-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Jan 18 14:27:09 2022 UTC
-- File 20210922_084723_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.674610 to 55.627048
-each separately from 10s to 55s and from 42755us to 829872us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.367856 in 1 packets: 18
      18.786539 in 1 packets: 36
      16.796887 in 1 packets: 54
MIU packet time from 2021-09-22T08:47:36.674.610.000 to 2021-09-22T08:48:21.627.048.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.367856 in 1 packets: 18
      18.786539 in 1 packets: 36
      16.796887 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.6223807 to 56.6222319
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999668 in 1 packets: 18
      17.9999405 in 1 packets: 36
      16.9999439 in 1 packets: 54
Temperature FPGA: 
      04 -> 49.8 degC in 72 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210922_084723_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.2969917 to 12.2972807
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.3678230  -- OK for SL --
   MIU time from 2021-09-22T08:47:36.674.610.000 to 2021-09-22T08:47:36.674.899.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.367856

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.6651037 to 21.6653927
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.7864790  -- OK for SL --
   MIU time from 2021-09-22T08:47:46.042.755.000 to 2021-09-22T08:47:46.043.044.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.786539

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 40.4518717 to 40.4521607
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.7968310  -- OK for SL --
   MIU time from 2021-09-22T08:48:04.829.583.000 to 2021-09-22T08:48:04.829.872.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.796887

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 57.2489917 to 57.2492807
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-22T08:48:21.626.759.000 to 2021-09-22T08:48:21.627.048.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.296991700 corrected by 40bit 10MHz counter of 11.622380700 obtained at 1Hz counter of 10. Result: 10.674613226
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-22T08:47:36.674.469.725
   Max 1st sample MIU time among all SL packets: 2021-09-22T08:48:21.626.907.725
   Min 1st sample XO time among all SL packets: 10.674472951
   Max 1st sample XO time among all SL packets: 55.626910594
   TM values from -245 to 182 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-22T08:47:36.674.469.725 (XO: 10.674472951 - 10.674778952, t=37.2 degC)  Rank: 00077
   TM values: Max =   182  Min =  -245  Ave =    -4.66  Std =    21.73
   Min TM value:-245 at an offset of 115.525 usec -> 2021-09-22T08:47:36.674.585.250 (XO: 10.674588476)
   Max TM value:182 at an offset of 115.213 usec -> 2021-09-22T08:47:36.674.584.937 (XO: 10.674588164)
   Max dTM/dt [1/us]:1.85e+03 at an offset of 117.888 usec -> 2021-09-22T08:47:36.674.587.612 (XO: 10.674590839)
   Min dTM/dt [1/us]:-1.63e+03 at an offset of 115.313 usec -> 2021-09-22T08:47:36.674.585.037 (XO: 10.674588264)

 - Block number: 2 starting at 2021-09-22T08:47:46.042.614.725 (XO: 20.042616066 - 20.042922067, t=37.2 degC)  Rank: 00075
   TM values: Max =   170  Min =  -231  Ave =    -4.70  Std =    25.61
   Min TM value:-231 at an offset of 204.638 usec -> 2021-09-22T08:47:46.042.819.363 (XO: 20.042820704)
   Max TM value:170 at an offset of 163.888 usec -> 2021-09-22T08:47:46.042.778.613 (XO: 20.042779954)
   Max dTM/dt [1/us]:1.46e+03 at an offset of 207.101 usec -> 2021-09-22T08:47:46.042.821.825 (XO: 20.042823166)
   Min dTM/dt [1/us]:-1.61e+03 at an offset of 161.113 usec -> 2021-09-22T08:47:46.042.775.838 (XO: 20.042777179)

 - Block number: 3 starting at 2021-09-22T08:48:04.829.442.725 (XO: 38.829446162 - 38.829752163, t=37.2 degC)  Rank: 00081
   TM values: Max =   175  Min =  -236  Ave =    -4.65  Std =    16.78
   Min TM value:-236 at an offset of 118.250 usec -> 2021-09-22T08:48:04.829.560.975 (XO: 38.829564413)
   Max TM value:175 at an offset of 120.825 usec -> 2021-09-22T08:48:04.829.563.550 (XO: 38.829566988)
   Max dTM/dt [1/us]:1.64e+03 at an offset of 120.650 usec -> 2021-09-22T08:48:04.829.563.375 (XO: 38.829566813)
   Min dTM/dt [1/us]:-1.48e+03 at an offset of 118.050 usec -> 2021-09-22T08:48:04.829.560.775 (XO: 38.829564213)

 - Block number: 4 starting at 2021-09-22T08:48:21.626.618.725 (XO: 55.626621593 - 55.626927594, t=37.2 degC)  Rank: 00087
   TM values: Max =   176  Min =  -221  Ave =    -4.60  Std =    19.04
   Min TM value:-221 at an offset of 128.125 usec -> 2021-09-22T08:48:21.626.746.850 (XO: 55.626749718)
   Max TM value:176 at an offset of 130.700 usec -> 2021-09-22T08:48:21.626.749.425 (XO: 55.626752293)
   Max dTM/dt [1/us]:1.36e+03 at an offset of 130.538 usec -> 2021-09-22T08:48:21.626.749.262 (XO: 55.626752131)
   Min dTM/dt [1/us]:-1.72e+03 at an offset of 127.888 usec -> 2021-09-22T08:48:21.626.746.612 (XO: 55.626749481)
