-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 14 11:24:14 2022 UTC
-- File 20210918_121407_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.338219 to 24.397176
-each separately from 6s to 24s and from 338219us to 397176us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.058379 in 1 packets: 18
MIU packet time from 2021-09-18T12:14:16.338.219.000 to 2021-09-18T12:14:34.397.176.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.058379 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.9505473 to 25.9504879
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.9999406 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210918_121407_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.2887674 to 8.2890564
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.0583190  -- OK for SL --
   MIU time from 2021-09-18T12:14:16.338.219.000 to 2021-09-18T12:14:16.338.508.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.058379

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.3473754 to 26.3476644
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-18T12:14:34.396.887.000 to 2021-09-18T12:14:34.397.176.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.288767400 corrected by 40bit 10MHz counter of 7.950547300 obtained at 1Hz counter of 6. Result: 6.338221216
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-18T12:14:16.338.078.725
   Max 1st sample MIU time among all SL packets: 2021-09-18T12:14:34.397.035.725
   Min 1st sample XO time among all SL packets: 6.338080941
   Max 1st sample XO time among all SL packets: 24.397037575
   TM values from -274 to 218 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-18T12:14:16.338.078.725 (XO: 6.338080941 - 6.338386942, t=37.2 degC)  Rank: 00081
   TM values: Max =   218  Min =  -267  Ave =    -4.69  Std =    26.17
   Min TM value:-267 at an offset of 123.875 usec -> 2021-09-18T12:14:16.338.202.600 (XO: 6.338204816)
   Max TM value:218 at an offset of 123.563 usec -> 2021-09-18T12:14:16.338.202.287 (XO: 6.338204504)
   Max dTM/dt [1/us]:2.08e+03 at an offset of 126.250 usec -> 2021-09-18T12:14:16.338.204.975 (XO: 6.338207191)
   Min dTM/dt [1/us]:-2.02e+03 at an offset of 123.675 usec -> 2021-09-18T12:14:16.338.202.400 (XO: 6.338204616)

 - Block number: 2 starting at 2021-09-18T12:14:34.396.746.725 (XO: 24.396748574 - 24.397054575, t=37.2 degC)  Rank: 00084
   TM values: Max =   204  Min =  -274  Ave =    -4.53  Std =    19.49
   Min TM value:-274 at an offset of 114.875 usec -> 2021-09-18T12:14:34.396.861.600 (XO: 24.396863449)
   Max TM value:204 at an offset of 114.575 usec -> 2021-09-18T12:14:34.396.861.300 (XO: 24.396863149)
   Max dTM/dt [1/us]:2.31e+03 at an offset of 117.263 usec -> 2021-09-18T12:14:34.396.863.987 (XO: 24.396865837)
   Min dTM/dt [1/us]:-2.29e+03 at an offset of 114.713 usec -> 2021-09-18T12:14:34.396.861.437 (XO: 24.396863287)
