-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 21 12:23:02 2022 UTC
-- File 20210925_134155_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.620936 to 32.382297
-each separately from 3s to 32s and from 382008us to 731784us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.110270 in 1 packets: 18
      14.650224 in 1 packets: 36
MIU packet time from 2021-09-25T13:42:01.620.936.000 to 2021-09-25T13:42:30.382.297.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.110270 in 1 packets: 18
      14.650224 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.8921163 to 33.8920195
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999532 in 1 packets: 18
      14.9999500 in 1 packets: 36
Temperature FPGA: 
      05 -> 48.5 degC in 54 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210925_134155_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.5130527 to 5.5133417
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.1102230  -- OK for SL --
   MIU time from 2021-09-25T13:42:01.620.936.000 to 2021-09-25T13:42:01.621.225.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.110270

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.6235647 to 19.6238537
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.6501750  -- OK for SL --
   MIU time from 2021-09-25T13:42:15.731.495.000 to 2021-09-25T13:42:15.731.784.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.650224

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.2740287 to 34.2743177
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-25T13:42:30.382.008.000 to 2021-09-25T13:42:30.382.297.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.513052700 corrected by 40bit 10MHz counter of 4.892116300 obtained at 1Hz counter of 3. Result: 3.620938449
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-25T13:42:01.620.795.725
   Max 1st sample MIU time among all SL packets: 2021-09-25T13:42:30.382.156.725
   Min 1st sample XO time among all SL packets: 3.620798174
   Max 1st sample XO time among all SL packets: 32.382159186
   TM values from -282 to 226 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-25T13:42:01.620.795.725 (XO: 3.620798174 - 3.621104175, t=37.2 degC)  Rank: 00080
   TM values: Max =   226  Min =  -282  Ave =    -4.48  Std =    27.42
   Min TM value:-282 at an offset of 183.526 usec -> 2021-09-25T13:42:01.620.979.250 (XO: 3.620981699)
   Max TM value:226 at an offset of 97.325 usec -> 2021-09-25T13:42:01.620.893.050 (XO: 3.620895499)
   Max dTM/dt [1/us]:1.87e+03 at an offset of 97.175 usec -> 2021-09-25T13:42:01.620.892.900 (XO: 3.620895349)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 183.413 usec -> 2021-09-25T13:42:01.620.979.138 (XO: 3.620981587)

 - Block number: 2 starting at 2021-09-25T13:42:15.731.354.725 (XO: 17.731357338 - 17.731663339, t=37.2 degC)  Rank: 00080
   TM values: Max =   218  Min =  -253  Ave =    -4.51  Std =    16.11
   Min TM value:-253 at an offset of 123.100 usec -> 2021-09-25T13:42:15.731.477.825 (XO: 17.731480439)
   Max TM value:218 at an offset of 120.800 usec -> 2021-09-25T13:42:15.731.475.525 (XO: 17.731478139)
   Max dTM/dt [1/us]:2.05e+03 at an offset of 120.675 usec -> 2021-09-25T13:42:15.731.475.400 (XO: 17.731478014)
   Min dTM/dt [1/us]:-2.00e+03 at an offset of 122.938 usec -> 2021-09-25T13:42:15.731.477.662 (XO: 17.731480276)

 - Block number: 3 starting at 2021-09-25T13:42:30.381.867.725 (XO: 32.381870185 - 32.382176186, t=37.2 degC)  Rank: 00079
   TM values: Max =   203  Min =  -245  Ave =    -4.56  Std =    22.49
   Min TM value:-245 at an offset of 119.138 usec -> 2021-09-25T13:42:30.381.986.862 (XO: 32.381989323)
   Max TM value:203 at an offset of 121.750 usec -> 2021-09-25T13:42:30.381.989.475 (XO: 32.381991936)
   Max dTM/dt [1/us]:1.74e+03 at an offset of 121.600 usec -> 2021-09-25T13:42:30.381.989.325 (XO: 32.381991786)
   Min dTM/dt [1/us]:-1.91e+03 at an offset of 118.950 usec -> 2021-09-25T13:42:30.381.986.675 (XO: 32.381989136)
