-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Jan 12 06:45:22 2022 UTC
-- File 20210915_231423_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.929129 to 83.678819
-each separately from 8s to 83s and from 21171us to 929418us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.557175 in 1 packets: 18
      15.937253 in 1 packets: 36
      15.357601 in 1 packets: 54
      7.238857 in 1 packets: 72
      19.657070 in 1 packets: 90
MIU packet time from 2021-09-15T23:14:34.929.129.000 to 2021-09-15T23:15:49.678.819.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.557175 in 1 packets: 18
      15.937253 in 1 packets: 36
      15.357601 in 1 packets: 54
      7.238857 in 1 packets: 72
      19.657070 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.9352450 to 84.9349955
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999435 in 1 packets: 18
      15.9999468 in 1 packets: 36
      14.9999501 in 1 packets: 54
      7.9999734 in 1 packets: 72
      18.9999367 in 1 packets: 90
Temperature FPGA: 
      04 -> 49.8 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      03 -> 51.1 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      05 -> 48.5 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C1 -> 37.2 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 72 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90, 91, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 72 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90, 91, ...
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20210915_231423_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.8643747 to 10.8646637
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.5571190  -- OK for SL --
   MIU time from 2021-09-15T23:14:34.929.129.000 to 2021-09-15T23:14:34.929.418.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.557175

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.4217827 to 27.4220717
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.9371990  -- OK for SL --
   MIU time from 2021-09-15T23:14:51.486.593.000 to 2021-09-15T23:14:51.486.882.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.937253

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.3592707 to 43.3595597
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.3575510  -- OK for SL --
   MIU time from 2021-09-15T23:15:07.424.135.000 to 2021-09-15T23:15:07.424.424.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.357601

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 58.7171107 to 58.7173997
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.2388310  -- OK for SL --
   MIU time from 2021-09-15T23:15:22.782.025.000 to 2021-09-15T23:15:22.782.314.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.238857

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 65.9562307 to 65.9565197
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.6570070  -- OK for SL --
   MIU time from 2021-09-15T23:15:30.021.171.000 to 2021-09-15T23:15:30.021.460.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.657070

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 85.6135267 to 85.6138157
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-15T23:15:49.678.530.000 to 2021-09-15T23:15:49.678.819.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 10.864374700 corrected by 40bit 10MHz counter of 9.935245000 obtained at 1Hz counter of 8. Result: 8.929132766
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-15T23:14:34.928.988.725
   Max 1st sample MIU time among all SL packets: 2021-09-15T23:15:49.678.678.725
   Min 1st sample XO time among all SL packets: 8.928992491
   Max 1st sample XO time among all SL packets: 83.678682165
   TM values from -273 to 244 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-15T23:14:34.928.988.725 (XO: 8.928992491 - 8.929298492, t=37.2 degC)  Rank: 00082
   TM values: Max =   231  Min =  -262  Ave =    -4.54  Std =    26.69
   Min TM value:-262 at an offset of 117.638 usec -> 2021-09-15T23:14:34.929.106.362 (XO: 8.929110129)
   Max TM value:231 at an offset of 117.375 usec -> 2021-09-15T23:14:34.929.106.100 (XO: 8.929109866)
   Max dTM/dt [1/us]:1.91e+03 at an offset of 120.063 usec -> 2021-09-15T23:14:34.929.108.787 (XO: 8.929112554)
   Min dTM/dt [1/us]:-2.58e+03 at an offset of 117.500 usec -> 2021-09-15T23:14:34.929.106.225 (XO: 8.929109991)

 - Block number: 2 starting at 2021-09-15T23:14:51.486.452.725 (XO: 25.486455579 - 25.486761580, t=37.2 degC)  Rank: 00075
   TM values: Max =   206  Min =  -233  Ave =    -4.53  Std =    18.21
   Min TM value:-233 at an offset of 112.738 usec -> 2021-09-15T23:14:51.486.565.462 (XO: 25.486568317)
   Max TM value:206 at an offset of 115.463 usec -> 2021-09-15T23:14:51.486.568.187 (XO: 25.486571042)
   Max dTM/dt [1/us]:1.79e+03 at an offset of 115.288 usec -> 2021-09-15T23:14:51.486.568.012 (XO: 25.486570867)
   Min dTM/dt [1/us]:-1.57e+03 at an offset of 112.600 usec -> 2021-09-15T23:14:51.486.565.325 (XO: 25.486568179)

