-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan 22 09:16:42 2022 UTC
-- File 20210927_113245_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.394963 to 21.004147
-each separately from 2s to 21s and from 3858us to 395252us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.608606 in 1 packets: 18
MIU packet time from 2021-09-27T11:32:50.394.963.000 to 2021-09-27T11:33:09.004.147.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.608606 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.9063327 to 22.9062695
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.9999368 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210927_113245_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.3012968 to 4.3015858
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.6085430  -- OK for SL --
   MIU time from 2021-09-27T11:32:50.394.963.000 to 2021-09-27T11:32:50.395.252.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.608606

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.9101288 to 22.9104178
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-27T11:33:09.003.858.000 to 2021-09-27T11:33:09.004.147.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.301296800 corrected by 40bit 10MHz counter of 3.906332700 obtained at 1Hz counter of 2. Result: 2.394965403
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-27T11:32:50.394.822.725
   Max 1st sample MIU time among all SL packets: 2021-09-27T11:33:09.004.006.725
   Min 1st sample XO time among all SL packets: 2.394825128
   Max 1st sample XO time among all SL packets: 21.004008039
   TM values from -225 to 190 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-27T11:32:50.394.822.725 (XO: 2.394825128 - 2.395131129, t=37.2 degC)  Rank: 00082
   TM values: Max =   176  Min =  -222  Ave =    -4.57  Std =    18.27
   Min TM value:-222 at an offset of 127.625 usec -> 2021-09-27T11:32:50.394.950.350 (XO: 2.394952753)
   Max TM value:176 at an offset of 130.175 usec -> 2021-09-27T11:32:50.394.952.900 (XO: 2.394955303)
   Max dTM/dt [1/us]:1.59e+03 at an offset of 129.988 usec -> 2021-09-27T11:32:50.394.952.712 (XO: 2.394955116)
   Min dTM/dt [1/us]:-1.65e+03 at an offset of 127.413 usec -> 2021-09-27T11:32:50.394.950.137 (XO: 2.394952541)

 - Block number: 2 starting at 2021-09-27T11:33:09.003.717.725 (XO: 21.003719038 - 21.004025039, t=37.2 degC)  Rank: 00083
   TM values: Max =   190  Min =  -225  Ave =    -4.48  Std =    22.93
   Min TM value:-225 at an offset of 133.313 usec -> 2021-09-27T11:33:09.003.851.037 (XO: 21.003852351)
   Max TM value:190 at an offset of 133.000 usec -> 2021-09-27T11:33:09.003.850.725 (XO: 21.003852038)
   Max dTM/dt [1/us]:1.50e+03 at an offset of 130.950 usec -> 2021-09-27T11:33:09.003.848.675 (XO: 21.003849988)
   Min dTM/dt [1/us]:-1.89e+03 at an offset of 133.113 usec -> 2021-09-27T11:33:09.003.850.837 (XO: 21.003852151)
