-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Jan 20 15:29:22 2022 UTC
-- File 20210924_100056_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.374631 to 53.193678
-each separately from 8s to 53s and from 20372us to 878429us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.503220 in 1 packets: 18
      11.141943 in 1 packets: 36
      17.172728 in 1 packets: 54
MIU packet time from 2021-09-24T10:01:07.374.631.000 to 2021-09-24T10:01:52.193.678.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.503220 in 1 packets: 18
      11.141943 in 1 packets: 36
      17.172728 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.9085537 to 54.9084038
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999468 in 1 packets: 18
      11.9999599 in 1 packets: 36
      16.9999434 in 1 packets: 54
Temperature FPGA: 
      04 -> 49.8 degC in 72 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210924_100056_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.2831854 to 10.2834744
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.5031670  -- OK for SL --
   MIU time from 2021-09-24T10:01:07.374.631.000 to 2021-09-24T10:01:07.374.920.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.503220

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.7866414 to 26.7869304
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.1419030  -- OK for SL --
   MIU time from 2021-09-24T10:01:23.878.140.000 to 2021-09-24T10:01:23.878.429.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.141943

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.9288334 to 37.9291224
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.1726710  -- OK for SL --
   MIU time from 2021-09-24T10:01:35.020.372.000 to 2021-09-24T10:01:35.020.661.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.172728

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 55.1017934 to 55.1020824
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-24T10:01:52.193.389.000 to 2021-09-24T10:01:52.193.678.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 10.283185400 corrected by 40bit 10MHz counter of 9.908553700 obtained at 1Hz counter of 8. Result: 8.374632936
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-24T10:01:07.374.490.725
   Max 1st sample MIU time among all SL packets: 2021-09-24T10:01:52.193.537.725
   Min 1st sample XO time among all SL packets: 8.374492661
   Max 1st sample XO time among all SL packets: 53.193538964
   TM values from -246 to 199 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-24T10:01:07.374.490.725 (XO: 8.374492661 - 8.374798662, t=37.2 degC)  Rank: 00078
   TM values: Max =   180  Min =  -209  Ave =    -4.49  Std =    17.33
   Min TM value:-209 at an offset of 118.225 usec -> 2021-09-24T10:01:07.374.608.950 (XO: 8.374610886)
   Max TM value:180 at an offset of 120.888 usec -> 2021-09-24T10:01:07.374.611.612 (XO: 8.374613549)
   Max dTM/dt [1/us]:1.39e+03 at an offset of 120.725 usec -> 2021-09-24T10:01:07.374.611.450 (XO: 8.374613386)
   Min dTM/dt [1/us]:-1.50e+03 at an offset of 117.975 usec -> 2021-09-24T10:01:07.374.608.700 (XO: 8.374610636)

 - Block number: 2 starting at 2021-09-24T10:01:23.877.999.725 (XO: 24.878003610 - 24.878309611, t=37.2 degC)  Rank: 00083
   TM values: Max =   199  Min =  -246  Ave =    -4.58  Std =    31.30
   Min TM value:-246 at an offset of 108.400 usec -> 2021-09-24T10:01:23.878.108.125 (XO: 24.878112011)
   Max TM value:199 at an offset of 108.125 usec -> 2021-09-24T10:01:23.878.107.850 (XO: 24.878111736)
   Max dTM/dt [1/us]:1.87e+03 at an offset of 110.775 usec -> 2021-09-24T10:01:23.878.110.500 (XO: 24.878114386)
   Min dTM/dt [1/us]:-2.31e+03 at an offset of 108.238 usec -> 2021-09-24T10:01:23.878.107.962 (XO: 24.878111848)

 - Block number: 3 starting at 2021-09-24T10:01:35.020.231.725 (XO: 36.020232792 - 36.020538793, t=37.2 degC)  Rank: 00082
   TM values: Max =   175  Min =  -217  Ave =    -4.51  Std =    17.43
   Min TM value:-217 at an offset of 105.500 usec -> 2021-09-24T10:01:35.020.337.225 (XO: 36.020338292)
   Max TM value:175 at an offset of 108.100 usec -> 2021-09-24T10:01:35.020.339.825 (XO: 36.020340892)
   Max dTM/dt [1/us]:1.43e+03 at an offset of 107.938 usec -> 2021-09-24T10:01:35.020.339.662 (XO: 36.020340730)
   Min dTM/dt [1/us]:-1.62e+03 at an offset of 105.275 usec -> 2021-09-24T10:01:35.020.337.000 (XO: 36.020338067)

 - Block number: 4 starting at 2021-09-24T10:01:52.193.248.725 (XO: 53.193249963 - 53.193555964, t=37.2 degC)  Rank: 00074
   TM values: Max =   185  Min =  -227  Ave =    -4.50  Std =    18.76
   Min TM value:-227 at an offset of 93.575 usec -> 2021-09-24T10:01:52.193.342.300 (XO: 53.193343538)
   Max TM value:185 at an offset of 96.175 usec -> 2021-09-24T10:01:52.193.344.900 (XO: 53.193346138)
   Max dTM/dt [1/us]:1.42e+03 at an offset of 96.000 usec -> 2021-09-24T10:01:52.193.344.725 (XO: 53.193345963)
   Min dTM/dt [1/us]:-1.70e+03 at an offset of 93.388 usec -> 2021-09-24T10:01:52.193.342.112 (XO: 53.193343351)
