-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jan 23 11:24:30 2022 UTC
-- File 20210929_102441_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.795142 to 23.152101
-each separately from 9s to 23s and from 151812us to 795431us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.356381 in 1 packets: 18
MIU packet time from 2021-09-29T10:24:53.795.142.000 to 2021-09-29T10:25:07.152.101.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.356381 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.9263239 to 24.9262772
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999533 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210929_102441_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.7214662 to 11.7217552
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.3563350  -- OK for SL --
   MIU time from 2021-09-29T10:24:53.795.142.000 to 2021-09-29T10:24:53.795.431.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.356381

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.0780902 to 25.0783792
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-29T10:25:07.151.812.000 to 2021-09-29T10:25:07.152.101.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 11.721466200 corrected by 40bit 10MHz counter of 10.926323900 obtained at 1Hz counter of 9. Result: 9.795144924
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-29T10:24:53.795.001.725
   Max 1st sample MIU time among all SL packets: 2021-09-29T10:25:07.151.960.725
   Min 1st sample XO time among all SL packets: 9.795004649
   Max 1st sample XO time among all SL packets: 23.151962226
   TM values from -250 to 204 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-29T10:24:53.795.001.725 (XO: 9.795004649 - 9.795310650, t=37.2 degC)  Rank: 00084
   TM values: Max =   204  Min =  -250  Ave =    -4.55  Std =    21.11
   Min TM value:-250 at an offset of 120.963 usec -> 2021-09-29T10:24:53.795.122.687 (XO: 9.795125611)
   Max TM value:204 at an offset of 123.463 usec -> 2021-09-29T10:24:53.795.125.187 (XO: 9.795128111)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 123.288 usec -> 2021-09-29T10:24:53.795.125.012 (XO: 9.795127936)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 120.763 usec -> 2021-09-29T10:24:53.795.122.487 (XO: 9.795125411)

 - Block number: 2 starting at 2021-09-29T10:25:07.151.671.725 (XO: 23.151673226 - 23.151979227, t=37.2 degC)  Rank: 00087
   TM values: Max =   178  Min =  -240  Ave =    -4.53  Std =    19.34
   Min TM value:-240 at an offset of 108.913 usec -> 2021-09-29T10:25:07.151.780.637 (XO: 23.151782138)
   Max TM value:178 at an offset of 111.500 usec -> 2021-09-29T10:25:07.151.783.225 (XO: 23.151784726)
   Max dTM/dt [1/us]:1.55e+03 at an offset of 111.325 usec -> 2021-09-29T10:25:07.151.783.050 (XO: 23.151784551)
   Min dTM/dt [1/us]:-1.75e+03 at an offset of 108.713 usec -> 2021-09-29T10:25:07.151.780.437 (XO: 23.151781938)
