
   MIU time from 2021-09-29T10:47:06.251.510.000 to 2021-09-29T10:47:06.251.799.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 26.044262

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.2238116 to 28.2241006
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.2586070  -- OK for SL --
   MIU time from 2021-09-29T10:47:32.296.061.000 to 2021-09-29T10:47:32.296.350.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.258650

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.4827076 to 41.4829966
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.8177270  -- OK for SL --
   MIU time from 2021-09-29T10:47:45.555.000.000 to 2021-09-29T10:47:45.555.289.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.817770

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      89 in 18 packets (all packets)  -- OK --
   10 MHz counter from 54.3007236 to 54.3010126
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-29T10:47:58.373.059.000 to 2021-09-29T10:47:58.373.348.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.179347600 corrected by 40bit 10MHz counter of 1.927836800 obtained at 1Hz counter of 0. Result: 0.251511655
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-29T10:47:06.251.369.725
   Max 1st sample MIU time among all SL packets: 2021-09-29T10:47:58.373.207.725
   Min 1st sample XO time among all SL packets: 0.251371380
   Max 1st sample XO time among all SL packets: 52.373210157
   TM values from -237 to 178 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-29T10:47:06.251.369.725 (XO: 0.251371380 - 0.251677381, t=37.2 degC)  Rank: 00078
   TM values: Max =   167  Min =  -184  Ave =    -4.42  Std =    18.24
   Min TM value:-184 at an offset of 126.800 usec -> 2021-09-29T10:47:06.251.496.525 (XO: 0.251498180)
   Max TM value:167 at an offset of 129.575 usec -> 2021-09-29T10:47:06.251.499.300 (XO: 0.251500955)
   Max dTM/dt [1/us]:1.39e+03 at an offset of 129.450 usec -> 2021-09-29T10:47:06.251.499.175 (XO: 0.251500830)
   Min dTM/dt [1/us]:-1.29e+03 at an offset of 114.288 usec -> 2021-09-29T10:47:06.251.484.012 (XO: 0.251485668)

 - Block number: 2 starting at 2021-09-29T10:47:32.295.920.725 (XO: 26.295922302 - 26.296228303, t=37.2 degC)  Rank: 00080
   TM values: Max =   178  Min =  -233  Ave =    -4.41  Std =    17.69
   Min TM value:-233 at an offset of 129.213 usec -> 2021-09-29T10:47:32.296.049.937 (XO: 26.296051514)
   Max TM value:178 at an offset of 131.738 usec -> 2021-09-29T10:47:32.296.052.462 (XO: 26.296054039)
   Max dTM/dt [1/us]:1.83e+03 at an offset of 131.575 usec -> 2021-09-29T10:47:32.296.052.300 (XO: 26.296053877)
   Min dTM/dt [1/us]:-1.91e+03 at an offset of 128.988 usec -> 2021-09-29T10:47:32.296.049.712 (XO: 26.296051289)

 - Block number: 3 starting at 2021-09-29T10:47:45.554.859.725 (XO: 39.554862456 - 39.555168457, t=37.2 degC)  Rank: 00076
   TM values: Max =   175  Min =  -227  Ave =    -4.56  Std =    19.70
   Min TM value:-227 at an offset of 116.025 usec -> 2021-09-29T10:47:45.554.975.750 (XO: 39.554978481)
   Max TM value:175 at an offset of 115.700 usec -> 2021-09-29T10:47:45.554.975.425 (XO: 39.554978156)
   Max dTM/dt [1/us]:1.75e+03 at an offset of 118.438 usec -> 2021-09-29T10:47:45.554.978.162 (XO: 39.554980894)
   Min dTM/dt [1/us]:-1.70e+03 at an offset of 115.825 usec -> 2021-09-29T10:47:45.554.975.550 (XO: 39.554978281)

 - Block number: 4 starting at 2021-09-29T10:47:58.372.918.725 (XO: 52.372921156 - 52.373227157, t=37.2 degC)  Rank: 00089
   TM values: Max =   177  Min =  -237  Ave =    -4.39  Std =    19.09
   Min TM value:-237 at an offset of 127.313 usec -> 2021-09-29T10:47:58.373.046.037 (XO: 52.373048469)
   Max TM value:177 at an offset of 129.925 usec -> 2021-09-29T10:47:58.373.048.650 (XO: 52.373051081)
   Max dTM/dt [1/us]:1.67e+03 at an offset of 129.738 usec -> 2021-09-29T10:47:58.373.048.462 (XO: 52.373050894)
   Min dTM/dt [1/us]:-1.86e+03 at an offset of 127.138 usec -> 2021-09-29T10:47:58.373.045.862 (XO: 52.373048294)
