-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Jan 12 04:23:06 2022 UTC
-- File 20210915_215820_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.423224 to 26.062981
-each separately from 3s to 26s and from 62692us to 423513us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.639179 in 1 packets: 18
MIU packet time from 2021-09-15T21:58:26.423.224.000 to 2021-09-15T21:58:49.062.981.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.639179 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.8930194 to 27.8929429
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.9999235 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210915_215820_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.3162445 to 5.3165335
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.6391030  -- OK for SL --
   MIU time from 2021-09-15T21:58:26.423.224.000 to 2021-09-15T21:58:26.423.513.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.639179

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.9556365 to 27.9559255
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-15T21:58:49.062.692.000 to 2021-09-15T21:58:49.062.981.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.316244500 corrected by 40bit 10MHz counter of 4.893019400 obtained at 1Hz counter of 3. Result: 3.423226497
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-15T21:58:26.423.083.725
   Max 1st sample MIU time among all SL packets: 2021-09-15T21:58:49.062.840.725
   Min 1st sample XO time among all SL packets: 3.423086221
   Max 1st sample XO time among all SL packets: 26.062842532
   TM values from -277 to 244 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-15T21:58:26.423.083.725 (XO: 3.423086221 - 3.423392222, t=37.2 degC)  Rank: 00081
   TM values: Max =   238  Min =  -273  Ave =    -4.56  Std =    25.97
   Min TM value:-273 at an offset of 128.300 usec -> 2021-09-15T21:58:26.423.212.025 (XO: 3.423214522)
   Max TM value:238 at an offset of 128.038 usec -> 2021-09-15T21:58:26.423.211.762 (XO: 3.423214259)
   Max dTM/dt [1/us]:2.59e+03 at an offset of 130.675 usec -> 2021-09-15T21:58:26.423.214.400 (XO: 3.423216897)
   Min dTM/dt [1/us]:-2.67e+03 at an offset of 128.150 usec -> 2021-09-15T21:58:26.423.211.875 (XO: 3.423214372)

 - Block number: 2 starting at 2021-09-15T21:58:49.062.551.725 (XO: 26.062553531 - 26.062859532, t=37.2 degC)  Rank: 00081
   TM values: Max =   244  Min =  -277  Ave =    -4.56  Std =    32.13
   Min TM value:-277 at an offset of 189.326 usec -> 2021-09-15T21:58:49.062.741.050 (XO: 26.062742857)
   Max TM value:244 at an offset of 173.413 usec -> 2021-09-15T21:58:49.062.725.138 (XO: 26.062726944)
   Max dTM/dt [1/us]:2.16e+03 at an offset of 122.125 usec -> 2021-09-15T21:58:49.062.673.850 (XO: 26.062675657)
   Min dTM/dt [1/us]:-2.65e+03 at an offset of 119.575 usec -> 2021-09-15T21:58:49.062.671.300 (XO: 26.062673107)
