-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan 22 11:42:44 2022 UTC
-- File 20210927_132058_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.426411 to 26.245377
-each separately from 5s to 26s and from 245088us to 426700us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.818388 in 1 packets: 18
MIU packet time from 2021-09-27T13:21:06.426.411.000 to 2021-09-27T13:21:27.245.377.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.818388 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.8497726 to 27.8497023
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.9999297 in 1 packets: 18
Temperature FPGA: 
      06 -> 47.2 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210927_132058_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.2761830 to 7.2764720
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.8183190  -- OK for SL --
   MIU time from 2021-09-27T13:21:06.426.411.000 to 2021-09-27T13:21:06.426.700.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.818388

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.0947910 to 28.0950800
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-27T13:21:27.245.088.000 to 2021-09-27T13:21:27.245.377.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.276183000 corrected by 40bit 10MHz counter of 6.849772600 obtained at 1Hz counter of 5. Result: 5.426411807
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-27T13:21:06.426.270.725
   Max 1st sample MIU time among all SL packets: 2021-09-27T13:21:27.245.236.725
   Min 1st sample XO time among all SL packets: 5.426271532
   Max 1st sample XO time among all SL packets: 26.245238234
   TM values from -237 to 198 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-27T13:21:06.426.270.725 (XO: 5.426271532 - 5.426577533, t=37.2 degC)  Rank: 00082
   TM values: Max =   185  Min =  -231  Ave =    -4.50  Std =    22.17
   Min TM value:-231 at an offset of 124.363 usec -> 2021-09-27T13:21:06.426.395.087 (XO: 5.426395895)
   Max TM value:185 at an offset of 126.963 usec -> 2021-09-27T13:21:06.426.397.687 (XO: 5.426398495)
   Max dTM/dt [1/us]:1.62e+03 at an offset of 126.763 usec -> 2021-09-27T13:21:06.426.397.487 (XO: 5.426398295)
   Min dTM/dt [1/us]:-1.61e+03 at an offset of 124.150 usec -> 2021-09-27T13:21:06.426.394.875 (XO: 5.426395682)

 - Block number: 2 starting at 2021-09-27T13:21:27.244.947.725 (XO: 26.244949233 - 26.245255234, t=37.2 degC)  Rank: 00078
   TM values: Max =   198  Min =  -237  Ave =    -4.60  Std =    18.75
   Min TM value:-237 at an offset of 124.100 usec -> 2021-09-27T13:21:27.245.071.825 (XO: 26.245073334)
   Max TM value:198 at an offset of 123.800 usec -> 2021-09-27T13:21:27.245.071.525 (XO: 26.245073034)
   Max dTM/dt [1/us]:1.93e+03 at an offset of 126.488 usec -> 2021-09-27T13:21:27.245.074.212 (XO: 26.245075721)
   Min dTM/dt [1/us]:-2.13e+03 at an offset of 123.913 usec -> 2021-09-27T13:21:27.245.071.637 (XO: 26.245073146)
