-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan 22 11:35:31 2022 UTC
-- File 20210927_131626_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.438929 to 30.101232
-each separately from 2s to 30s and from 100943us to 570526us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.131019 in 1 packets: 18
      14.530417 in 1 packets: 36
MIU packet time from 2021-09-27T13:16:31.438.929.000 to 2021-09-27T13:16:59.101.232.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.131019 in 1 packets: 18
      14.530417 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.8620445 to 31.8619511
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.9999566 in 1 packets: 18
      14.9999500 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210927_131626_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.3009748 to 4.3012638
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.1309750  -- OK for SL --
   MIU time from 2021-09-27T13:16:31.438.929.000 to 2021-09-27T13:16:31.439.218.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.131019

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.4322388 to 17.4325278
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.5303670  -- OK for SL --
   MIU time from 2021-09-27T13:16:44.570.237.000 to 2021-09-27T13:16:44.570.526.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.530417

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.9628948 to 31.9631838
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-27T13:16:59.100.943.000 to 2021-09-27T13:16:59.101.232.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.300974800 corrected by 40bit 10MHz counter of 3.862044500 obtained at 1Hz counter of 2. Result: 2.438931792
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-27T13:16:31.438.788.725
   Max 1st sample MIU time among all SL packets: 2021-09-27T13:16:59.101.091.725
   Min 1st sample XO time among all SL packets: 2.438791517
   Max 1st sample XO time among all SL packets: 30.101092769
   TM values from -257 to 216 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-27T13:16:31.438.788.725 (XO: 2.438791517 - 2.439097518, t=37.2 degC)  Rank: 00080
   TM values: Max =   216  Min =  -257  Ave =    -4.57  Std =    23.96
   Min TM value:-257 at an offset of 155.738 usec -> 2021-09-27T13:16:31.438.944.463 (XO: 2.438947255)
   Max TM value:216 at an offset of 125.725 usec -> 2021-09-27T13:16:31.438.914.450 (XO: 2.438917242)
   Max dTM/dt [1/us]:1.75e+03 at an offset of 125.538 usec -> 2021-09-27T13:16:31.438.914.262 (XO: 2.438917055)
   Min dTM/dt [1/us]:-2.22e+03 at an offset of 127.613 usec -> 2021-09-27T13:16:31.438.916.337 (XO: 2.438919130)

 - Block number: 2 starting at 2021-09-27T13:16:44.570.096.725 (XO: 15.570099306 - 15.570405307, t=37.2 degC)  Rank: 00084
   TM values: Max =   197  Min =  -257  Ave =    -4.51  Std =    20.87
   Min TM value:-257 at an offset of 124.125 usec -> 2021-09-27T13:16:44.570.220.850 (XO: 15.570223432)
   Max TM value:197 at an offset of 123.825 usec -> 2021-09-27T13:16:44.570.220.550 (XO: 15.570223132)
   Max dTM/dt [1/us]:2.01e+03 at an offset of 126.475 usec -> 2021-09-27T13:16:44.570.223.200 (XO: 15.570225782)
   Min dTM/dt [1/us]:-1.91e+03 at an offset of 123.938 usec -> 2021-09-27T13:16:44.570.220.662 (XO: 15.570223244)

 - Block number: 3 starting at 2021-09-27T13:16:59.100.802.725 (XO: 30.100803768 - 30.101109769, t=37.2 degC)  Rank: 00087
   TM values: Max =   177  Min =  -235  Ave =    -4.44  Std =    19.36
   Min TM value:-235 at an offset of 112.050 usec -> 2021-09-27T13:16:59.100.914.775 (XO: 30.100915818)
   Max TM value:177 at an offset of 114.638 usec -> 2021-09-27T13:16:59.100.917.362 (XO: 30.100918406)
   Max dTM/dt [1/us]:1.58e+03 at an offset of 114.425 usec -> 2021-09-27T13:16:59.100.917.150 (XO: 30.100918193)
   Min dTM/dt [1/us]:-1.83e+03 at an offset of 111.825 usec -> 2021-09-27T13:16:59.100.914.550 (XO: 30.100915593)
