-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Jan 12 05:54:42 2022 UTC
-- File 20210915_224530_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.258944 to 54.709949
-each separately from 12s to 54s and from 129467us to 709949us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.899927 in 1 packets: 18
      24.970018 in 1 packets: 36
      10.579904 in 1 packets: 54
MIU packet time from 2021-09-15T22:45:45.258.944.000 to 2021-09-15T22:46:27.709.949.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.899927 in 1 packets: 18
      24.970018 in 1 packets: 36
      10.579904 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.9038156 to 55.9036762
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.9999768 in 1 packets: 18
      24.9999169 in 1 packets: 36
      9.9999669 in 1 packets: 54
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 36 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Temperature XO: 
      C1 -> 37.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20210915_224530_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.1627611 to 14.1630501
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.8999030  -- OK for SL --
   MIU time from 2021-09-15T22:45:45.258.944.000 to 2021-09-15T22:45:45.259.233.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.899927

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.0629531 to 21.0632421
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.9699350  -- OK for SL --
   MIU time from 2021-09-15T22:45:52.159.160.000 to 2021-09-15T22:45:52.159.449.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.970018

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.0331771 to 46.0334661
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.5798710  -- OK for SL --
   MIU time from 2021-09-15T22:46:17.129.467.000 to 2021-09-15T22:46:17.129.756.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.579904

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 56.6133371 to 56.6136261
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-15T22:46:27.709.660.000 to 2021-09-15T22:46:27.709.949.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.162761100 corrected by 40bit 10MHz counter of 13.903815600 obtained at 1Hz counter of 12. Result: 12.258946355
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-15T22:45:45.258.803.725
   Max 1st sample MIU time among all SL packets: 2021-09-15T22:46:27.709.808.725
   Min 1st sample XO time among all SL packets: 12.258806079
   Max 1st sample XO time among all SL packets: 54.709811967
   TM values from -285 to 242 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-15T22:45:45.258.803.725 (XO: 12.258806079 - 12.259112080, t=37.2 degC)  Rank: 00080
   TM values: Max =   233  Min =  -240  Ave =    -4.50  Std =    20.57
   Min TM value:-240 at an offset of 120.700 usec -> 2021-09-15T22:45:45.258.924.425 (XO: 12.258926779)
   Max TM value:233 at an offset of 123.338 usec -> 2021-09-15T22:45:45.258.927.062 (XO: 12.258929417)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 118.388 usec -> 2021-09-15T22:45:45.258.922.112 (XO: 12.258924467)
   Min dTM/dt [1/us]:-2.15e+03 at an offset of 120.525 usec -> 2021-09-15T22:45:45.258.924.250 (XO: 12.258926604)

 - Block number: 2 starting at 2021-09-15T22:45:52.159.019.725 (XO: 19.159020950 - 19.159326951, t=37.2 degC)  Rank: 00078
   TM values: Max =   242  Min =  -261  Ave =    -4.53  Std =    26.06
   Min TM value:-261 at an offset of 118.588 usec -> 2021-09-15T22:45:52.159.138.312 (XO: 19.159139538)
   Max TM value:242 at an offset of 105.725 usec -> 2021-09-15T22:45:52.159.125.450 (XO: 19.159126675)
   Max dTM/dt [1/us]:2.30e+03 at an offset of 103.688 usec -> 2021-09-15T22:45:52.159.123.412 (XO: 19.159124638)
   Min dTM/dt [1/us]:-2.40e+03 at an offset of 105.813 usec -> 2021-09-15T22:45:52.159.125.537 (XO: 19.159126763)

 - Block number: 3 starting at 2021-09-15T22:46:17.129.326.725 (XO: 44.129327952 - 44.129633953, t=37.2 degC)  Rank: 00076
   TM values: Max =   206  Min =  -258  Ave =    -4.50  Std =    20.12
   Min TM value:-258 at an offset of 105.738 usec -> 2021-09-15T22:46:17.129.432.462 (XO: 44.129433690)
   Max TM value:206 at an offset of 108.325 usec -> 2021-09-15T22:46:17.129.435.050 (XO: 44.129436277)
   Max dTM/dt [1/us]:1.87e+03 at an offset of 108.163 usec -> 2021-09-15T22:46:17.129.434.887 (XO: 44.129436115)
   Min dTM/dt [1/us]:-1.79e+03 at an offset of 105.513 usec -> 2021-09-15T22:46:17.129.432.237 (XO: 44.129433465)

 - Block number: 4 starting at 2021-09-15T22:46:27.709.519.725 (XO: 54.709522966 - 54.709828967, t=37.2 degC)  Rank: 00081
   TM values: Max =   230  Min =  -285  Ave =    -4.47  Std =    32.46
   Min TM value:-285 at an offset of 164.113 usec -> 2021-09-15T22:46:27.709.683.838 (XO: 54.709687079)
   Max TM value:230 at an offset of 107.450 usec -> 2021-09-15T22:46:27.709.627.175 (XO: 54.709630417)
   Max dTM/dt [1/us]:2.18e+03 at an offset of 110.113 usec -> 2021-09-15T22:46:27.709.629.837 (XO: 54.709633079)
   Min dTM/dt [1/us]:-2.54e+03 at an offset of 107.575 usec -> 2021-09-15T22:46:27.709.627.300 (XO: 54.709630542)
