-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan 22 01:15:02 2022 UTC
-- File 20210926_062740_1227.dat, 126  packets found.
APID:
      1227 (SL) in 126 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 126 packets (all packets)
Total number of packets: 
      18 in 126 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.877676 to 101.229507
-each separately from 8s to 101s and from 62980us to 877965us.
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.719911 in 1 packets: 18
      18.998991 in 1 packets: 36
      18.715838 in 1 packets: 54
      9.749408 in 1 packets: 72
      8.060234 in 1 packets: 90
      25.105426 in 1 packets: 108
MIU packet time from 2021-09-26T06:27:51.877.676.000 to 2021-09-26T06:29:24.229.507.000
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.719911 in 1 packets: 18
      18.998991 in 1 packets: 36
      18.715838 in 1 packets: 54
      9.749408 in 1 packets: 72
      8.060234 in 1 packets: 90
      25.105426 in 1 packets: 108
MIU orbit number: 
      60000 in 126 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 126 packets (all packets)
MIU orbit information / number of events: 
      0 in 126 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 126 packets (all packets)
Instrument anomalies: 
      0 (OK) in 126 packets (all packets)
FPGA configuration version: 
      211 in 126 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.8582255 to 102.8579149
   Differences in neigboring packets: 
      0.0000000 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999599 in 1 packets: 18
      18.9999365 in 1 packets: 36
      18.9999367 in 1 packets: 54
      9.9999666 in 1 packets: 72
      7.9999732 in 1 packets: 90
      24.9999165 in 1 packets: 108
Temperature FPGA: 
      03 -> 51.1 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 54 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Temperature XO: 
      C1 -> 37.2 degC in 126 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 126 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 126 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 54 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
      2 in 35 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108, 109, ...
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 54 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
      2 in 35 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108, 109, ...
Survey memory error: 
      0 in 126 packets (all packets)
Event memory error: 
      0 in 126 packets (all packets)
Configuration error: 
      0 in 126 packets (all packets)
SDRAM CRC error: 
      0 in 126 packets (all packets)
Radio gain: 
      0 in 126 packets (all packets)
Radio bandwidth: 
      0 in 126 packets (all packets)
Digital gain: 
      0 in 126 packets (all packets)
Analog gain: 
      0 in 126 packets (all packets)
 
SURVEY DATA READER
-- File 20210926_062740_1227.dat, 7 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      58 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.7359024 to 10.7361914
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.7198710  -- OK for SL --
   MIU time from 2021-09-26T06:27:51.877.676.000 to 2021-09-26T06:27:51.877.965.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.719911

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.4560624 to 22.4563514
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.9989270  -- OK for SL --
   MIU time from 2021-09-26T06:28:03.597.876.000 to 2021-09-26T06:28:03.598.165.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.998991

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.4552784 to 41.4555674
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.7157750  -- OK for SL --
   MIU time from 2021-09-26T06:28:22.597.156.000 to 2021-09-26T06:28:22.597.445.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.715838

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      69 in 18 packets (all packets)  -- OK --
   10 MHz counter from 60.1713424 to 60.1716314
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.7493750  -- OK for SL --
   MIU time from 2021-09-26T06:28:41.313.283.000 to 2021-09-26T06:28:41.313.572.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.749408

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 69.9210064 to 69.9212954
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.0602070  -- OK for SL --
   MIU time from 2021-09-26T06:28:51.062.980.000 to 2021-09-26T06:28:51.063.269.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.060234

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 77.9815024 to 77.9817914
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.1053430  -- OK for SL --
   MIU time from 2021-09-26T06:28:59.123.503.000 to 2021-09-26T06:28:59.123.792.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.105426

 - Block number: 7 from the sequence of 18 packets.
   Block numbers in the data:
      7 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      71 in 18 packets (all packets)  -- OK --
   10 MHz counter from 103.0871344 to 103.0874234
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-26T06:29:24.229.218.000 to 2021-09-26T06:29:24.229.507.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 126 points based on 104 XO calibration points at 1Hz counter from 0 to 103
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 10.735902400 corrected by 40bit 10MHz counter of 9.858225500 obtained at 1Hz counter of 8. Result: 8.877679796
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-26T06:27:51.877.535.725
   Max 1st sample MIU time among all SL packets: 2021-09-26T06:29:24.229.366.725
   Min 1st sample XO time among all SL packets: 8.877539521
   Max 1st sample XO time among all SL packets: 101.229368982
   TM values from -391 to 817 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-26T06:27:51.877.535.725 (XO: 8.877539521 - 8.877845522, t=37.2 degC)  Rank: 00058
   TM values: Max =   149  Min =  -203  Ave =    -4.73  Std =    19.69
   Min TM value:-203 at an offset of 112.725 usec -> 2021-09-26T06:27:51.877.648.450 (XO: 8.877652246)
   Max TM value:149 at an offset of 115.250 usec -> 2021-09-26T06:27:51.877.650.975 (XO: 8.877654771)
   Max dTM/dt [1/us]:1.52e+03 at an offset of 115.138 usec -> 2021-09-26T06:27:51.877.650.862 (XO: 8.877654659)
   Min dTM/dt [1/us]:-1.43e+03 at an offset of 72.313 usec -> 2021-09-26T06:27:51.877.608.037 (XO: 8.877611834)

