-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 21 07:45:30 2022 UTC
-- File 20210925_102235_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.141854 to 29.061304
-each separately from 2s to 29s and from 61015us to 142143us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      26.918872 in 1 packets: 18
MIU packet time from 2021-09-25T10:22:40.141.854.000 to 2021-09-25T10:23:07.061.304.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      26.918872 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.8685057 to 30.8684160
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      26.9999103 in 1 packets: 18
Temperature FPGA: 
      03 -> 51.1 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210925_102235_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.0103605 to 4.0106495
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 26.9187830  -- OK for SL --
   MIU time from 2021-09-25T10:22:40.141.854.000 to 2021-09-25T10:22:40.142.143.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 26.918872

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.9294325 to 30.9297215
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-25T10:23:07.061.015.000 to 2021-09-25T10:23:07.061.304.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.010360500 corrected by 40bit 10MHz counter of 3.868505700 obtained at 1Hz counter of 2. Result: 2.141855268
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-25T10:22:40.141.713.725
   Max 1st sample MIU time among all SL packets: 2021-09-25T10:23:07.061.163.725
   Min 1st sample XO time among all SL packets: 2.141714993
   Max 1st sample XO time among all SL packets: 29.061165427
   TM values from -227 to 186 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-25T10:22:40.141.713.725 (XO: 2.141714993 - 2.142020994, t=37.2 degC)  Rank: 00080
   TM values: Max =   186  Min =  -227  Ave =    -4.58  Std =    23.70
   Min TM value:-227 at an offset of 124.625 usec -> 2021-09-25T10:22:40.141.838.350 (XO: 2.141839618)
   Max TM value:186 at an offset of 124.288 usec -> 2021-09-25T10:22:40.141.838.012 (XO: 2.141839281)
   Max dTM/dt [1/us]:1.67e+03 at an offset of 126.950 usec -> 2021-09-25T10:22:40.141.840.675 (XO: 2.141841943)
   Min dTM/dt [1/us]:-1.92e+03 at an offset of 124.388 usec -> 2021-09-25T10:22:40.141.838.112 (XO: 2.141839381)

 - Block number: 2 starting at 2021-09-25T10:23:07.060.874.725 (XO: 29.060876426 - 29.061182427, t=37.2 degC)  Rank: 00079
   TM values: Max =   172  Min =  -215  Ave =    -4.54  Std =    19.87
   Min TM value:-215 at an offset of 125.525 usec -> 2021-09-25T10:23:07.061.000.250 (XO: 29.061001951)
   Max TM value:172 at an offset of 128.063 usec -> 2021-09-25T10:23:07.061.002.787 (XO: 29.061004489)
   Max dTM/dt [1/us]:1.75e+03 at an offset of 127.875 usec -> 2021-09-25T10:23:07.061.002.600 (XO: 29.061004301)
   Min dTM/dt [1/us]:-1.75e+03 at an offset of 125.275 usec -> 2021-09-25T10:23:07.061.000.000 (XO: 29.061001701)