 - Block number: 3 starting at 2021-09-15T23:15:07.423.994.725 (XO: 41.423996524 - 41.424302525, t=37.2 degC)  Rank: 00079
   TM values: Max =   204  Min =  -264  Ave =    -4.64  Std =    18.54
   Min TM value:-264 at an offset of 115.450 usec -> 2021-09-15T23:15:07.424.110.175 (XO: 41.424111975)
   Max TM value:204 at an offset of 115.175 usec -> 2021-09-15T23:15:07.424.109.900 (XO: 41.424111700)
   Max dTM/dt [1/us]:1.57e+03 at an offset of 117.913 usec -> 2021-09-15T23:15:07.424.112.637 (XO: 41.424114437)
   Min dTM/dt [1/us]:-2.00e+03 at an offset of 115.275 usec -> 2021-09-15T23:15:07.424.110.000 (XO: 41.424111800)

 - Block number: 4 starting at 2021-09-15T23:15:22.781.884.725 (XO: 56.781887605 - 56.782193606, t=37.2 degC)  Rank: 00078
   TM values: Max =   203  Min =  -265  Ave =    -4.56  Std =    22.93
   Min TM value:-265 at an offset of 124.400 usec -> 2021-09-15T23:15:22.782.009.125 (XO: 56.782012006)
   Max TM value:203 at an offset of 124.100 usec -> 2021-09-15T23:15:22.782.008.825 (XO: 56.782011706)
   Max dTM/dt [1/us]:1.94e+03 at an offset of 126.775 usec -> 2021-09-15T23:15:22.782.011.500 (XO: 56.782014381)
   Min dTM/dt [1/us]:-2.18e+03 at an offset of 124.200 usec -> 2021-09-15T23:15:22.782.008.925 (XO: 56.782011806)

 - Block number: 5 starting at 2021-09-15T23:15:30.021.030.725 (XO: 64.021031694 - 64.021337695, t=37.2 degC)  Rank: 00083
   TM values: Max =   244  Min =  -273  Ave =    -4.53  Std =    30.34
   Min TM value:-273 at an offset of 171.251 usec -> 2021-09-15T23:15:30.021.201.975 (XO: 64.021202945)
   Max TM value:244 at an offset of 127.988 usec -> 2021-09-15T23:15:30.021.158.712 (XO: 64.021159682)
   Max dTM/dt [1/us]:2.14e+03 at an offset of 130.638 usec -> 2021-09-15T23:15:30.021.161.362 (XO: 64.021162332)
   Min dTM/dt [1/us]:-2.60e+03 at an offset of 128.113 usec -> 2021-09-15T23:15:30.021.158.837 (XO: 64.021159807)

 - Block number: 6 starting at 2021-09-15T23:15:49.678.389.725 (XO: 83.678393164 - 83.678699165, t=37.2 degC)  Rank: 00077
   TM values: Max =   188  Min =  -237  Ave =    -4.51  Std =    20.02
   Min TM value:-237 at an offset of 112.888 usec -> 2021-09-15T23:15:49.678.502.612 (XO: 83.678506052)
   Max TM value:188 at an offset of 128.413 usec -> 2021-09-15T23:15:49.678.518.137 (XO: 83.678521577)
   Max dTM/dt [1/us]:1.63e+03 at an offset of 128.213 usec -> 2021-09-15T23:15:49.678.517.937 (XO: 83.678521377)
   Min dTM/dt [1/us]:-1.78e+03 at an offset of 112.675 usec -> 2021-09-15T23:15:49.678.502.400 (XO: 83.678505839)
