-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan 22 11:46:28 2022 UTC
-- File 20210927_132238_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.992152 to 64.925967
-each separately from 0s to 64s and from 295561us to 994484us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      28.341552 in 1 packets: 18
      3.659913 in 1 packets: 36
      25.301077 in 1 packets: 54
      6.629828 in 1 packets: 72
MIU packet time from 2021-09-27T13:22:41.992.152.000 to 2021-09-27T13:23:45.925.967.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      28.341552 in 1 packets: 18
      3.659913 in 1 packets: 36
      25.301077 in 1 packets: 54
      6.629828 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.8536803 to 65.8534668
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      28.9999031 in 1 packets: 18
      2.9999901 in 1 packets: 36
      25.9999133 in 1 packets: 54
      5.9999800 in 1 packets: 72
Temperature FPGA: 
      04 -> 49.8 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 73 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Number of generated alerts: 
      0 in 73 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20210927_132238_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.8458330 to 2.8461220
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 28.3414550  -- OK for SL --
   MIU time from 2021-09-27T13:22:41.992.152.000 to 2021-09-27T13:22:41.992.441.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 28.341552

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.1875770 to 31.1878660
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 3.6599030  -- OK for SL --
   MIU time from 2021-09-27T13:23:10.333.993.000 to 2021-09-27T13:23:10.334.282.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 3.659913

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.8477690 to 34.8480580
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.3009910  -- OK for SL --
   MIU time from 2021-09-27T13:23:13.994.195.000 to 2021-09-27T13:23:13.994.484.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.301077

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 60.1490490 to 60.1493380
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.6298070  -- OK for SL --
   MIU time from 2021-09-27T13:23:39.295.561.000 to 2021-09-27T13:23:39.295.850.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.629828

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 66.7791450 to 66.7794340
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-27T13:23:45.925.678.000 to 2021-09-27T13:23:45.925.967.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.845833000 corrected by 40bit 10MHz counter of 1.853680300 obtained at 1Hz counter of 0. Result: 0.992156073
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-27T13:22:41.992.011.725
   Max 1st sample MIU time among all SL packets: 2021-09-27T13:23:45.925.826.725
   Min 1st sample XO time among all SL packets: 0.992015798
   Max 1st sample XO time among all SL packets: 64.925830073
   TM values from -244 to 205 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-27T13:22:41.992.011.725 (XO: 0.992015798 - 0.992321799, t=37.2 degC)  Rank: 00078
   TM values: Max =   201  Min =  -244  Ave =    -4.53  Std =    22.75
   Min TM value:-244 at an offset of 123.713 usec -> 2021-09-27T13:22:41.992.135.437 (XO: 0.992139511)
   Max TM value:201 at an offset of 126.225 usec -> 2021-09-27T13:22:41.992.137.950 (XO: 0.992142023)
   Max dTM/dt [1/us]:2.09e+03 at an offset of 126.050 usec -> 2021-09-27T13:22:41.992.137.775 (XO: 0.992141848)
   Min dTM/dt [1/us]:-2.37e+03 at an offset of 123.538 usec -> 2021-09-27T13:22:41.992.135.262 (XO: 0.992139336)

 - Block number: 2 starting at 2021-09-27T13:23:10.333.852.725 (XO: 29.333854427 - 29.334160428, t=37.2 degC)  Rank: 00080
   TM values: Max =   202  Min =  -236  Ave =    -4.59  Std =    23.10
   Min TM value:-236 at an offset of 112.463 usec -> 2021-09-27T13:23:10.333.965.187 (XO: 29.333966890)
   Max TM value:202 at an offset of 112.188 usec -> 2021-09-27T13:23:10.333.964.912 (XO: 29.333966615)
   Max dTM/dt [1/us]:1.89e+03 at an offset of 114.825 usec -> 2021-09-27T13:23:10.333.967.550 (XO: 29.333969252)
   Min dTM/dt [1/us]:-2.28e+03 at an offset of 112.313 usec -> 2021-09-27T13:23:10.333.965.037 (XO: 29.333966740)

 - Block number: 3 starting at 2021-09-27T13:23:13.994.054.725 (XO: 32.994058505 - 32.994364506, t=37.2 degC)  Rank: 00078
   TM values: Max =   190  Min =  -232  Ave =    -4.72  Std =    24.41
   Min TM value:-232 at an offset of 124.463 usec -> 2021-09-27T13:23:13.994.179.187 (XO: 32.994182968)
   Max TM value:190 at an offset of 124.163 usec -> 2021-09-27T13:23:13.994.178.887 (XO: 32.994182668)
   Max dTM/dt [1/us]:1.91e+03 at an offset of 126.850 usec -> 2021-09-27T13:23:13.994.181.575 (XO: 32.994185356)
   Min dTM/dt [1/us]:-2.04e+03 at an offset of 124.275 usec -> 2021-09-27T13:23:13.994.179.000 (XO: 32.994182781)

 - Block number: 4 starting at 2021-09-27T13:23:39.295.420.725 (XO: 58.295422900 - 58.295728901, t=37.2 degC)  Rank: 00078
   TM values: Max =   199  Min =  -241  Ave =    -4.56  Std =    23.99
   Min TM value:-241 at an offset of 111.638 usec -> 2021-09-27T13:23:39.295.532.362 (XO: 58.295534538)
   Max TM value:199 at an offset of 111.313 usec -> 2021-09-27T13:23:39.295.532.037 (XO: 58.295534213)
   Max dTM/dt [1/us]:1.95e+03 at an offset of 113.975 usec -> 2021-09-27T13:23:39.295.534.700 (XO: 58.295536875)
   Min dTM/dt [1/us]:-2.16e+03 at an offset of 111.425 usec -> 2021-09-27T13:23:39.295.532.150 (XO: 58.295534325)

 - Block number: 5 starting at 2021-09-27T13:23:45.925.537.725 (XO: 64.925541072 - 64.925847073, t=37.2 degC)  Rank: 00083
   TM values: Max =   205  Min =  -232  Ave =    -4.58  Std =    21.44
   Min TM value:-232 at an offset of 111.975 usec -> 2021-09-27T13:23:45.925.649.700 (XO: 64.925653047)
   Max TM value:205 at an offset of 114.538 usec -> 2021-09-27T13:23:45.925.652.262 (XO: 64.925655610)
   Max dTM/dt [1/us]:1.91e+03 at an offset of 109.688 usec -> 2021-09-27T13:23:45.925.647.412 (XO: 64.925650760)
   Min dTM/dt [1/us]:-2.05e+03 at an offset of 111.850 usec -> 2021-09-27T13:23:45.925.649.575 (XO: 64.925652922)
