-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 21 05:00:18 2022 UTC
-- File 20210924_162956_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.972884 to 71.772863
-each separately from 6s to 71s and from 32724us to 973173us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.059551 in 1 packets: 18
      20.339539 in 1 packets: 36
      17.020136 in 1 packets: 54
      18.379308 in 1 packets: 72
MIU packet time from 2021-09-24T16:30:05.972.884.000 to 2021-09-24T16:31:10.772.863.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.059551 in 1 packets: 18
      20.339539 in 1 packets: 36
      17.020136 in 1 packets: 54
      18.379308 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.8818683 to 72.8816503
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999665 in 1 packets: 18
      19.9999329 in 1 packets: 36
      16.9999430 in 1 packets: 54
      17.9999396 in 1 packets: 72
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
      06 -> 47.2 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 54 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 54 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20210924_162956_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.8547524 to 8.8550414
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.0595190  -- OK for SL --
   MIU time from 2021-09-24T16:30:05.972.884.000 to 2021-09-24T16:30:05.973.173.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.059551

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.9145604 to 17.9148494
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.3394710  -- OK for SL --
   MIU time from 2021-09-24T16:30:15.032.724.000 to 2021-09-24T16:30:15.033.013.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.339539

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 38.2543204 to 38.2546094
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.0200790  -- OK for SL --
   MIU time from 2021-09-24T16:30:35.372.552.000 to 2021-09-24T16:30:35.372.841.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.020136

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 55.2746884 to 55.2749774
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.3792470  -- OK for SL --
   MIU time from 2021-09-24T16:30:52.392.977.000 to 2021-09-24T16:30:52.393.266.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.379308

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 73.6542244 to 73.6545134
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-24T16:31:10.772.574.000 to 2021-09-24T16:31:10.772.863.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.854752400 corrected by 40bit 10MHz counter of 7.881868300 obtained at 1Hz counter of 6. Result: 6.972887311
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-24T16:30:05.972.743.725
   Max 1st sample MIU time among all SL packets: 2021-09-24T16:31:10.772.722.725
   Min 1st sample XO time among all SL packets: 6.972747035
   Max 1st sample XO time among all SL packets: 71.772725452
   TM values from -280 to 235 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-24T16:30:05.972.743.725 (XO: 6.972747035 - 6.973053036, t=37.2 degC)  Rank: 00081
   TM values: Max =   211  Min =  -258  Ave =    -4.50  Std =    25.98
   Min TM value:-258 at an offset of 120.550 usec -> 2021-09-24T16:30:05.972.864.275 (XO: 6.972867585)
   Max TM value:211 at an offset of 123.200 usec -> 2021-09-24T16:30:05.972.866.925 (XO: 6.972870235)
   Max dTM/dt [1/us]:1.52e+03 at an offset of 123.063 usec -> 2021-09-24T16:30:05.972.866.787 (XO: 6.972870098)
   Min dTM/dt [1/us]:-2.06e+03 at an offset of 120.388 usec -> 2021-09-24T16:30:05.972.864.112 (XO: 6.972867423)

 - Block number: 2 starting at 2021-09-24T16:30:15.032.583.725 (XO: 16.032585436 - 16.032891437, t=37.2 degC)  Rank: 00080
   TM values: Max =   204  Min =  -262  Ave =    -4.57  Std =    20.94
   Min TM value:-262 at an offset of 121.413 usec -> 2021-09-24T16:30:15.032.705.137 (XO: 16.032706849)
   Max TM value:204 at an offset of 119.113 usec -> 2021-09-24T16:30:15.032.702.837 (XO: 16.032704549)
   Max dTM/dt [1/us]:2.07e+03 at an offset of 123.813 usec -> 2021-09-24T16:30:15.032.707.537 (XO: 16.032709249)
   Min dTM/dt [1/us]:-2.18e+03 at an offset of 121.213 usec -> 2021-09-24T16:30:15.032.704.937 (XO: 16.032706649)

 - Block number: 3 starting at 2021-09-24T16:30:35.372.411.725 (XO: 36.372413691 - 36.372719692, t=37.2 degC)  Rank: 00077
   TM values: Max =   216  Min =  -262  Ave =    -4.47  Std =    22.86
   Min TM value:-262 at an offset of 119.088 usec -> 2021-09-24T16:30:35.372.530.812 (XO: 36.372532779)
   Max TM value:216 at an offset of 121.700 usec -> 2021-09-24T16:30:35.372.533.425 (XO: 36.372535392)
   Max dTM/dt [1/us]:1.86e+03 at an offset of 121.488 usec -> 2021-09-24T16:30:35.372.533.212 (XO: 36.372535179)
   Min dTM/dt [1/us]:-1.90e+03 at an offset of 118.900 usec -> 2021-09-24T16:30:35.372.530.625 (XO: 36.372532592)

 - Block number: 4 starting at 2021-09-24T16:30:52.392.836.725 (XO: 53.392838761 - 53.393144762, t=37.2 degC)  Rank: 00084
   TM values: Max =   235  Min =  -280  Ave =    -4.49  Std =    33.83
   Min TM value:-280 at an offset of 181.063 usec -> 2021-09-24T16:30:52.393.017.788 (XO: 53.393019824)
   Max TM value:235 at an offset of 113.688 usec -> 2021-09-24T16:30:52.392.950.412 (XO: 53.392952449)
   Max dTM/dt [1/us]:2.19e+03 at an offset of 113.575 usec -> 2021-09-24T16:30:52.392.950.300 (XO: 53.392952336)
   Min dTM/dt [1/us]:-2.67e+03 at an offset of 111.038 usec -> 2021-09-24T16:30:52.392.947.762 (XO: 53.392949799)

 - Block number: 5 starting at 2021-09-24T16:31:10.772.433.725 (XO: 71.772436451 - 71.772742452, t=37.2 degC)  Rank: 00078
   TM values: Max =   213  Min =  -241  Ave =    -4.48  Std =    20.97
   Min TM value:-241 at an offset of 121.850 usec -> 2021-09-24T16:31:10.772.555.575 (XO: 71.772558302)
   Max TM value:213 at an offset of 124.488 usec -> 2021-09-24T16:31:10.772.558.212 (XO: 71.772560939)
   Max dTM/dt [1/us]:1.86e+03 at an offset of 124.288 usec -> 2021-09-24T16:31:10.772.558.012 (XO: 71.772560739)
   Min dTM/dt [1/us]:-2.25e+03 at an offset of 121.675 usec -> 2021-09-24T16:31:10.772.555.400 (XO: 71.772558127)
