-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed Jan 12 04:19:44 2022 UTC
-- File 20210915_215547_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.826223 to 65.505379
-each separately from 11s to 65s and from 45673us to 976165us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.149364 in 1 packets: 18
      15.069508 in 1 packets: 36
      11.229347 in 1 packets: 54
      11.229492 in 1 packets: 72
MIU packet time from 2021-09-15T21:56:01.826.223.000 to 2021-09-15T21:56:55.505.379.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.149364 in 1 packets: 18
      15.069508 in 1 packets: 36
      11.229347 in 1 packets: 54
      11.229492 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.8966226 to 66.8964432
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999468 in 1 packets: 18
      15.9999469 in 1 packets: 36
      10.9999634 in 1 packets: 54
      10.9999635 in 1 packets: 72
Temperature FPGA: 
      04 -> 49.8 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 71 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 71 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20210915_215547_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.7228460 to 13.7231350
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.1493110  -- OK for SL --
   MIU time from 2021-09-15T21:56:01.826.223.000 to 2021-09-15T21:56:01.826.512.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.149364

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.8724460 to 29.8727350
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0694550  -- OK for SL --
   MIU time from 2021-09-15T21:56:17.975.876.000 to 2021-09-15T21:56:17.976.165.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.069508

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 44.9421900 to 44.9424790
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.2293110  -- OK for SL --
   MIU time from 2021-09-15T21:56:33.045.673.000 to 2021-09-15T21:56:33.045.962.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.229347

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 56.1717900 to 56.1720790
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.2294550  -- OK for SL --
   MIU time from 2021-09-15T21:56:44.275.309.000 to 2021-09-15T21:56:44.275.598.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.229492

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 67.4015340 to 67.4018230
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-15T21:56:55.505.090.000 to 2021-09-15T21:56:55.505.379.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 13.722846000 corrected by 40bit 10MHz counter of 12.896622600 obtained at 1Hz counter of 11. Result: 11.826226127
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-15T21:56:01.826.082.725
   Max 1st sample MIU time among all SL packets: 2021-09-15T21:56:55.505.238.725
   Min 1st sample XO time among all SL packets: 11.826085851
   Max 1st sample XO time among all SL packets: 65.505241243
   TM values from -258 to 261 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-15T21:56:01.826.082.725 (XO: 11.826085851 - 11.826391852, t=37.2 degC)  Rank: 00082
   TM values: Max =   233  Min =  -258  Ave =    -4.53  Std =    25.85
   Min TM value:-258 at an offset of 115.700 usec -> 2021-09-15T21:56:01.826.198.425 (XO: 11.826201551)
   Max TM value:233 at an offset of 118.225 usec -> 2021-09-15T21:56:01.826.200.950 (XO: 11.826204076)
   Max dTM/dt [1/us]:2.84e+03 at an offset of 118.050 usec -> 2021-09-15T21:56:01.826.200.775 (XO: 11.826203901)
   Min dTM/dt [1/us]:-2.27e+03 at an offset of 115.500 usec -> 2021-09-15T21:56:01.826.198.225 (XO: 11.826201351)

 - Block number: 2 starting at 2021-09-15T21:56:17.975.735.725 (XO: 27.975739545 - 27.976045546, t=37.2 degC)  Rank: 00084
   TM values: Max =   261  Min =  -256  Ave =    -4.54  Std =    20.58
   Min TM value:-256 at an offset of 102.413 usec -> 2021-09-15T21:56:17.975.838.137 (XO: 27.975841958)
   Max TM value:261 at an offset of 100.300 usec -> 2021-09-15T21:56:17.975.836.025 (XO: 27.975839845)
   Max dTM/dt [1/us]:2.20e+03 at an offset of 104.825 usec -> 2021-09-15T21:56:17.975.840.550 (XO: 27.975844370)
   Min dTM/dt [1/us]:-2.29e+03 at an offset of 102.325 usec -> 2021-09-15T21:56:17.975.838.050 (XO: 27.975841870)

 - Block number: 3 starting at 2021-09-15T21:56:33.045.532.725 (XO: 43.045533575 - 43.045839576, t=37.2 degC)  Rank: 00076
   TM values: Max =   221  Min =  -248  Ave =    -4.55  Std =    23.27
   Min TM value:-248 at an offset of 101.663 usec -> 2021-09-15T21:56:33.045.634.387 (XO: 43.045635238)
   Max TM value:221 at an offset of 101.325 usec -> 2021-09-15T21:56:33.045.634.050 (XO: 43.045634901)
   Max dTM/dt [1/us]:1.94e+03 at an offset of 104.000 usec -> 2021-09-15T21:56:33.045.636.725 (XO: 43.045637576)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 101.413 usec -> 2021-09-15T21:56:33.045.634.137 (XO: 43.045634988)

 - Block number: 4 starting at 2021-09-15T21:56:44.275.168.725 (XO: 54.275170933 - 54.275476934, t=37.2 degC)  Rank: 00080
   TM values: Max =   238  Min =  -219  Ave =    -4.57  Std =    21.18
   Min TM value:-219 at an offset of 125.938 usec -> 2021-09-15T21:56:44.275.294.662 (XO: 54.275296871)
   Max TM value:238 at an offset of 128.638 usec -> 2021-09-15T21:56:44.275.297.362 (XO: 54.275299571)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 123.650 usec -> 2021-09-15T21:56:44.275.292.375 (XO: 54.275294583)
   Min dTM/dt [1/us]:-2.00e+03 at an offset of 113.700 usec -> 2021-09-15T21:56:44.275.282.425 (XO: 54.275284633)

 - Block number: 5 starting at 2021-09-15T21:56:55.504.949.725 (XO: 65.504952242 - 65.505258243, t=37.2 degC)  Rank: 00078
   TM values: Max =   203  Min =  -242  Ave =    -4.63  Std =    26.48
   Min TM value:-242 at an offset of 124.625 usec -> 2021-09-15T21:56:55.505.074.350 (XO: 65.505076867)
   Max TM value:203 at an offset of 127.313 usec -> 2021-09-15T21:56:55.505.077.037 (XO: 65.505079555)
   Max dTM/dt [1/us]:1.83e+03 at an offset of 127.175 usec -> 2021-09-15T21:56:55.505.076.900 (XO: 65.505079417)
   Min dTM/dt [1/us]:-1.98e+03 at an offset of 112.450 usec -> 2021-09-15T21:56:55.505.062.175 (XO: 65.505064692)