 - Block number: 2 starting at 2021-09-26T06:28:03.597.735.725 (XO: 20.597738698 - 20.598044699, t=37.2 degC)  Rank: 00080
   TM values: Max =   817  Min =  -289  Ave =    -4.47  Std =    25.65
   Min TM value:-289 at an offset of 127.900 usec -> 2021-09-26T06:28:03.597.863.625 (XO: 20.597866598)
   Max TM value:817 at an offset of 127.275 usec -> 2021-09-26T06:28:03.597.863.000 (XO: 20.597865973)
   Max dTM/dt [1/us]:5.95e+03 at an offset of 127.225 usec -> 2021-09-26T06:28:03.597.862.950 (XO: 20.597865923)
   Min dTM/dt [1/us]:-3.23e+03 at an offset of 127.463 usec -> 2021-09-26T06:28:03.597.863.187 (XO: 20.597866160)

 - Block number: 3 starting at 2021-09-26T06:28:22.597.015.725 (XO: 39.597018195 - 39.597324196, t=37.2 degC)  Rank: 00082
   TM values: Max =   191  Min =  -391  Ave =    -4.54  Std =    24.20
   Min TM value:-391 at an offset of 112.725 usec -> 2021-09-26T06:28:22.597.128.450 (XO: 39.597130921)
   Max TM value:191 at an offset of 205.251 usec -> 2021-09-26T06:28:22.597.220.975 (XO: 39.597223446)
   Max dTM/dt [1/us]:1.91e+03 at an offset of 205.213 usec -> 2021-09-26T06:28:22.597.220.938 (XO: 39.597223408)
   Min dTM/dt [1/us]:-2.09e+03 at an offset of 205.288 usec -> 2021-09-26T06:28:22.597.221.013 (XO: 39.597223483)

 - Block number: 4 starting at 2021-09-26T06:28:41.313.142.725 (XO: 58.313144590 - 58.313450591, t=37.2 degC)  Rank: 00069
   TM values: Max =   136  Min =  -165  Ave =    -4.59  Std =    21.73
   Min TM value:-165 at an offset of 111.250 usec -> 2021-09-26T06:28:41.313.253.975 (XO: 58.313255840)
   Max TM value:136 at an offset of 156.913 usec -> 2021-09-26T06:28:41.313.299.638 (XO: 58.313301503)
   Max dTM/dt [1/us]:1.49e+03 at an offset of 156.851 usec -> 2021-09-26T06:28:41.313.299.575 (XO: 58.313301440)
   Min dTM/dt [1/us]:-1.30e+03 at an offset of 135.688 usec -> 2021-09-26T06:28:41.313.278.412 (XO: 58.313280278)

 - Block number: 5 starting at 2021-09-26T06:28:51.062.839.725 (XO: 68.062841139 - 68.063147140, t=37.2 degC)  Rank: 00077
   TM values: Max =   146  Min =  -263  Ave =    -4.51  Std =    23.86
   Min TM value:-263 at an offset of 88.000 usec -> 2021-09-26T06:28:51.062.927.725 (XO: 68.062929139)
   Max TM value:146 at an offset of 90.050 usec -> 2021-09-26T06:28:51.062.929.775 (XO: 68.062931189)
   Max dTM/dt [1/us]:1.39e+03 at an offset of 105.363 usec -> 2021-09-26T06:28:51.062.945.087 (XO: 68.062946502)
   Min dTM/dt [1/us]:-1.57e+03 at an offset of 105.250 usec -> 2021-09-26T06:28:51.062.944.975 (XO: 68.062946389)

 - Block number: 6 starting at 2021-09-26T06:28:59.123.362.725 (XO: 76.123364144 - 76.123670145, t=37.2 degC)  Rank: 00087
   TM values: Max =   156  Min =  -218  Ave =    -4.18  Std =    28.25
   Min TM value:-218 at an offset of 117.725 usec -> 2021-09-26T06:28:59.123.480.450 (XO: 76.123481870)
   Max TM value:156 at an offset of 187.288 usec -> 2021-09-26T06:28:59.123.550.013 (XO: 76.123551433)
   Max dTM/dt [1/us]:1.68e+03 at an offset of 177.926 usec -> 2021-09-26T06:28:59.123.540.650 (XO: 76.123542070)
   Min dTM/dt [1/us]:-1.56e+03 at an offset of 187.326 usec -> 2021-09-26T06:28:59.123.550.050 (XO: 76.123551470)

 - Block number: 7 starting at 2021-09-26T06:29:24.229.077.725 (XO: 101.229079981 - 101.229385982, t=37.2 degC)  Rank: 00071
   TM values: Max =   267  Min =  -350  Ave =    -4.53  Std =    46.89
   Min TM value:-350 at an offset of 111.575 usec -> 2021-09-26T06:29:24.229.189.300 (XO: 101.229191556)
   Max TM value:267 at an offset of 89.138 usec -> 2021-09-26T06:29:24.229.166.862 (XO: 101.229169119)
   Max dTM/dt [1/us]:2.56e+03 at an offset of 167.001 usec -> 2021-09-26T06:29:24.229.244.725 (XO: 101.229246982)
   Min dTM/dt [1/us]:-2.50e+03 at an offset of 174.838 usec -> 2021-09-26T06:29:24.229.252.563 (XO: 101.229254819)
