-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Jan 11 23:35:31 2022 UTC
-- File 20210915_162336_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.844088 to 17.724343
-each separately from 3s to 17s and from 724054us to 844377us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.879677 in 1 packets: 18
MIU packet time from 2021-09-15T16:23:42.844.088.000 to 2021-09-15T16:23:56.724.343.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.879677 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.7405148 to 18.7404683
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999535 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets (all packets)
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      1 in 1 packets: 1
      2 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      3 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      3 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20210915_162336_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.5846033 to 5.5848923
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.8796310  -- OK for SL --
   MIU time from 2021-09-15T16:23:42.844.088.000 to 2021-09-15T16:23:42.844.377.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.879677

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.4645233 to 19.4648123
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2021-09-15T16:23:56.724.054.000 to 2021-09-15T16:23:56.724.343.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.584603300 corrected by 40bit 10MHz counter of 4.740514800 obtained at 1Hz counter of 3. Result: 3.844091370
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2021-09-15T16:23:42.843.947.725
   Max 1st sample MIU time among all SL packets: 2021-09-15T16:23:56.724.202.725
   Min 1st sample XO time among all SL packets: 3.843951094
   Max 1st sample XO time among all SL packets: 17.724206115
   TM values from -249 to 248 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2021-09-15T16:23:42.843.947.725 (XO: 3.843951094 - 3.844257095, t=37.2 degC)  Rank: 00086
   TM values: Max =   243  Min =  -249  Ave =    -4.52  Std =    22.45
   Min TM value:-249 at an offset of 112.950 usec -> 2021-09-15T16:23:42.844.060.675 (XO: 3.844064045)
   Max TM value:243 at an offset of 127.388 usec -> 2021-09-15T16:23:42.844.075.112 (XO: 3.844078482)
   Max dTM/dt [1/us]:2.23e+03 at an offset of 127.263 usec -> 2021-09-15T16:23:42.844.074.987 (XO: 3.844078357)
   Min dTM/dt [1/us]:-2.05e+03 at an offset of 124.525 usec -> 2021-09-15T16:23:42.844.072.250 (XO: 3.844075620)

 - Block number: 2 starting at 2021-09-15T16:23:56.723.913.725 (XO: 17.723917114 - 17.724223115, t=37.2 degC)  Rank: 00083
   TM values: Max =   248  Min =  -236  Ave =    -4.57  Std =    21.25
   Min TM value:-236 at an offset of 122.563 usec -> 2021-09-15T16:23:56.724.036.287 (XO: 17.724039677)
   Max TM value:248 at an offset of 125.225 usec -> 2021-09-15T16:23:56.724.038.950 (XO: 17.724042339)
   Max dTM/dt [1/us]:1.97e+03 at an offset of 125.100 usec -> 2021-09-15T16:23:56.724.038.825 (XO: 17.724042214)
   Min dTM/dt [1/us]:-2.13e+03 at an offset of 122.350 usec -> 2021-09-15T16:23:56.724.036.075 (XO: 17.724039464)
