-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 21 10:37:47 2022 UTC
-- File 20210925_124708_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.398746 to 31.658209
-each separately from 10s to 31s and from 398746us to 998401us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.599077 in 1 packets: 18
      4.659519 in 1 packets: 36
MIU packet time from 2021-09-25T12:47:21.398.746.000 to 2021-09-25T12:47:42.658.209.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.599077 in 1 packets: 18
      4.659519 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.8545866 to 32.8545164
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999464 in 1 packets: 18
      4.9999834 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20210925_124708_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.2533331 to 12.2536221
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.5990230  -- OK for SL --
   MIU time from 2021-09-25T12:47:21.398.746.000 to 2021-09-25T12:47:21.399.035.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.599077

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      93 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.8526451 to 28.8529341
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.6595030  -- OK for SL --
   MIU time from 2021-09-25T12:47:37.998.112.000 to 2021-09-25T12:47:37.998.401.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.659519

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.5124371 to 33.5127261
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-25T12:47:42.657.920.000 to 2021-09-25T12:47:42.658.209.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.253333100 corrected by 40bit 10MHz counter of 11.854586600 obtained at 1Hz counter of 10. Result: 10.398747816
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-25T12:47:21.398.605.725
   Max 1st sample MIU time among all SL packets: 2021-09-25T12:47:42.658.068.725
   Min 1st sample XO time among all SL packets: 10.398607540
   Max 1st sample XO time among all SL packets: 31.658071662
   TM values from -252 to 203 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-25T12:47:21.398.605.725 (XO: 10.398607540 - 10.398913541, t=37.2 degC)  Rank: 00078
   TM values: Max =   193  Min =  -252  Ave =    -4.58  Std =    24.15
   Min TM value:-252 at an offset of 118.313 usec -> 2021-09-25T12:47:21.398.724.037 (XO: 10.398725853)
   Max TM value:193 at an offset of 120.850 usec -> 2021-09-25T12:47:21.398.726.575 (XO: 10.398728391)
   Max dTM/dt [1/us]:1.81e+03 at an offset of 120.713 usec -> 2021-09-25T12:47:21.398.726.437 (XO: 10.398728253)
   Min dTM/dt [1/us]:-2.04e+03 at an offset of 118.113 usec -> 2021-09-25T12:47:21.398.723.837 (XO: 10.398725653)

 - Block number: 2 starting at 2021-09-25T12:47:37.997.971.725 (XO: 26.997975218 - 26.998281219, t=37.2 degC)  Rank: 00093
   TM values: Max =   195  Min =  -252  Ave =    -4.58  Std =    24.56
   Min TM value:-252 at an offset of 117.575 usec -> 2021-09-25T12:47:37.998.089.300 (XO: 26.998092794)
   Max TM value:195 at an offset of 120.188 usec -> 2021-09-25T12:47:37.998.091.912 (XO: 26.998095406)
   Max dTM/dt [1/us]:1.83e+03 at an offset of 120.000 usec -> 2021-09-25T12:47:37.998.091.725 (XO: 26.998095219)
   Min dTM/dt [1/us]:-2.16e+03 at an offset of 117.425 usec -> 2021-09-25T12:47:37.998.089.150 (XO: 26.998092644)

 - Block number: 3 starting at 2021-09-25T12:47:42.657.779.725 (XO: 31.657782661 - 31.658088663, t=37.2 degC)  Rank: 00080
   TM values: Max =   203  Min =  -233  Ave =    -4.43  Std =    22.97
   Min TM value:-233 at an offset of 90.650 usec -> 2021-09-25T12:47:42.657.870.375 (XO: 31.657873312)
   Max TM value:203 at an offset of 90.400 usec -> 2021-09-25T12:47:42.657.870.125 (XO: 31.657873062)
   Max dTM/dt [1/us]:1.94e+03 at an offset of 93.075 usec -> 2021-09-25T12:47:42.657.872.800 (XO: 31.657875737)
   Min dTM/dt [1/us]:-2.21e+03 at an offset of 90.513 usec -> 2021-09-25T12:47:42.657.870.237 (XO: 31.657873174)
