-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jan 15 08:33:21 2022 UTC
-- File 20210919_155408_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.424833 to 24.485551
-each separately from 1s to 24s and from 424833us to 485551us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.060140 in 1 packets: 18
MIU packet time from 2021-09-19T15:54:12.424.833.000 to 2021-09-19T15:54:35.485.551.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      23.060140 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.9096529 to 25.9095765
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.9999236 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 1 packets: 19
      3 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 1 packets: 19
      3 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210919_155408_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.3344869 to 3.3347759
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.0600630  -- OK for SL --
   MIU time from 2021-09-19T15:54:12.424.833.000 to 2021-09-19T15:54:12.425.122.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.060140

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.3948389 to 26.3951279
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-19T15:54:35.485.262.000 to 2021-09-19T15:54:35.485.551.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.334486900 corrected by 40bit 10MHz counter of 2.909652900 obtained at 1Hz counter of 1. Result: 1.424835402
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-19T15:54:12.424.692.725
   Max 1st sample MIU time among all SL packets: 2021-09-19T15:54:35.485.410.725
   Min 1st sample XO time among all SL packets: 1.424695126
   Max 1st sample XO time among all SL packets: 24.485412727
   TM values from -277 to 251 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-19T15:54:12.424.692.725 (XO: 1.424695126 - 1.425001128, t=37.2 degC)  Rank: 00084
   TM values: Max =   222  Min =  -255  Ave =    -4.58  Std =    30.02
   Min TM value:-255 at an offset of 120.625 usec -> 2021-09-19T15:54:12.424.813.350 (XO: 1.424815752)
   Max TM value:222 at an offset of 123.288 usec -> 2021-09-19T15:54:12.424.816.012 (XO: 1.424818414)
   Max dTM/dt [1/us]:1.98e+03 at an offset of 123.150 usec -> 2021-09-19T15:54:12.424.815.875 (XO: 1.424818277)
   Min dTM/dt [1/us]:-1.84e+03 at an offset of 108.463 usec -> 2021-09-19T15:54:12.424.801.187 (XO: 1.424803589)

 - Block number: 2 starting at 2021-09-19T15:54:35.485.121.725 (XO: 24.485123726 - 24.485429727, t=37.2 degC)  Rank: 00085
   TM values: Max =   251  Min =  -277  Ave =    -4.55  Std =    28.50
   Min TM value:-277 at an offset of 98.450 usec -> 2021-09-19T15:54:35.485.220.175 (XO: 24.485222176)
   Max TM value:251 at an offset of 98.200 usec -> 2021-09-19T15:54:35.485.219.925 (XO: 24.485221926)
   Max dTM/dt [1/us]:2.25e+03 at an offset of 100.888 usec -> 2021-09-19T15:54:35.485.222.612 (XO: 24.485224614)
   Min dTM/dt [1/us]:-2.77e+03 at an offset of 98.325 usec -> 2021-09-19T15:54:35.485.220.050 (XO: 24.485222051)
