-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan 14 01:18:02 2022 UTC
-- File 20210917_162137_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.493716 to 16.943811
-each separately from 12s to 16s and from 493716us to 943811us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.449517 in 1 packets: 18
MIU packet time from 2021-09-17T16:21:52.493.716.000 to 2021-09-17T16:21:56.943.811.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.449517 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.9049859 to 17.9049726
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      3.9999867 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      03 -> 51.1 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      2 in 36 packets (all packets)
Number of generated alerts: 
      2 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210917_162137_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.3987029 to 14.3989919
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.4495030  -- OK for SL --
   MIU time from 2021-09-17T16:21:52.493.716.000 to 2021-09-17T16:21:52.494.005.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.449517

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.8484949 to 18.8487839
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-17T16:21:56.943.522.000 to 2021-09-17T16:21:56.943.811.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.398702900 corrected by 40bit 10MHz counter of 13.904985900 obtained at 1Hz counter of 12. Result: 12.493718629
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-17T16:21:52.493.575.725
   Max 1st sample MIU time among all SL packets: 2021-09-17T16:21:56.943.670.725
   Min 1st sample XO time among all SL packets: 12.493578354
   Max 1st sample XO time among all SL packets: 16.943674139
   TM values from -247 to 218 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-17T16:21:52.493.575.725 (XO: 12.493578354 - 12.493884355, t=37.2 degC)  Rank: 00083
   TM values: Max =   218  Min =  -247  Ave =    -4.65  Std =    21.27
   Min TM value:-247 at an offset of 35.675 usec -> 2021-09-17T16:21:52.493.611.400 (XO: 12.493614029)
   Max TM value:218 at an offset of 38.288 usec -> 2021-09-17T16:21:52.493.614.012 (XO: 12.493616641)
   Max dTM/dt [1/us]:1.53e+03 at an offset of 38.175 usec -> 2021-09-17T16:21:52.493.613.900 (XO: 12.493616529)
   Min dTM/dt [1/us]:-2.12e+03 at an offset of 35.488 usec -> 2021-09-17T16:21:52.493.611.212 (XO: 12.493613841)

 - Block number: 2 starting at 2021-09-17T16:21:56.943.381.725 (XO: 16.943385138 - 16.943691139, t=37.2 degC)  Rank: 00078
   TM values: Max =   217  Min =  -238  Ave =    -4.66  Std =    22.38
   Min TM value:-238 at an offset of 120.738 usec -> 2021-09-17T16:21:56.943.502.462 (XO: 16.943505876)
   Max TM value:217 at an offset of 123.363 usec -> 2021-09-17T16:21:56.943.505.087 (XO: 16.943508501)
   Max dTM/dt [1/us]:1.95e+03 at an offset of 123.125 usec -> 2021-09-17T16:21:56.943.504.850 (XO: 16.943508264)
   Min dTM/dt [1/us]:-1.92e+03 at an offset of 180.976 usec -> 2021-09-17T16:21:56.943.562.700 (XO: 16.943566114)
