-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan 15 11:58:51 2022 UTC
-- File 20210919_205041_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.745326 to 52.204011
-each separately from 5s to 52s and from 54596us to 745615us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.559568 in 1 packets: 18
      20.749124 in 1 packets: 36
      16.148837 in 1 packets: 54
MIU packet time from 2021-09-19T20:50:49.745.326.000 to 2021-09-19T20:51:36.204.011.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.559568 in 1 packets: 18
      20.749124 in 1 packets: 36
      16.148837 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.7345387 to 53.7343823
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999668 in 1 packets: 18
      20.9999300 in 1 packets: 36
      15.9999468 in 1 packets: 54
Temperature FPGA: 
      03 -> 51.1 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210919_205041_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.4798650 to 7.4801540
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.5595350  -- OK for SL --
   MIU time from 2021-09-19T20:50:49.745.326.000 to 2021-09-19T20:50:49.745.615.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.559568

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      73 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.0396890 to 17.0399780
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.7490550  -- OK for SL --
   MIU time from 2021-09-19T20:50:59.305.183.000 to 2021-09-19T20:50:59.305.472.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.749124

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.7890330 to 37.7893220
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.1487830  -- OK for SL --
   MIU time from 2021-09-19T20:51:20.054.596.000 to 2021-09-19T20:51:20.054.885.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.148837

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 53.9381050 to 53.9383940
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-19T20:51:36.203.722.000 to 2021-09-19T20:51:36.204.011.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.479865000 corrected by 40bit 10MHz counter of 6.734538700 obtained at 1Hz counter of 5. Result: 5.745328834
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-19T20:50:49.745.185.725
   Max 1st sample MIU time among all SL packets: 2021-09-19T20:51:36.203.870.725
   Min 1st sample XO time among all SL packets: 5.745188559
   Max 1st sample XO time among all SL packets: 52.203872098
   TM values from -285 to 218 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-19T20:50:49.745.185.725 (XO: 5.745188559 - 5.745494560, t=37.2 degC)  Rank: 00078
   TM values: Max =   196  Min =  -276  Ave =    -4.69  Std =    18.56
   Min TM value:-276 at an offset of 120.600 usec -> 2021-09-19T20:50:49.745.306.325 (XO: 5.745309159)
   Max TM value:196 at an offset of 120.288 usec -> 2021-09-19T20:50:49.745.306.012 (XO: 5.745308847)
   Max dTM/dt [1/us]:1.54e+03 at an offset of 118.100 usec -> 2021-09-19T20:50:49.745.303.825 (XO: 5.745306659)
   Min dTM/dt [1/us]:-2.06e+03 at an offset of 120.400 usec -> 2021-09-19T20:50:49.745.306.125 (XO: 5.745308959)

 - Block number: 2 starting at 2021-09-19T20:50:59.305.042.725 (XO: 15.305044262 - 15.305350263, t=37.2 degC)  Rank: 00073
   TM values: Max =   204  Min =  -281  Ave =    -4.56  Std =    22.41
   Min TM value:-281 at an offset of 109.550 usec -> 2021-09-19T20:50:59.305.152.275 (XO: 15.305153813)
   Max TM value:204 at an offset of 112.075 usec -> 2021-09-19T20:50:59.305.154.800 (XO: 15.305156338)
   Max dTM/dt [1/us]:1.94e+03 at an offset of 111.913 usec -> 2021-09-19T20:50:59.305.154.637 (XO: 15.305156175)
   Min dTM/dt [1/us]:-2.18e+03 at an offset of 109.350 usec -> 2021-09-19T20:50:59.305.152.075 (XO: 15.305153613)

 - Block number: 3 starting at 2021-09-19T20:51:20.054.455.725 (XO: 36.054457405 - 36.054763406, t=37.2 degC)  Rank: 00077
   TM values: Max =   184  Min =  -206  Ave =    -4.64  Std =    23.88
   Min TM value:-206 at an offset of 106.050 usec -> 2021-09-19T20:51:20.054.561.775 (XO: 36.054563455)
   Max TM value:184 at an offset of 121.513 usec -> 2021-09-19T20:51:20.054.577.237 (XO: 36.054578918)
   Max dTM/dt [1/us]:1.98e+03 at an offset of 121.375 usec -> 2021-09-19T20:51:20.054.577.100 (XO: 36.054578780)
   Min dTM/dt [1/us]:-1.66e+03 at an offset of 93.088 usec -> 2021-09-19T20:51:20.054.548.812 (XO: 36.054550493)

 - Block number: 4 starting at 2021-09-19T20:51:36.203.581.725 (XO: 52.203583097 - 52.203889098, t=37.2 degC)  Rank: 00082
   TM values: Max =   218  Min =  -285  Ave =    -4.56  Std =    30.64
   Min TM value:-285 at an offset of 121.150 usec -> 2021-09-19T20:51:36.203.702.875 (XO: 52.203704247)
   Max TM value:218 at an offset of 120.863 usec -> 2021-09-19T20:51:36.203.702.587 (XO: 52.203703960)
   Max dTM/dt [1/us]:2.12e+03 at an offset of 123.550 usec -> 2021-09-19T20:51:36.203.705.275 (XO: 52.203706647)
   Min dTM/dt [1/us]:-3.04e+03 at an offset of 121.025 usec -> 2021-09-19T20:51:36.203.702.750 (XO: 52.203704122)
