-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Jan 18 16:35:20 2022 UTC
-- File 20210922_095934_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.472917 to 55.061940
-each separately from 2s to 55s and from 61651us to 868145us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.394650 in 1 packets: 18
      16.676312 in 1 packets: 36
      12.516905 in 1 packets: 54
MIU packet time from 2021-09-22T09:59:39.472.917.000 to 2021-09-22T10:00:32.061.940.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.394650 in 1 packets: 18
      16.676312 in 1 packets: 36
      12.516905 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.9119696 to 56.9117938
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.9999239 in 1 packets: 18
      16.9999435 in 1 packets: 36
      12.9999568 in 1 packets: 54
Temperature FPGA: 
      05 -> 48.5 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      04 -> 49.8 degC in 18 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210922_095934_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.3848863 to 4.3851753
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.3945750  -- OK for SL --
   MIU time from 2021-09-22T09:59:39.472.917.000 to 2021-09-22T09:59:39.473.206.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.394650

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.7797503 to 27.7800393
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.6762550  -- OK for SL --
   MIU time from 2021-09-22T10:00:02.867.856.000 to 2021-09-22T10:00:02.868.145.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.676312

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 44.4562943 to 44.4565833
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.5168630  -- OK for SL --
   MIU time from 2021-09-22T10:00:19.544.457.000 to 2021-09-22T10:00:19.544.746.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.516905

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 56.9734463 to 56.9737353
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-22T10:00:32.061.651.000 to 2021-09-22T10:00:32.061.940.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.384886300 corrected by 40bit 10MHz counter of 3.911969600 obtained at 1Hz counter of 2. Result: 2.472918261
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-22T09:59:39.472.776.725
   Max 1st sample MIU time among all SL packets: 2021-09-22T10:00:32.061.799.725
   Min 1st sample XO time among all SL packets: 2.472777985
   Max 1st sample XO time among all SL packets: 55.061801429
   TM values from -244 to 204 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-22T09:59:39.472.776.725 (XO: 2.472777985 - 2.473083986, t=37.2 degC)  Rank: 00080
   TM values: Max =   184  Min =  -244  Ave =    -4.54  Std =    22.76
   Min TM value:-244 at an offset of 88.413 usec -> 2021-09-22T09:59:39.472.865.137 (XO: 2.472866398)
   Max TM value:184 at an offset of 119.850 usec -> 2021-09-22T09:59:39.472.896.575 (XO: 2.472897836)
   Max dTM/dt [1/us]:1.56e+03 at an offset of 85.963 usec -> 2021-09-22T09:59:39.472.862.687 (XO: 2.472863948)
   Min dTM/dt [1/us]:-1.64e+03 at an offset of 116.938 usec -> 2021-09-22T09:59:39.472.893.662 (XO: 2.472894923)

 - Block number: 2 starting at 2021-09-22T10:00:02.867.715.725 (XO: 25.867719388 - 25.868025389, t=37.2 degC)  Rank: 00082
   TM values: Max =   204  Min =  -237  Ave =    -4.55  Std =    17.65
   Min TM value:-237 at an offset of 119.650 usec -> 2021-09-22T10:00:02.867.835.375 (XO: 25.867839039)
   Max TM value:204 at an offset of 117.363 usec -> 2021-09-22T10:00:02.867.833.087 (XO: 25.867836751)
   Max dTM/dt [1/us]:1.78e+03 at an offset of 117.200 usec -> 2021-09-22T10:00:02.867.832.925 (XO: 25.867836589)
   Min dTM/dt [1/us]:-2.02e+03 at an offset of 119.413 usec -> 2021-09-22T10:00:02.867.835.137 (XO: 25.867838801)

 - Block number: 3 starting at 2021-09-22T10:00:19.544.316.725 (XO: 42.544318876 - 42.544624877, t=37.2 degC)  Rank: 00077
   TM values: Max =   182  Min =  -235  Ave =    -4.61  Std =    18.59
   Min TM value:-235 at an offset of 108.225 usec -> 2021-09-22T10:00:19.544.424.950 (XO: 42.544427101)
   Max TM value:182 at an offset of 107.925 usec -> 2021-09-22T10:00:19.544.424.650 (XO: 42.544426801)
   Max dTM/dt [1/us]:1.90e+03 at an offset of 110.613 usec -> 2021-09-22T10:00:19.544.427.337 (XO: 42.544429489)
   Min dTM/dt [1/us]:-2.01e+03 at an offset of 108.050 usec -> 2021-09-22T10:00:19.544.424.775 (XO: 42.544426926)

 - Block number: 4 starting at 2021-09-22T10:00:32.061.510.725 (XO: 55.061512428 - 55.061818429, t=37.2 degC)  Rank: 00086
   TM values: Max =   188  Min =  -235  Ave =    -4.65  Std =    19.95
   Min TM value:-235 at an offset of 116.575 usec -> 2021-09-22T10:00:32.061.627.300 (XO: 55.061629003)
   Max TM value:188 at an offset of 119.100 usec -> 2021-09-22T10:00:32.061.629.825 (XO: 55.061631528)
   Max dTM/dt [1/us]:1.77e+03 at an offset of 118.938 usec -> 2021-09-22T10:00:32.061.629.662 (XO: 55.061631366)
   Min dTM/dt [1/us]:-2.06e+03 at an offset of 116.363 usec -> 2021-09-22T10:00:32.061.627.087 (XO: 55.061628791)
